“From microtiter plates to droplets” tools for micro-fluidic droplet processing
Droplet-based microfluidic allows high throughput experimentation in with low volume droplets. Essential fluidic process steps are on the one hand the proper control of the droplet composition and on the other hand the droplet processing, manipulation and storage. Beside integrated fluidic chips, standard PTFE-tubings and fluid connectors can be used in combination with appropriate pumps to realize almost all necessary fluidic processes. The segmented flow technique usually operates with droplets of about 100–500 nL volume. These droplets are embedded in an immiscible fluid and confined by channel walls. For the integration of segmented flow applications in established research workflows—which are usually base on microtiter plates—robotic interface tools for parallel/serial and serial/parallel transfer operations are necessary. Especially dose–response experiments are well suited for the segmented flow technique. We developed different transfer tools including an automated “gradient take-up tool” for the generation of segment sequences with gradually changing composition of the individual droplets. The general working principles are introduced and the fluidic characterizations are given. As exemplary application for a dose–response experiment the inhibitory effect of antibiotic tetracycline on Escherichia coli bacteria cultivated inside nanoliter droplets was investigated.
Financial support from the BMWI-Project “PharmTest” FKZ:KS2731202AK0, BMBF-Project “BactoKat” FKZ:031A161A, and the German Federal Environmental Foundation (DBU 20009/009) is gratefully acknowledged.
- Cao JL, Kürsten D, Schneider S, Köhler JM (2012b) Stimulation and inhibition of bacterial growth by caffeine dependent on chloramphenicol and a phenolic uncoupler-A ternary toxicity study using microfluid segment technique. J Biomed Nanotechnol 8(5):770–778. doi: 10.1166/jbn.2012.1447 CrossRefGoogle Scholar
- Chalmers G, Kozak GK, Hillyer E, Reid-Smith RJ, Boerlin P (2010) Low minimum inhibitory concentrations associated with the tetracycline-resistance gene tet(C) in Escherichia coli. Can J Vet Res-Rev Can Rech Vet 74(2):145–148Google Scholar
- Funfak A, Hartung R, Cao J, Martin K, Wiesmueller K-H, Wolfbeis OS, Köhler JM (2009) Highly resolved dose-response functions for drug-modulated bacteria cultivation obtained by fluorometric and photometric flow-through sensing in microsegmented flow. Sens Actuators B Chem 142(1):66–72. doi: 10.1016/j.snb.2009.07.017 CrossRefGoogle Scholar
- Hartung R, Brösing A, Sczcepankiewicz G, Liebert U, Haefner N, Duerst M, Felbel J, Lassner D, Köhler JM (2009) Application of an asymmetric helical tube reactor for fast identification of gene transcripts of pathogenic viruses by micro flow-through PCR. Biomed Microdevices 11(3):685–692. doi: 10.1007/s10544-008-9280-6 CrossRefGoogle Scholar
- Miller OJ, El Harrak A, Mangeat T, Baret J-C, Frenz L, El Debs B, Mayot E, Samuels ML, Rooney EK, Dieu P, Galvan M, Link DR, Griffiths AD (2012) High-resolution dose-response screening using droplet-based microfluidics. Proc Natl Acad Sci USA 109(2):378–383. doi: 10.1073/pnas.1113324109 CrossRefGoogle Scholar