Measurements of carbon monoxide mixing ratios in Houston using a compact high-power CW DFB-QCL-based QEPAS sensor
- 281 Downloads
Measurements of carbon monoxide (CO) mixing ratios in Houston, Texas, during the period from May 16, 2013 to May 28, 2013 were performed using a sensitive, selective, compact, and portable quartz-enhanced photoacoustic spectroscopy (QEPAS)-based CO sensor employing a high-power continuous wave (CW) distributed feedback quantum cascade laser (DFB-QCL). The minimum detectable CO concentration was 3 ppbv for the strong, interference-free R(6) absorption line at 2,169.2 cm−1 and a 5 s data acquisition time. The average CO concentration during the measurement period was 299.1 ± 81.4 ppb with observed minimum and maximum values of 210.5 and 4,307.9 ppb, respectively. A commercially available electrochemical sensor was employed in-line for simultaneous measurements to confirm the response of the CW DFB-QCL-based QEPAS sensor to variations of the CO mixing ratios. Moderate agreement (R 2 = 0.7) was found between both sets of CO measurements.
KeywordsContinuous Wave Modulation Depth Control Electronics Unit Quartz Tuning Fork QEPAS Signal
The Rice University group acknowledges financial support from a National Science Foundation (NSF) Grant EEC-0540832 entitled “Mid-Infrared Technologies for Health and the Environment (MIRTHE)”, a NSF-ANR award for international collaboration in chemistry “Next generation of Compact Infrared Laser based Sensor for environmental monitoring (NexCILAS)” and the Robert Welch Foundation Grant C-0586.
- 6.An introduction to indoor air quality (IAQ). Carbon monoxide (CO). http://www.epa.gov/iaq/co.html#Sources, United States Environmental Protection Agency, Web., Cited: 07/03/2013
- 9.R. Lewicki et al, in The wonder of nanotechnology: quantum optoelectronic devices and applications, Chapter. 23, eds by M. Razeghi, L. Esaki, K.V. Klitzing, (SPIE Press, 2013), pp. 597–632Google Scholar
- 12.Q.Y. Lu et al., Appl. Phys. Lett. 97, 04210 (2010)Google Scholar
- 16.University of Houston Moody Tower (UH-MT), www5.tceq.texas.gov/tamis/index.cfm?fuseaction=report.view_site&CAMS=695, Texas Commision on Environmental Quality, Web, Cited: 7/03/2013
- 20.Y. Qin, S.C. Kot, Atmos. Environ. B Urb. 27, 3 (1993)Google Scholar