Temperature change during non-contact diode laser irradiation of implant surfaces
- 410 Downloads
A temperature increase of more than 10°C can compromise bone vitality. Laser radiation with different wavelengths has been used for the treatment of peri-implantitis, but little is known about the effect of laser irradiation on temperature rise on the implant surface. In this study, the temperature gradient (∆T) generated by laser irradiation of implant surface using two diode lasers (810 nm and a 980 nm) with 2 W of power has been recorded by two thermocouples (one in the cervical area and one in the apical area) and studied. The 810-nm diode laser showed the following results: after 60 s of irradiation with 2 W of continuous mode the temperature gradient in the cervical area of the implant (∆Tc) was 37.2°C, while in the apical area (∆Ta) was 27.2°C. The 980-nm diode laser showed the following results: after 60 s of irradiation with 2 W continuous mode ∆Tc was 41.1°C, and ∆Ta was 30.6°C. The 810-nm diode laser with 2 W continuous mode generated a temperature increase of 10°C after only 14 s. The 980-nm diode lasers groups produced a much more rapid temperature increase. In only 12 s, the continuous wave of 980 nm reached the 10°C temperature rise. From the present in vitro study it was concluded that the irradiation of implant surfaces with diode lasers may produce a temperature increase above the critical threshold (10°C ) after only 10 s.
KeywordsDental implants Diode laser Laser irradiation Temperature increase
The author would like to thank Dr. Carlo Ercoli (Division of Prosthodontics), for providing the machines for temperature measurements for this in vitro study.
- 20.Chu RT, Watanabe L, White JM, et al (1992) Temperature rise and surface modification of lased titanium cylinders. J Dent Res 71144, special issue, abstract no 312Google Scholar