Electrophysiological detection of scalar changing perimodiolar cochlear electrode arrays: a long term follow-up study
- 233 Downloads
The position of the cochlear electrode array within the scala tympani is essential for an optimal hearing benefit. An intraoperative NRT-ratio was established, which can provide information about the intraoperative intracochlear electrode array position for perimodiolar electrodes. The aims of this study were to verify the longterm reliability for the NRT-ratio in perimodiolar electrodes. In a retrospective controlled study in a Tertiary Referral Center the electrophysiological data sets of 123 patients with implanted Nucleus Contour Advance electrodes were enclosed. Intraoperative and up to 1 year follow-up Auto-NRTs were evaluated. A NRT-ratio was calculated by dividing the average Auto-NRT data from electrode 16 to 18 with the average from electrode 5 to 7. Using a flat panel tomography system, the position of the electrode array was certified radiological. 31 patients with perimodiolar electrodes with 1 year follow-up data were included in the study. Eleven patients showed regular follow-up NRT-ratio with a correlated and radiologically confirmed electrode position. 20 patients showed mismatches between the NRT-ratio and the radiological position. These patients were highly variable in terms of duration of deafness and neural spectrum disorders. The NRT-ratio can be used to determine the intracochlear position of the electrode array for perimodiolar electrodes. Intraoperatively the NRT-ratio predicts the array position within the cochlea highly reliable for perimodiolar electrodes. We showed that after 6 months and a year, the NRT-ratio remains unchanged in most of the cases and shows a good correlation to the radiological determined position of the array. Nevertheless, the condition of the neural structures is highly important for reproducible responses. Limited validity is given in patients with degenerative and structural neural disorders.
KeywordsCochlear implant NRT Radiology Quality control
This study was funded by Cochlear Pty. Sydney, Australia (IIR 1209).
Compliance with ethical standards
Conflict of interest
The Unfallkrankenhaus Berlin received research grants from Cochlear Pty.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards. For this type of study formal consent is not required.
- 5.Carelsen B, Grolman W, Tange R, Streekstra GJ, van Kemenade P, Jansen RJ, Freling NJ, White M, Maat B, Fokkens WJ (2007) Cochlear implant electrode array insertion monitoring with intra-operative 3D rotational X-ray. Clin Otolaryngol 32(1):46–50. doi: 10.1111/j.1365-2273.2007.01319.x CrossRefPubMedGoogle Scholar
- 8.Svrakic M, Friedmann DR, Berman PM, Davis AJ, Roland JT Jr, Svirsky MA (2015) Measurement of cochlear implant electrode position from intraoperative post-insertion skull radiographs: a validation study. Otol Neurotol 36(9):1486–1491. doi: 10.1097/MAO.0000000000000852 CrossRefPubMedPubMedCentralGoogle Scholar
- 10.Holden LK, Finley CC, Firszt JB, Holden TA, Brenner C, Potts LG, Gotter BD, Vanderhoof SS, Mispagel K, Heydebrand G, Skinner MW (2013) Factors affecting open-set word recognition in adults with cochlear implants. Ear Hear 34(3):342–360. doi: 10.1097/AUD.0b013e3182741aa7 CrossRefPubMedPubMedCentralGoogle Scholar
- 16.Wanna GB, Noble JH, Gifford RH, Dietrich MS, Sweeney AD, Zhang D, Dawant BM, Rivas A, Labadie RF (2015) Impact of intrascalar electrode location, electrode type, and angular insertion depth on residual hearing in cochlear implant patients: preliminary results. Otol Neurotol 36(8):1343–1348. doi: 10.1097/MAO.0000000000000829 CrossRefPubMedGoogle Scholar
- 17.Boyer E, Karkas A, Attye A, Lefournier V, Escude B, Schmerber S (2015) Scalar localization by cone-beam computed tomography of cochlear implant carriers: a comparative study between straight and periomodiolar precurved electrode arrays. Otol Neurotol 36(3):422–429. doi: 10.1097/MAO.0000000000000705 CrossRefPubMedGoogle Scholar
- 18.Mittmann P, Rademacher G, Mutze S, Hassepass F, Ernst A, Todt I (2015) Evaluation of the relationship between the NRT-ratio, cochlear anatomy, and insertions depth of perimodiolar cochlear implant electrodes. BioMed Res Int 2015:706253. doi: 10.1155/2015/706253 CrossRefPubMedPubMedCentralGoogle Scholar
- 21.Henkin Y, Kaplan-Neeman R, Kronenberg J, Migirov L, Hildesheimer M, Muchnik C (2006) A longitudinal study of electrical stimulation levels and electrode impedance in children using the Clarion cochlear implant. Acta Otolaryngol 126(6):581–586. doi: 10.1080/00016480500443391 CrossRefPubMedGoogle Scholar