Can the radiopaque marker in surgical swabs scratch orthopaedic implant surfaces?
- 17 Downloads
To determine whether the radiopaque marker strip, which is woven in surgical swabs, causes measureable wear on metal implants at pressures typically used to wipe off fluid from their surface.
Materials and methods
Finger pressure used to wipe a surface was measured and used as a reference pressure for further testing. A tribological wear rig was then used to analyse the wear caused on polished titanium plates by a cobalt chromium pin (the control test), the pin covered by a surgical swab and the pin covered by a radiopaque marker strip.
It was found that the cotton part or the radiopaque marker of surgical swabs on polished medical grade titanium plates caused no significant wear. In contrast severe scratching was observed from the cobalt chromium pin on its own.
To our knowledge, this is the first study in the literature analysing the wear caused by the surgical swabs and radiopaque strip on metal implants. The results suggest that surgical swabs are safe to use on metallic implants at pressures typical of a wiping motion.
KeywordsSwab Wear Radiopaque marker X-ray Implant Metal Titanium
This work was supported by the NIHR (National Institute for Health Research) through funding of the LMBRU (Leeds Musculoskeletal Biomedical Research Unit).
Compliance with ethical standards
Conflict of interest
The authors declare that they have no competing interests.
- 2.Sadeghifar AR, Saeed AR (2013) Infected textiloma, 35 years after the operation for femur fracture, an extremely rare occurrence. Arch Bone Jt Surg 1:41–43Google Scholar
- 4.Sansone V, Pagani D, Melato M (2013) The effects on bone cells of metal ions released from orthopaedic implants. A review. Clin Cases Miner Bone Metab 10:34–40Google Scholar
- 5.Archard Equation. Wikipedia. https://en.wikipedia.org/wiki/Archard_equation. Accessed 18 Aug 2018
- 8.Billi F, Campbell P (2010) Nanotoxicology of metal wear particles in total joint arthroplasty: a review of current concepts. J Appl Biomater Biomech 8:1–6Google Scholar
- 15.Harris WH (1995) The problem is osteolysis. Clin Orthop Relat Res 311:46–53Google Scholar