Extension and characterization of pressure-sensitive molecular film
Pressure-sensitive paint (PSP) has the potential as a diagnostic tool for pressure measurement in high Knudsen number regime because it works as a so-called “molecular sensor”. However, there are few reports concerning application of PSP to micro-devices, because conventional PSPs are too thick owing to polymer binders. In our previous work, we adopted the Langmuir–Blodgett (LB) technique to fabricate the pressure-sensitive molecular film (PSMF) using Pd(II) Mesoporphyrin IX (PdMP), which has pressure sensitivity only in the low pressure range (below 130 Pa). In this study, aiming for pressure measurement under an atmospheric pressure condition, we have constructed four samples of PSMFs composed of Pt(II) Mesoporphyrin IX (PtMP), Pt(II) Mesoporphyrin IX dimethylester (PtMPDME), Pt(II) Protoporphyrin IX (PtPP) and Cu(II) Mesoporphyrin IX dimethylester (CuMPDME) as luminescent molecules. The pressure sensitivity of those PSMFs was measured, and it was clarified that the pressure sensitivity of PSMF-PtMP is the highest among the four samples. Moreover, the temperature dependency of PSMF-PtMP was investigated, and we found that the temperature dependency of PSMF is dominated not by the oxygen diffusion in the layer, but by non-radiative deactivation process of excited luminescent molecules.
KeywordsLuminescent Intensity Pressure Sensitivity Oxygen Quenching Volmer Plot Langmuir Adsorption Model
The present work was supported by a grant-in-aid for Scientific Research of the Ministry of Education, Culture, Sports, Science and Technology. The authors would like to express our gratitude to Prof. Tokuji Miyashita and Prof. Masaya Mitsuishi of the Tohoku University and Prof. Takahiro Seki of the Nagoya University for their helpful advice about the LB film technique.
- Asai K (2001) Status and prospect of pressure-sensitive paint technology. Kashika Joho Gakkaishi (J Vis Soc Jpn) 21(83):203–208Google Scholar
- Bird GA (1994) Molecular gas dynamics and the direct simulation of gas flows. Oxford University Press, OxfordGoogle Scholar
- Engler RH et al (1991) Description and assessment of a new optical pressure measurement system (OPMS) demonstrated in the high speed wind tunnel of DLR in Göttingen. DLR-FB 92-24Google Scholar
- Huang CY et al (2002) Molecular sensors for MEMS, AIAA2002-0256. 40th Aerospace sciences meeting & exhibit, Reno, NVGoogle Scholar
- Liu T, Sullivan JP (2005) Pressure and temperature-sensitive paints. Springer, BerlinGoogle Scholar
- Nagai H et al (2006) Flow diagnostic using PSP and TSP technique in a supersonic micronozzle. In: 12th International symposium on flow visualization. Göttinge, GermanyGoogle Scholar
- Ulman A (1991) An introduction to ultrathin organic films from Langmuir–Blodgett to self-assembly. Academic Press, DublinGoogle Scholar