Fracture load and failure types of different veneered polyetheretherketone fixed dental prostheses
- 833 Downloads
The aim of this study is to investigate the fracture load of different veneered PEEK 3-unit fixed dental prosthesis (FDPs) after different aging regimens.
Congruently anatomically shaped 3-unit FDPs were milled using a master stl-data set and randomly divided into four groups (N = 120, n = 30 per veneering group), which were veneered using different veneering methods: (i) digital veneering with breCAM.HIPC, (ii) conventional veneering with crea.lign, (iii) conventional with crea.lign paste, and (iv) using pre-manufactured veneers visio.lign. The FDPs were then adhesively cemented on a metal abutment and fracture loads were measured in a universal testing machine (1 mm/min) before and after aging (10,000 thermal cycles, 5/55 °C). Two- and one-way ANOVA followed by post hoc Scheffé tests were used for data analysis (p < 0.05).
This investigation showed an influence of the veneering method on the fracture load results independent of the aging level. The highest fracture load was measured for the FDPs with digital veneering (1882 ± 152 N at baseline, 2021 ± 184 N after thermocycling). The remaining groups showed comparable results, and no impact of thermal aging was observed. Digital and conventional veneers showed cracks in the pontic region starting from the connector area as a main failure type after loading, while the pre-manufactured veneers showed predominantly adhesive failures.
The digital veneering method showed the highest fracture load resistance. Thermal aging showed no impact on the fracture load of all tested veneered PEEK 3-unit FDPs.
According to this study results, reliable veneering of PEEK FDPs can be achieved with digital veneering.
KeywordsFracture load PEEK Digital veneering Veneering resin composite
The authors would like to thank bredent for material support of this study.
Compliance with ethical standards
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards.
Conflict of interest
The authors declare that they have no conflict of interest.
This study received no funding.
This article does not contain any studies with human participants or animals performed by any of the authors.
For this type of study, formal consent is not required.
- 1.Salamone JC (1996) Polymeric materials encyclopedia 4. CRC PressGoogle Scholar
- 2.Ebnesajjad S (2013) Handbook ofpolymer applications in medicine and medical devices. William Andrew PublishingGoogle Scholar
- 7.Klingler JH, Krüger MT, Sircar R, Kogias E, Scholz C, Volz F, Scheiwe C, Hubbe U, et al. (2014) PEEK cages versus PMMA spacers in anterior cervical discectomy: comparison of fusion, subsidence, sagittal alignment, and clinical outcome with a minimum 1-year follow-up. Sci World J. 398396Google Scholar
- 23.Stawarczyk B, Thrun H, Eichberger M, Roos M, Edelhoff D, Schweiger J, Schmidlin PR, et al. (2015) Effect of different surface pretreatments and adhesives on the load-bearing capacity of veneered 3-unit PEEK FDPs. J Prosthet Dent 114:666–673Google Scholar