We propose a new flexible piezoelectric micromachined ultrasonic transducer (pMUT) array integrated on flexible polydimethylsiloxane (PDMS) that can be used in studying brain stimulation by ultrasound. To achieve the technical demands of a high sound pressure level and flexibility, a diaphragm-type piezoelectric ultrasound transducer array was manufactured with 55 μm-thick bulk lead zirconate titanate (PZT) that was thinned after bonding with a silicon wafer. The ultrasound transducer array was then strongly bonded onto a PDMS substrate using an oxygen-plasma treatment followed by precise dicing with a fixed pitch to achieve flexibility. The radius of curvature was smaller than 5 mm, which is sufficient for attachment to the surface of a mouse brain. After a thinning process for the PZT layer, we observed that the PZT layer still maintained a high ferroelectric property. The measured remnant polarization (Pr) and coercive field (Ec) were 28.26 μC/cm2 and 79 kV/cm, respectively. The resonant frequencies of fabricated pMUT elements with different membrane sizes of 700, 800, 900, 1200 μm in diameter were measured to be 694.4, 565.4, 430.8, and 289.3 kHz, respectively. By measuring the ultrasound output pressure, a pMUT showed a sound intensity (Isppa) of 44 mW/cm2 at 80 V, which is high enough for low-intensity ultrasound brain stimulation.
PDMS Deep Brain Stimulation Brain Stimulation Chemical Mechanical Polishing PDMS Layer
These keywords were added by machine and not by the authors. This process is experimental and the keywords may be updated as the learning algorithm improves.
This is a preview of subscription content, log in to check access.
This work was supported financially by the KIST Institutional Program (Project No. 2E25590).
Aktakka EE, Peterson RL, Najafi K (2011) Thinned-PZT on SOI process and design optimization for piezoelectric inertial energy harvesting. IEEE Transducers Proc 1649–1652Google Scholar
Che L, Halvorsen E, Chen X (2011) An optimized one-step wet etching process of Pb (Zr 0.52 Ti 0.48) O3 thin films for microelectromechanical system applications. J Micromech Microeng 21:105008. doi:10.1088/0960-1317/21/10/105008
Choi HS, Ding JL, Bandyopadhyay A, Anderson MJ, Bose S (2008) Characterization and modeling of a piezoelectric micromachined ultrasonic transducer with a very large length/width aspect ratio. J Micromech Microeng 18:025037. doi:10.1088/0960-1317/18/2/025037CrossRefGoogle Scholar
Eccardt PC, Niederer K (2000) Micromachined ultrasound transducers with improved coupling factors from a CMOS compatible process. Ultrasonics 38:774–780CrossRefGoogle Scholar
Jung J, Kim S, Lee W, Choi H (2013) Fabrication of a two-dimensional piezoelectric micromachined ultrasonic transducer array using a top-crossover-to-bottom structure and metal bridge connections. J Micromech Microeng 23:125037. doi:10.1088/0960-1317/23/12/125037CrossRefGoogle Scholar