Combined acquisition technique (CAT) for high-field neuroimaging with reduced RF power
- 142 Downloads
Clinical 3 T MRI systems are rapidly increasing and MRI systems with a static field of 7 T or even more have been installed. The RF power deposition is proportional to the square of the static magnetic field strength and is characterized by the specific absorption rate (SAR). Therefore, there exist defined safety limits to avoid heating of the patient. Here, we describe a hybrid method to significantly reduce the SAR compared to a turbo-spin-echo (TSE) sequence.
Materials and methods
We investigate the potential benefits of a combined acquisition technique (CAT) for high-field neuroimaging at 3 and 7 T. The TSE/EPI CAT experiments were performed on volunteers and patients and compared with standard TSE and GRASE protocols. Problems and solutions regarding T2 weighted CAT imaging are discussed.
We present in vivo images with T2 and proton density contrast obtained on 3 and 7 T with significant SAR reduction (up to 60 %) compared with standard TSE. Image quality is comparable to TSE but CAT shows fewer artifacts than a GRASE sequence.
CAT is a promising candidate for neuroimaging at high fields up to 7 T. The SAR reduction allows one to shorten the waiting time between two excitations or to image more slices thereby reducing the overall measurement time.
KeywordsHybrid imaging CAT SAR Dual bandwidth
The authors thank Melisa Okanovic for providing the GRASE sequence. This work was funded by the German Research Association (DFG JA 827/9-1). Technical support was supplied by Siemens Medical, Erlangen, Germany. We thank the Leibniz Institute for Neurobiology in Magdeburg for access to the 7 T.
- 2.Hennig J (1988) Multiecho imaging sequences with low refocusing flip angles. JMR 78(3):397–407Google Scholar
- 7.Hillenbrand C, Hahn D, Haase A, Jakob PM (2000) MR CAT scan: a modular approach for hybrid imaging. Magn Reson Mater Phy 10(3):183–199Google Scholar
- 9.Mugler JP III, Brookeman JR (2004) Efficient spatially-selective single-slab 3D turbo-spin-echo imaging. In: Proceedings of the 12th scientific meeting, international society for magnetic resonance in medicine, Kyoto, p 695Google Scholar