Intra-examiner repeatability and agreement of corneal pachymetry map measurement by time-domain and Fourier-domain optical coherence tomography
- 159 Downloads
To assess the intra-examiner repeatability and agreement of corneal pachymetry maps measured by time-domain (Visante OCT) and Fourier-domain optical coherence tomography (RTVue OCT).
This observational cross-sectional study enrolled 72 eyes from 72 volunteers. Only one eye of each patient was chosen randomly to receive repeated scanning with both devices by the same examiner, in order to test the intra-observer repeatability. The first scan by each device from all enrolled eyes was used to analyze the difference and agreement between the two devices. The agreement between the two devices was analyzed by the Bland–Altman method. Intra-observer repeatability of each OCT device was analyzed by intra-class correlation (ICC).
The mean corneal thickness of the central 2 mm zone was 524.3 ± 35.7 µm and 525.4 ± 35.3 µm by the Visante OCT and the RTVue OCT respectively (p = 0.089). The intra-observer repeatability of the RTVue OCT (ICC = 0.994) was superior to that of the Visante OCT (ICC = 0.989) in the central 2 mm zone. However, the intra-observer repeatability of the RTVue OCT in the pericentral 2 to 5 mm zone was not superior to that of the Visante OCT. Both OCT devices had similar repeatability in the pericentral 2 to 5 mm zone [ICC of the Visante OCT = 0.991; 95% confidence interval (CI) 0.986–0.995), ICC of the RTVue OCT = 0.991; 95% CI 0.985–0.994)).
The difference in CCT measurement by the Visante OCT and the RTVue OCT is probably too small to influence clinical decision making for refractive surgery and glaucoma management. The RTVue OCT demonstrated better intra-observer repeatability in the central 2 mm zone, which probably was related to its rapid image acquisition capability.
KeywordsCorneal thickness Time domain Fourier domain Optical coherence tomography
- 6.Gordon MO, Beiser JA, Brandt JD, Heuer DK, Higginbotham EJ, Johnson CA, Keltner JL, Miller JP, Parrish RK 2nd, Wilson MR, Kass MA (2002) The Ocular Hypertension Treatment Study: baseline factors that predict the onset of primary open-angle glaucoma. Arch Ophthalmol 120:714–720; discussion 829–730, doi: ecs20044 [pii]PubMedGoogle Scholar
- 7.Dueker DK, Singh K, Lin SC, Fechtner RD, Minckler DS, Samples JR, Schuman JS (2007) Corneal thickness measurement in the management of primary open-angle glaucoma: a report by the American Academy of Ophthalmology. Ophthalmology 114:1779–1787, doi: S0161-6420(07)00773-7 [pii] 10.1016/j.ophtha.2007.04.068 CrossRefPubMedGoogle Scholar
- 13.Li H, Leung CK, Wong L, Cheung CY, Pang CP, Weinreb RN, Lam DS (2008) Comparative study of central corneal thickness measurement with slit-lamp optical coherence tomography and visante optical coherence tomography. Ophthalmology 115:796–801, doi: S0161-6420(07)00775-0 [pii] 10.1016/j.ophtha.2007.07.006, e792CrossRefPubMedGoogle Scholar
- 16.Prospero Ponce CM, Rocha KM, Smith SD, Krueger RR (2009) Central and peripheral corneal thickness measured with optical coherence tomography, Scheimpflug imaging, and ultrasound pachymetry in normal, keratoconus-suspect, and post-laser in situ keratomileusis eyes. J Cataract Refract Surg 35:1055–1062, doi: S0886-3350(09)00224-7 [pii] 10.1016/j.jcrs.2009.01.022 CrossRefPubMedGoogle Scholar
- 20.Mohamed S, Lee GK, Rao SK, Wong AL, Cheng AC, Li EY, Chi SC, Lam DS (2007) Repeatability and reproducibility of pachymetric mapping with Visante anterior segment-optical coherence tomography. Invest Ophthalmol Vis Sci 48:5499–5504, doi: 48/12/5499 [pii] 10.1167/iovs.07-0591 CrossRefPubMedGoogle Scholar
- 22.Li EY, Mohamed S, Leung CK, Rao SK, Cheng AC, Cheung CY, Lam DS (2007) Agreement among 3 methods to measure corneal thickness: ultrasound pachymetry, Orbscan II, and Visante anterior segment optical coherence tomography. Ophthalmology 114:1842–1847, doi: S0161-6420(07)00189-3 [pii] 10.1016/j.ophtha.2007.02.017 CrossRefPubMedGoogle Scholar
- 23.Prakash G, Agarwal A, Jacob S, Kumar DA, Banerjee R (2009) Comparison of fourier-domain and time-domain optical coherence tomography for assessment of corneal thickness and intersession repeatability. Am J Ophthalmol 148:282–290, doi: S0002-9394(09)00214-1 [pii] 10.1016/j.ajo.2009.03.012, e282CrossRefPubMedGoogle Scholar