Better performance of RTVue than Cirrus spectral-domain optical coherence tomography in detecting band atrophy of the optic nerve
- 236 Downloads
To assess the agreement and diagnostic performance between retinal nerve fiber layer (RNFL) thickness measurements obtained using the Cirrus (Carl Zeiss Meditec) and RTVue (Optovue Inc.) devices for detection of band atrophy (BA) in patients with permanent temporal hemianopia.
In this retrospective study, 26 eyes with BA and 64 control eyes were enrolled. The Cirrus optic disc cube protocol and the RTVue optic nerve head map protocol were used. The Cirrus measurements were extracted and regrouped to be topographically matched with the RTVue measurements. Concordance correlation and 95 % limits of agreement were assessed. Areas under the receiver operating characteristic curves (AUC) and the Spearman’s correlation coefficient between average Humphrey total deviation in the temporal hemifield and average RNFL thickness were calculated.
RTVue measured consistently thicker values than Cirrus in controls, whereas in eyes with BA, nasal segment measurements from the RTVue were thinner than those obtained using the Cirrus. Each quadrant showed moderate to close agreement in controls, whereas in eyes with BA, the nasal and temporal quadrants exhibited poor agreement. The RTVue measurements demonstrated significantly higher AUCs for nasal segments just above (0.95) and below (0.96) the horizontal meridian than Cirrus measurements (0.80 and 0.66, respectively) and a significant correlation with visual field loss (rs = 0.46, P = 0.02 for RTVue vs. rs = 0.26, P = 0.22 for Cirrus).
The RTVue RNFL thickness measurements in nasal sectors showed better diagnostic performance in detecting BA and higher correlations with temporal hemianopia than the Cirrus measurements.
KeywordsBand atrophy Chiasmal compression Optical coherence tomography Retinal nerve fiber layer thickness Bitemporal hemianopia
This study was supported in part by Grants-in-Aid 22390324 (A.N., Y.Y., M.N.) and 20592043 (M.N., A.N.) from the Ministry of Education, Culture, Sports, and Science and Technology of the Japanese government.
- 7.Danesh-Meyer HV, Carroll SC, Foroozan R, Savino PJ, Fan J, Jiang Y, Vander Hoorn S (2006) Relationship between retinal nerve fiber layer and visual field sensitivity as measured by optical coherence tomography in chiasmal compression. Investig Ophthalmol Vis Sci 47(11):4827–4835CrossRefGoogle Scholar
- 20.Watson GM, Keltner JL, Chin EK, Harvey D, Nguyen A, Park SS (2011) Comparison of retinal nerve fiber layer and central macular thickness measurements among five different optical coherence tomography instruments in patients with multiple sclerosis and optic neuritis. J Neuro-ophthalmol 31(2):110–116CrossRefGoogle Scholar
- 24.Gabriele ML, Ishikawa H, Wollstein G, Bilonick RA, Townsend KA, Kagemann L, Wojtkowski M, Srinivasan VJ, Fujimoto JG, Duker JS, Schuman JS (2008) Optical coherence tomography scan circle location and mean retinal nerve fiber layer measurement variability. Investig Ophthalmol Vis Sci 49(6):2315–2321CrossRefGoogle Scholar