Effect of multiple autoclave cycles on the surface roughness of HyFlex CM and HyFlex EDM files: an atomic force microscopy study
- 111 Downloads
To compare the effect of autoclave cycles on the surface topography and roughness of HyFlex CM and HyFlex EDM instruments using atomic force microscopy (AFM) analysis.
Eight new files of each brand were subdivided into four subgroups (n = 2/each subgroup). One group was allocated as the control group and not subjected to autoclave sterilization. The other three groups were subjected to different numbers (1, 5, and 10) of autoclave sterilization cycles. After the cycle instruments were subjected to AFM analysis. Roughness average (Ra) and the root mean square (RMS) values were chosen to investigate the surface features of endodontic files. The data was analyzed using one-way ANOVA and post hoc Tamhane tests at 5% significant level.
The lowest Ra and RMS values were observed in the HyFlex EDM files that served as the control and in those subjected to a single cycle of autoclave sterilization (P < 0.05). The highest Ra and RMS values were observed in the HyFlex CM and HyFlex EDM files that were subjected to 10 cycles of autoclave sterilization (P < 0.05). The surface roughness values of the HyFlex CM group showed a significant increase after ten autoclave cycles, whereas those of the HyFlex EDM group exhibited a significant change after five autoclave cycles (P < 0.05).
Although the initial surface roughness values of the HyFlex EDM files were lower than those of the HyFlex CM files, the surface roughness values of the EDM files showed a statistically significant increase after 5 cycles of autoclave sterilization. In contrast, the surface roughness values of the HyFlex CM files did not increase until 10 cycles of autoclave sterilization.
Present study indicated that autoclave sterilization negatively affected the surface roughness of the tested NiTi files.
KeywordsAFM Autoclave sterilization HyFlex CM HyFlex EDM Nickel–titanium
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
For this type of study, formal consent is not required.
- 9.Daneshmand S, Kahrizi EF, Abedi E, Abdolhosseini MM (2013) Influence of machining parameters on electro discharge machining of NiTi shape memory alloys. Int J Electrochem Sci 8:3095–3104Google Scholar
- 21.ColteneEndo HyFlex EDM Brochure. Available at: https://www.coltene.com/fileadmin/Data/EN/Products/Endodontics/Root_Canal_Shaping/HyFlex_EDM/31328A_HyFlexEDM_Brochure_US.pdf. Accessed 15 June 2017
- 22.ColteneEndo HyFlex CM Brochure. Available at: http://www.hyflexcm.com/DevDownloads/HyFlexFamily_bro.pdf. Accessed 15 June 2017
- 31.Iacono F, Pirani C, Generali L, Sassatelli P, Nucci C, Gandolfi MG, Prati C (2016) Wear analysis and cyclic fatigue resistance of electro discharge machined NiTi rotary instruments. G Ital Endod 30:64–68Google Scholar