Helmet-based physiological signal monitoring system
- 472 Downloads
A helmet-based system that was able to monitor the drowsiness of a soldier was developed. The helmet system monitored the electrocardiogram, electrooculogram and electroencephalogram (alpha waves) without constraints. Six dry electrodes were mounted at five locations on the helmet: both temporal sides, forehead region and upper and lower jaw strips. The electrodes were connected to an amplifier that transferred signals to a laptop computer via Bluetooth wireless communication. The system was validated by comparing the signal quality with conventional recording methods. Data were acquired from three healthy male volunteers for 12 min twice a day whilst they were sitting in a chair wearing the sensor-installed helmet. Experimental results showed that physiological signals for the helmet user were measured with acceptable quality without any intrusions on physical activities. The helmet system discriminated between the alert and drowsiness states by detecting blinking and heart rate variability (HRV) parameters extracted from ECG. Blinking duration and eye reopening time were increased during the sleepiness state compared to the alert state. Also, positive peak values of the sleepiness state were much higher, and the negative peaks were much lower than that of the alert state. The LF/HF ratio also decreased during drowsiness. This study shows the feasibility for using this helmet system: the subjects’ health status and mental states could be monitored without constraints whilst they were working.
KeywordsHelmet Dry electrodes ECG EOG EEG Alpha wave Drowsiness
This study was supported by a grant from the Advanced Biometric Research Center (ABRC) and the Korea Science and Engineering Foundation (KOSEF).
- Andreassi JL (2000) Psychophysiology: human behavior and physiological response. Lawrence Erlbaum, MahwahGoogle Scholar
- Galley N, Schleicher R, Galley L (2004) Blink parameters as indicators of driver’s sleepiness—possibilities and limitations. In: Gale A (ed) Vision in vehicles X. Elsevier, Amsterdam, pp 189–196Google Scholar
- Kaefer G, Prochart G, Weiss R (2003) Wearable alertness monitoring for industrial applications. In: Seventh IEEE international symposium on wearable computers, New York pp 254–255Google Scholar
- Kong X, Wilson G (1998) A new EOG-based eyeblink detection algorithm. Behav Res Methods Instrum Comput 30:713–719Google Scholar
- Papadelis C, Chen Z, Kourtidou-Papadeli C, Bamidis PD, Chouvarda I, Bekiaris E, Maglaveras N (2007) Monitoring sleepiness with on-board electrophysiological recordings for preventing sleep-deprived traffic accidents. Clin Neurophysiol 118:1906–1922. doi: 10.1016/j.clinph.2007.04.031 PubMedCrossRefGoogle Scholar
- Scherrer JF, Xian H, Lyons MJ, Goldberg J, Eisen SA, True WR, Tsuang M, Bucholz KK, Koenen KC (2008) Posttraumatic stress disorder; combat exposure; and nicotine dependence, alcohol dependence, and major depression in male twins. Compr Psychiatry 49:297–304. doi: 10.1016/j.comppsych.2007.11.001 PubMedCrossRefGoogle Scholar
- Stern RM, Ray WJ, Quigley KS (2001) Psychophysiological recording. Oxford University Press, New YorkGoogle Scholar
- Svensson U (2004) Blink behaviour based drowsiness detection: method development and validation. Department of Biomedical Engineering, Linköping University, Sweden, p 20Google Scholar
- Webster JG (1998) Medical instrumentation: application and design. Wiley, CanadaGoogle Scholar