C1–C2 posterior screw fixation in atlantoaxial fractures revisited: technical update based on 127 cases
Posterior fusion of traumatic odontoid fractures by C1 lateral mass and C2 isthmic screws (modified Harms–Goel technique) is a viable alternative to transarticular screw fixation due to its universal applicability. This retrospective study reports on a series of 127 patients.
Material and methods
Our clinical database was screened for patients with fractures of the upper cervical spine incorporating a C2 fracture, operated on between 2007 and 2015. Patients were included if fused by internal fixation via C1 lateral mass screws, C2 isthmic screws in freehand technique under lateral fluoroscopy. Screw placement was controlled postoperatively by computed tomography and rated using the Gertzbein & Robbins classification. Surgery-related complications, consecutive treatment, revision surgeries and duration of surgeries were registered.
In total, 127 patients were identified with altogether 572 screws. Correct screw positions of grade A and B according to Gertzbein & Robbins were achieved in a total of 539 (94.2%) screws (grade A: 453 (79.2%); grade B: 86 (15%)), grade C screw malpositions noted in 21 (3.7%), grade D in 10 (1.7%) and grade E in 2 (0.3%) screws. Vertebral artery canal breaches occurred in 29 screws (5.1%), with vertebral artery occlusion in 4 patients. Coiling of injured vertebral artery had to be performed in one patient. None of these patients suffered clinically apparent cerebrovascular complications. Revision surgery was performed in 8 patients (6%).
Posterior fixation of atlantoaxial fractures by C1 lateral mass and C2 isthmic screws with fluoroscopy without navigation is a safe and feasible method but not free of risk of vertebral artery injuries.
These slides can be retrieved under Electronic Supplementary Material.
KeywordsAtlantoaxial fracture Posterior screw fixation Accuracy C1–C2 Odontoid fracture
Compliance with ethical standards
Conflict of interest
All authors state that they have no conflicts of interest.
- 3.Chapman J, Smith JS, Kopjar B, Vaccaro AR, Arnold P, Shaffrey CI, Fehlings MG (2013) The AOSpine North America geriatric odontoid fracture mortality study: a retrospective review of mortality outcomes for operative versus nonoperative treatment of 322 patients with long-term follow-up. Spine 38(13):1098–1104. https://doi.org/10.1097/BRS.0b013e318286f0cf CrossRefPubMedPubMedCentralGoogle Scholar
- 4.Vaccaro AR, Kepler CK, Kopjar B, Chapman J, Shaffrey C, Arnold P, Gokaslan Z, Brodke D, France J, Dekutoski M, Sasso R, Yoon ST, Bono C, Harrop J, Fehlings MG (2013) Functional and quality-of-life outcomes in geriatric patients with type-II dens fracture. J Bone Joint Surg Am 95(8):729–735. https://doi.org/10.2106/JBJS.K.01636 CrossRefPubMedGoogle Scholar
- 5.Ryang YM, Torok E, Janssen I, Reinke A, Buchmann N, Gempt J, Ringel F, Meyer B (2016) Early morbidity and mortality in 50 very elderly patients after posterior atlantoaxial fusion for traumatic odontoid fractures. World Neurosurg 87:381–391. https://doi.org/10.1016/j.wneu.2015.12.026 CrossRefPubMedGoogle Scholar
- 8.Magerl F, Seemann P-S (1987) Stable posterior fusion of the atlas and axis by transarticular screw fixation. In: Cervical spine I. Springer, pp 322–327Google Scholar
- 17.Gautschi OP, Payer M, Corniola MV, Smoll NR, Schaller K, Tessitore E (2014) Clinically relevant complications related to posterior atlanto-axial fixation in atlanto-axial instability and their management. Clin Neurol Neurosurg 123:131–135. https://doi.org/10.1016/j.clineuro.2014.05.020 CrossRefPubMedGoogle Scholar
- 18.Akinduro OO, Baum GR, Howard BM, Pradilla G, Grossberg JA, Rodts GE Jr, Ahmad FU (2016) Neurological outcomes following iatrogenic vascular injury during posterior atlanto-axial instrumentation. Clin Neurol Neurosurg 150:110–116. https://doi.org/10.1016/j.clineuro.2016.08.012 CrossRefPubMedGoogle Scholar
- 20.Scheufler KM, Franke J, Eckardt A, Dohmen H (2011) Accuracy of image-guided pedicle screw placement using intraoperative computed tomography-based navigation with automated referencing, part I: cervicothoracic spine. Neurosurgery 69(4):782–795. https://doi.org/10.1227/NEU.0b013e318222ae16 CrossRefPubMedGoogle Scholar
- 30.Elliott RE, Tanweer O, Boah A, Morsi A, Ma T, Frempong-Boadu A, Smith ML (2014) Outcome comparison of atlantoaxial fusion with transarticular screws and screw-rod constructs: meta-analysis and review of literature. J Spinal Disord Tech 27(1):11–28. https://doi.org/10.1097/BSD.0b013e318277da19 CrossRefPubMedGoogle Scholar
- 31.Costa F, Ortolina A, Attuati L, Cardia A, Tomei M, Riva M, Balzarini L, Fornari M (2015) Management of C1–2 traumatic fractures using an intraoperative 3D imaging-based navigation system. Journal of neurosurgery Spine 22(2):128–133. https://doi.org/10.3171/2014.10.SPINE14122 CrossRefPubMedGoogle Scholar
- 33.Yang YL, Zhou DS, He JL (2013) Comparison of isocentric C-arm 3-dimensional navigation and conventional fluoroscopy for C1 lateral mass and C2 pedicle screw placement for atlantoaxial instability. J Spinal Disord Tech 26(3):127–134. https://doi.org/10.1097/BSD.0b013e31823d36b6 CrossRefPubMedGoogle Scholar
- 35.Czabanka M, Haemmerli J, Hecht N, Foehre B, Arden K, Liebig T, Woitzik J, Vajkoczy P (2017) Spinal navigation for posterior instrumentation of C1–2 instability using a mobile intraoperative CT scanner. J Neurosurg Spine 27(3):268–275. https://doi.org/10.3171/2017.1.SPINE16859 CrossRefPubMedGoogle Scholar
- 36.Akins PT, Harris J, Alvarez JL, Chen Y, Paxton EW, Bernbeck J, Guppy KH (2015) Risk factors associated with 30-day readmissions after instrumented spine surgery in 14,939 patients: 30-day readmissions after instrumented spine surgery. Spine (Phila Pa 1976) 40(13):1022–1032. https://doi.org/10.1097/BRS.0000000000000916 CrossRefGoogle Scholar