Biomechanical advantages of supplemental accessory and satellite rods with and without interbody cages implantation for the stabilization of pedicle subtraction osteotomy
To investigate the effect of anterior interbody cages, accessory and satellite rods usage on primary stability and rod strains for PSO stabilization.
Seven human cadaveric spine segments (T12-S1) underwent PSO at L4 with posterior fixation from L2 to S1. In vitro flexibility tests were performed under pure moments in flexion/extension (FE), lateral bending (LB) and axial rotation (AR) to determine the range of motion, while measuring the strains on the primary rods with strain gauge rosettes. Six constructs with 2, 3 and 4 rods, with and without interbody cages implantation adjacent to the PSO site, were compared.
All constructs had comparable effects in reducing spine kinematics compared to the intact condition (− 94% in FE and LB; − 80% in AR). Supplementation of 2 rods with lateral accessory rods (4 rods) was the most effective strategy in minimizing primary rod strains, particularly when coupled to cages (p ≤ 0.005; − 50% in FE, − 42% in AR and − 11% in LB); even without cages, the strains were significantly reduced (p ≤ 0.009; − 26%, − 37%, − 9%). The addition of a central satellite rod with laminar hooks (3 rods) effectively reduced rod strains in FE (p ≤ 0.005; − 30%) only in combination with cages.
The study supports the current clinical practice providing a strong biomechanical rationale to recommend 4-rod constructs based on accessory rods combined with cages adjacent to PSO site. Although weaker, the usage of accessory rods without cages and of a central satellite rod with hooks in combination with interbody spacers may also be justified.
KeywordsPedicle subtraction osteotomy Primary stability Revision Rod breakage Implant failure Accessory rods Satellite rod In vitro study Strain gauge Spine Biomechanics
This study was funded by the Scoliosis Research Society through a New Investigator Grant. The authors gratefully acknowledge DePuy Synthes (Raynham, MA, USA), Medtronic Sofamor Danek (Minneapolis, MN, USA) and NuVasive (San Diego, CA, USA) for providing the implants and surgical tools. The authors gratefully acknowledge Gloria Casaroli Ph.D., Maria Luisa Ruspi, Lisa Flachmüller and Theodor Di Pauli von Treuheim for their assistance during specimens’ preparation. Tito Bassani Ph.D. is gratefully acknowledged for comments regarding statistical analysis.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest related to the content of the current study.
- 8.Smith JS, Shaffrey CI, Klineberg E, Lafage V, Schwab F, Lafage R, Kim HJ, Hostin R, Mundis GM Jr, Gupta M, Liabaud B, Scheer JK, Diebo BG, Protopsaltis TS, Kelly MP, Deviren V, Hart R, Burton D, Bess S, Ames CP, on behalf of the International Spine Study Group (2017) Complication rates associated with 3-column osteotomy in 82 adult spinal deformity patients: retrospective review of a prospectively collected multicenter consecutive series with 2-year follow-up. J Neurosurg Spine 27(4):444–457. https://doi.org/10.3171/2016.10.SPINE16849 CrossRefPubMedGoogle Scholar
- 9.Gupta S, Eksi MS, Ames CP, Deviren V, Durbin-Johnson B, Smith JS, Gupta MC (2017) A Novel 4-rod technique offers potential to reduce rod breakage and pseudarthrosis in pedicle subtraction osteotomies for adult spinal deformity correction. Oper Neurosurg (Hagerstown). https://doi.org/10.1093/ons/opx151 CrossRefGoogle Scholar
- 11.Smith JS, Shaffrey E, Klineberg E, Shaffrey CI, Lafage V, Schwab FJ, Protopsaltis T, Scheer JK, Mundis GM, Fu KMG, Gupta MC, Hostin R, Deviren V, Kebaish K, Hart R, Burton DC, Line B, Bess S, Ames CP (2014) Prospective multicenter assessment of risk factors for rod fracture following surgery for adult spinal deformity. J Neurosurg Spine 21(6):994–1003. https://doi.org/10.3171/2014.9.SPINE131176 CrossRefPubMedGoogle Scholar
- 12.Smith JS, Shaffrey CI, Ames CP, Demakakos J, Fu KMG, Keshavarzi S, Li CMY, Deviren V, Schwab FJ, Lafage V, Bess S (2012) Assessment of symptomatic rod fracture after posterior instrumented fusion for adult spinal deformity. Neurosurgery 71(4):862–867. https://doi.org/10.1227/NEU.0b013e3182672aab CrossRefPubMedGoogle Scholar
- 18.ISSG, GuptaM Henry J, Lafage V, Schwab FJ, Ames CP et al (2015) Reducing rod breakage and nonunion in pedicle subtraction osteotomy: the importance of rod number and configuration in 264 patients with 2-years follow-up. Spine J 15:154S–155S. https://doi.org/10.1016/j.spinee.2015.07.173 CrossRefGoogle Scholar
- 21.Hallager DW, Gehrchen M, Dahl B, Harris JA, Gudipally M, Jenkins S, Wu AM, Bucklen BS (2016) Use of supplemental short pre-contoured accessory rods and cobalt chrome alloy posterior rods reduces primary rod strain and range of motion across the pedicle subtraction osteotomy level: an in vitro biomechanical study. Spine 41(7):E388–E395. https://doi.org/10.1097/BRS.0000000000001282 CrossRefPubMedGoogle Scholar
- 22.Luca A, Ottardi C, Sasso M, Prosdocimo L, La Barbera L, Brayda-Bruno M, Galbusera F, Villa T (2017) Instrumentation failure following pedicle subtraction osteotomy: the role of rod material, diameter, and multi-rod constructs. Eur Spine J 26(3):764–770. https://doi.org/10.1007/s00586-016-4859-8 CrossRefPubMedGoogle Scholar
- 23.Luca A, Ottardi C, Lovi A, Brayda-Bruno M, Villa T, Galbusera F (2017) Anterior support reduces the stresses on the posterior instrumentation after pedicle subtraction osteotomy: a finite-element study. Eur Spine J 26(Suppl 4):450–456. https://doi.org/10.1007/s00586-017-5084-9 CrossRefPubMedGoogle Scholar
- 29.Dahl BT, Harris JA, Gudipally M, Moldavsky M, Khalil S, Bucklen BS (2017) Kinematic efficacy of supplemental anterior lumbar interbody fusion at lumbosacral levels in thoracolumbosacral deformity correction with and without pedicle subtraction osteotomy at L3: an in vitro cadaveric study. Eur Spine J 26(11):2773–2781. https://doi.org/10.1007/s00586-017-5222-4 CrossRefPubMedGoogle Scholar
- 31.Berti F, La Barbera L, Piovesan A, Allegretti D, Ottardi C, Villa T, Pennati G (2018) Residual stresses in titanium spinal rods: effects of two contouring methods and material plastic properties. J Mech Behav Biomed Mater. Submitted on 10/01/2018Google Scholar
- 32.Tang JA, Leasure JM, Smith JS, Buckley JM, Kondrashov D, Ames CP (2013) Effect of severity of rod contour on posterior rod failure in the setting of lumbar pedicle subtraction osteotomy (PSO): a biomechanical study. Neurosurgery 72(2):276–282. https://doi.org/10.1227/neu.0b013e31827ba066 (discussion 283) CrossRefPubMedGoogle Scholar
- 34.Ottardi C (2015) Ph.D. thesis. Politecnico di Milano, Dec 2015Google Scholar
- 35.Wilke HJ, Rohlmann A, Neller S, Schultheiss M, Bergmann G, Graichen F, Claes LE (2001) Is it possible to simulate physiologic loading conditions by applying pure moments? A comparison of in vivo and in vitro load components in an internal fixator. Spine (Phila Pa 1976) 15 26(6):636–642CrossRefGoogle Scholar