Preservation of skin DNA for oligonucleotide array CGH studies: a feasibility study
Array-based comparative genomic hybridization (a-CGH) is a promising tool for clinical genomic studies. However, pre-analytical sample preparation methods have not been fully evaluated for this purpose. Parallel sections of normal male human skin biopsy samples were collected and immediately immersed in saline, formalin and a molecular fixative for 8, 12 and 24 h. Genomic DNA was isolated from the samples and subjected to amplification and labeling. Labeled samples were then co-hybridized with normal reference female DNA to Agilent oligonucleotide-based a-CGH 44k slides. Pre-analytic parameters such as DNA yield, quality of genomic DNA and labeling efficacy were evaluated. Also microarray analytical variables, including the feature signal intensity, data distribution dynamic range, signal to noise ratio and background intensity levels were assessed for data quality. DNA yield and quality of genomic DNA—as evaluated by spectrophotometry and gel electrophoresis—were similar for fresh and molecular fixative-exposed samples. In addition, labeling efficacy of dye incorporation was not drastically different. There was no difference between fresh and molecular fixative material comparing scan parameters and stem plot analysis of a-CGH result. Formalin-fixed samples, on the other hand, showed various errors such as oversaturation, non-uniformity in replicates, and decreased signal to noise ratio. Overall, the a-CGH result of formalin samples was not interpretable. DNA extracted from formalin-fixed tissue samples is not suitable for oligonucleotide-based a-CGH studies. On the other hand, the molecular fixative preserves tissue DNA similar to its fresh state with no discernable analytical differences.
KeywordsGenomics Array comparative genomic hybridization Fixatives Tissue preservation
Applications of new technologies have resulted in major advancements in laboratory medicine . Bringing these advances into clinical practice, however, requires careful evaluation and validation. Array-based comparative genomic hybridization (a-CGH) is an extremely powerful tool that can generate high resolution mapping of chromosomal abnormalities. Advances in microarray technology and bioinformatics have now made a-CGH easily available and affordable [14, 15]. With a-CGH’s potential for clinical application, it is important that guidelines for proper sample preparation and control of quality are developed. Since conventional methods of clinical tissue preparation commonly employ formalin fixation, we studied the suitability of formalin for array CGH studies and compared the results to that of clinical samples that were preserved in a newly developed, molecular-friendly fixative.
The study was approved by the University of Miami Institutional Review Board. Three separate normal skin biopsies from one volunteer healthy male were immediately sliced in three parts each 0.1 × 0.2 × 0.2 cm. One part was immersed in normal saline solution, one part in a methanol-based molecular tissue fixative, UMFIX (Universal Molecular Fixative, marketed as Tissue-Tek® Xpress™ Molecular Fixative, Sakura Finetek, Torrance, CA), and the third slice was fixed in 10% neutral buffered formalin. Volume of fixative/preservative was 150 ml and incubation was performed at room temperature. After 8 (set 1), 12 (set 2) and 24 h (set 3), the genomic DNA was extracted from the samples using Puregene DNA Purification System tissue kit (Gentra, Minneapolis, MN). One microliter of extracted DNA solution was diluted in Tris–EDTA (TE) buffer and evaluated on ND-1000 spectrophotometer (NanoDrop Technologies, Rockland, DE). Gel electrophoresis (0.8% agarose gel) was performed to evaluate the quality of genomic DNA.
Spectrophotometric result of total DNA yield, DNA purity, and labeling efficiency, for fresh, formalin-fixed and UMFIX-exposed samples
Total DNA (μg)
Sample-cy5 labeling efficiency (pmol/μg)
Control-cy3 labeling efficiency (pmol/μg)
Mean ± SD
3.25 ± 0.55
1.84 ± 0.07
1.74 ± 0.29
55.47 ± 14.19
99.20 ± 10.87
Mean ± SD
4.38 ± 1.88
1.86 ± 0.01
1.88 ± 0.11
58.80 ± 17.16
93.27 ± 14.82
Mean ± SD
4.61 ± 0.68
1.73 ± 0.18
1.34 ± 0.65
45.73 ± 12.44
97.73 ± 38.73
Genomic DNA from the skin samples and the control female DNA (Promega, Madison, WI) were then subjected to amplification according to Agilent’s (Palo Alto, CA) protocol for oligonucleotide array-based CGH for genomic DNA (version 2.0 August 2005). Amplification, of both male genomic DNA and female control DNA, 100 ng each, was performed using Qiagen REPLI-g Amplification Kit. Amplified DNA was digested during a 2 h incubation at 37°C, with Alu I and Rsa I (10 U/μl; 5μl/reaction) restriction enzymes (Promega, Madison, WI). Purification of digested DNA was performed with QIAprep Spin Miniprep Kit (Qiagen, Valencia, CA). Digested DNA was subjected to electrophoresis to evaluate quality of amplified DNA by visual inspection of its uniformity and range.
The amplification efficiency of DNA from formalin-fixed samples was less than fresh and UMFIX samples, as evidenced by the size and fluorescent intensity of the bands (Fig. 1b).
We also used PCR for GAPDH primers to gauge the quality of extracted DNA. PCR was performed using glyceraldehyde-3-phosphate dehydrogenase primers (GAPDH, Clonetech, Palo Alto, CA) using 0.5 μg of RNase-treated isolated DNA and Qiagen TaqPCR Mastermix (Qiagen, Valencia, CA). The conditions for DNA PCR were as follows: 95°C, 15 min; 35 cycles at 94°C, 45 s; 60°C, 45 s; 72°C, 2 min. As seen in Fig. 1c, a 450 bp band was detected in all samples, although the intensity was considerably lower in formalin-fixed material.
Amplified Genomic DNA was then labeled using BioPrime Array CGH Genomic Labeling kit (Invitrogen, Carlsbad, CA). Quality analysis and quantitation was performed with ND-1000 (NanoDrop Technologies, Wilmington, DE) spectrophotometer to calculate the labeling efficiency. Labeled and purified samples were combined with hybridization master mix and applied to Agilent Human Genome CGH Microarray 44B slides for 40 h at 65°C. To minimize the impact of environmental oxidants on signal intensities, slides were scanned immediately using Agilent microarray scanner 4800B. Array images were analyzed using Agilent feature extraction software (v8.1) and CGH explorer (v2.51) . Microarray analytical variables, including the feature signal intensity, data distribution dynamic range, signal to noise ratio and background intensity levels, and the number of saturated and undetected features was used to assess microarray quality.
Representative data of microarray scan quality measurement
Reproducibility: non-control replicated probes median %CV
Mean ± SD
67 ± 40
218 ± 140
5 ± 1
6 ± 2
Mean ± SD
158 ± 170
292 ± 233
8 ± 4
9 ± 3
Mean ± SD
2,175 ± 882
210 ± 80
69 ± 30
9 ± 3
324 ± 38
These findings may be attributed to erratic and random fragmentation of DNA in formalin-fixed samples. The fragmentation itself results in increase background noise and aberrant signal intensity due to random hybridization.
Array CGH studies have great potential for clinical application. Since the technique does not utilize live cells, it is considerably more advantageous when compared to conventional karyotyping techniques. Array CGH has its own limitations, such as failure to detect translocations; nevertheless it offers unprecedented spatial resolution [6, 11]. Recently, it has been shown that low-level gene copy number change is associated with changes in expression level of its transcripts . Therefore, results of transcriptomics studies can be used to study DNA markers, which are more stable and easier to study than RNA.
Array GGH was originally based on BAC (Bacterial Artificial Chromosomes), but more recently it utilizes oligonucleotides, with its ability to provide whole genome-covering resolution. Synthetic oligonucleotides obviously have the advantage because their exact sequence and length are known for each element on the array. Oligonucleotide array CGH (oa-CGH) has the benefit of overcoming the difficulties inherent to BAC arrays, such as the amount of available DNA, clone management, probe identity due to PCR contaminations, and mapping inaccuracies. Using oa-CGH, it is also possible to eliminate another source of error, which is the batch-to-batch variability of Cot-1 DNA, used to block repetitive DNA sequences, since the oligonucleotide probes are designed to be repeat-free .
Samples used in nearly all a-CGH studies have been fresh or fresh-frozen tissue. Such material, although useful in research settings, is impractical and cumbersome to use in clinical practice. Furthermore, because diagnostic biopsy samples are relatively small, the amount of residual tissue for additional ancillary testing may be inadequate. Hence there is a great need to develop test strategies that require minimal amounts of tissue and are robust enough to withstand pre-analytical sample preparation. Simplified schemes for sample preservation that allow reliable histomorphology along with preservation of high quality macromolecules are desirable.
We have previously described a novel tissue fixation and processing technique that beside providing adequate histomorphology also preserves high quality HMW RNA, akin to fresh samples . This fixative also protects HMW tissue DNA for use in PCR studies . Here, we further evaluated the suitability of skin tissue DNA for array CGH studies using same methods applied routinely for fresh samples. This study demonstrates that using the novel fixative it is possible to preserve and extract high molecular weight genomic DNA, supported by high labeling efficiency, comparable to that from fresh tissue. Furthermore, no artifacts were seen using this DNA in microarray scanning or analysis for array CGH.
Array CGH studies have been performed on formalin-fixed tissues with variable and irreproducible results. Most of the prior studies were based on low-resolution BAC arrays without detailed description of DNA quality or array metrics [10, 21]. Besides, there was often inadequate documentation of fixation time, the volumetric ratio of fixative to tissue, or buffering status. Only few studies have adequately addressed analytical aspects of fixation and processing on array CGH. Ghazani et al.  studied the effect of formalin fixation on MCF-7 cell line and on one breast cancer tissue sample. They used two different BAC clone arrays; 1.7k for the cell line and 19k for the breast cancer sample . There was no mention of fixative volume or quality of genomic DNA. They showed that long term (1 week) fixation results in extensive loss of HMW DNA. In cell lines, the concordance between fresh and formalin-fixed samples was around 85%. When genomic DNA was amplified, the concordance decreased to 75%.
Johnson et al.  also studied the effect of formalin fixation on a-CGH using BAC arrays. Our results support their conclusion that analysis of DNA samples on agarose gels does not offer any advantage in prediction of suitability of formalin-fixed tissue samples for a-CGH. They also show that the quality of a-CGH depends on the integrity of DNA samples with the requirement that extracted DNA supports the PCR amplification of an amplicon of 300 bp or longer. We showed that it is possible to detect amplicons of up to 450 bp in formalin-fixed sample; (albeit with lower intensity when compared to fresh or UMFIX-exposed samples). The studies by Johnson and others show the possibility of having acceptable results with DNA from formalin-fixed samples using BAC arrays but there are no data that supports its suitability for oa-CGH arrays. This may be due to larger sequence of probes presented in BAC array. Other investigators have shown that using alternative approaches to DNA extraction, quantification, amplification or labeling may produce improved results. However, all other authors do agree that the procedures for improving DNA quality are neither substitute for high quality DNA nor they could obtain consistent results from formalin-fixed samples [7, 10]. This is mostly due to complex chemical effect of formalin on tissue, which is still poorly understood. Formalin has a tissue penetration rate of 2.4 mm in 24 h and adequate fixation requires at least a ten to one volumetric ratio of fixative to tissue . Therefore, many clinical specimens are only partially fixed by formalin before processing. Furthermore during processing, they are exposed to formalin and alcohol, introducing formalin and alcohol-related tissue artifacts . Formaldehyde is a dipolar molecule and can react with amino, or imino group of the anionic forms of the amino acids. This reaction is time and temperature dependent [2, 9, 16, 17, 18]. More recent findings show that the deleterious effect of formalin fixation might also result from cumulative action of other reagents and processing conditions. Conversely, no DNA or nucleoside reactions have been reported with ethanol and none would be expected under physiological conditions . By changing the three dimensional structure of the proteins, alcohols prevent protein functions. Therefore, rapid fixation of samples prevents alteration and degradation of biomolecules and preserves them in their native form .
In summary, we show that by using a new molecular fixative it is possible to preserve skin tissue DNA that is suitable for array CGH studies; identical to fresh tissue. This can be achieved using same methods and protocols used for fresh samples.
- 9.Lingjaerde OC, Baumbusch LO, Liestol K, Glad IK, Borresen-Dale AL (2005) CGH-explorer: a program for analysis of array-CGH data. Bioinformatics 21:817–820Google Scholar
- 12.Pollack JR, Sorlie T, Perou CM, Rees CA, Jeffrey SS, Lonning PE, Tibshirani R, Botstein D, Borresen-Dale AL, Brown PO (2002) Microarray analysis reveals a major direct role of DNA copy number alteration in the transcriptional program of human breast tumors. Proc Natl Acad Sci USA 99:12963–12968PubMedCrossRefGoogle Scholar
- 16.Theis ER (1944) The protein-formaldehyde reaction. I Collagen. J Biol Chem 154:87–97Google Scholar
- 17.Theis ER (1945) The protein–formaldehyde reaction. III. The effect of temperature and denaturation. J Biol Chem 157:7–14Google Scholar