In vitro validation of near-infrared transillumination at 780 nm for the detection of caries on proximal surfaces
- 27 Downloads
The aim of the present in vitro study was to determine the ability of Diagnocam to detect caries at an early stage and to compare it with digital radiography.
Materials and methods
One hundred twenty proximal surfaces composed equally of sound and decayed surfaces were evaluated by assessing images captured with Diagnocam (Kavo, Biberach, Germany) and digital radiography (DR). All images were assessed twice by two calibrated dentists with a minimum interval of 2 weeks between examinations. The results were compared with μCT scans.
Inter-rater reliability showed nearly perfect agreement; a high intra-rater reliability was calculated. Spearman’s rank correlation coefficients showed a strong correlation of Diagnocam and μCT (0.82). DR and μCT achieved a slightly lower correlation (0.73). The surfaces were categorized into sound surfaces, enamel lesions, and dentin lesions to determine intraclass correlation coefficients (ICCs), sensitivity and specificity. ICCs between μCT and DR ranged from 0.20 to 0.63. ICCs between Diagnocam and DR were higher in all categories and ranged from 0.56 to 0.83. Sensitivity, specificity, and accuracy of Diagnocam achieved mostly higher values than DR. In detecting enamel lesions, sensitivity was 0.36 for DR and 0.59 for the DC. The areas under the receiver operating characteristic (ROC) curves of Diagnocam were larger than those of DR in all categories.
Diagnocam is more capable of detecting initial proximal lesions than DR and also has a higher sensitivity for dentin lesions. However, caries progression in dentin cannot be reliably determined with Diagnocam.
Diagnocam may be suitable as a supplement to X-ray diagnostics in clinical use.
KeywordsTransillumination Near-infrared light Proximal caries Caries detection X-ray microtomography In vitro
The work was supported by the Department of Conservative Dentistry and Periodontology of the University Hospital of the LMU Munich.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors. All experimental procedures were approved by the Ethics Committee of the Faculty of Medicine of the Ludwig Maximilians University in Munich, Germany (488-15 UE).
For this type of study, formal consent is not required.
- 2.Poorterman JH, Aartman IH, Kieft JA, Kalsbeek H (2002) Klinische onderschatting van de prevalentie van approximale dentinelaesies en inadequate restauraties (Clinical underestimation of the prevalence of approximal dentin lesions and inadequate restorations). Ned Tijdschr Tandheelkd 109(2):47–50Google Scholar
- 8.Abdelaziz M, Krejci I (2015) DIAGNOcam--a near infrared digital imaging transillumination (NIDIT) technology. Int J Esthet Dent 10(1):158–165Google Scholar
- 17.Banting D, Eggertsson H, Ekstrand K, Zandoná AF, Ismail A, Longbottom C, Pitts N, Reich E, Ricketts D, Selwitz R (2005) Rationale and evidence for the international caries detection and assessment system (ICDAS II). Ann Arbor 1001:48109–41078Google Scholar
- 18.Marthaler TM (1966) A standardized system of recording dental conditions. Helv Odontol Acta 10(1):1–18Google Scholar
- 21.Schindelin J, Arganda-Carreras I, Frise E, Kaynig V, Longair M, Pietzsch T, Preibisch S, Rueden C, Saalfeld S, Schmid B, Tinevez J-Y, White DJ, Hartenstein V, Eliceiri K, Tomancak P, Cardona A (2012) Fiji: an open-source platform for biological-image analysis. Nat Methods 9(7):676–682 https://www.nature.com/articles/nmeth.2019#supplementary-information. Accessed 21 Jan 2019
- 22.Kunzelmann KH (2012) ImageJ I/O utilities. http://www.kunzelmann.de/6_software-imagej-import-export-utilities.html. Accessed 8 August 2018
- 26.Ozkan G, Kanli A, Baseren NM, Arslan U, Tatar I (2015) Validation of micro-computed tomography for occlusal caries detection: an in vitro study. Braz Oral Res 29(1):S1806-83242015000100309. https://doi.org/10.1590/1807-3107BOR-2015.vol29.0132 CrossRefGoogle Scholar
- 28.Haiter-Neto F, dos Anjos Pontual A, Frydenberg M, Wenzel A (2008) Detection of non-cavitated approximal caries lesions in digital images from seven solid-state receptors with particular focus on task-specific enhancement filters. An ex vivo study in human teeth. Clin Oral Investig 12(3):217–223. https://doi.org/10.1007/s00784-007-0173-5 CrossRefGoogle Scholar
- 33.Moritz A, Beer F (2006) Orale Lasertherapie. Quintessenz-Verlag, BerlinGoogle Scholar