Initial clinical evaluation of stationary digital chest tomosynthesis in adult patients with cystic fibrosis
- 68 Downloads
The imaging evaluation of cystic fibrosis currently relies on chest radiography or computed tomography. Recently, digital chest tomosynthesis has been proposed as an alternative. We have developed a stationary digital chest tomosynthesis (s-DCT) system based on a carbon nanotube (CNT) linear x-ray source array. This system enables tomographic imaging without movement of the x-ray tube and allows for physiological gating. The goal of this study was to evaluate the feasibility of clinical CF imaging with the s-DCT system.
Materials and methods
CF patients undergoing clinically indicated chest radiography were recruited for the study and imaged on the s-DCT system. Three board-certified radiologists reviewed both the CXR and s-DCT images for image quality relevant to CF. CF disease severity was assessed by Brasfield score on CXR and chest tomosynthesis score on s-DCT. Disease severity measures were also evaluated against subject pulmonary function tests.
Fourteen patients underwent s-DCT imaging within 72 h of their chest radiograph imaging. Readers scored the visualization of proximal bronchi, small airways and vascular pattern higher on s-DCT than CXR. Correlation between the averaged Brasfield score and averaged tomosynthesis disease severity score for CF was -0.73, p = 0.0033. The CF disease severity score system for tomosynthesis had high correlation with FEV1 (r = -0.685) and FEF 25–75% (r = -0.719) as well as good correlation with FVC (r = -0.582).
We demonstrate the potential of CNT x-ray-based s-DCT for use in the evaluation of cystic fibrosis disease status in the first clinical study of s-DCT.
• Carbon nanotube-based linear array x-ray tomosynthesis systems have the potential to provide diagnostically relevant information for patients with cystic fibrosis without the need for a moving gantry.
• Despite the short angular span in this prototype system, lung features such as the proximal bronchi, small airways and pulmonary vasculature have improved visualization on s-DCT compared with CXR. Further improvements are anticipated with longer linear x-ray array tubes.
• Evaluation of disease severity in CF patients is possible with s-DCT, yielding improved visualization of important lung features and high correlation with pulmonary function tests at a relatively low dose.
KeywordsTomography Nanotubes, Carbon X-rays Cystic fibrosis Scoring methods
Adapted fan-beam volume reconstruction
Digital chest tomosynthesis
- FEF 25–75%
Forced expiratory flow at 25–75% of the pulmonary volume
Forced expiratory volume during the first second
Forced vital capacity
Intra-class correlation coefficients
Lateral costophrenic angles
Pulmonary function tests
Stationary digital chest tomosynthesis
This project was supported by a Translational Team Science Award with funding provided by the UNC School of Medicine and the National Center for Advancing Translational Sciences (NCATS), National Institutes of Health, through grant award no. UL1TR001111. This work is partially supported by the NCI-funded Carolina Center for Cancer Nanotechnology Excellence (5U54CA151652), a grant from the NCI (5R21CA185741) and The University of North Carolina School of Medicine Team Translation Science Award (TTSA001P2). Dr. Goralski was supported by the Cystic Fibrosis Foundation, grant award nos. GORALS16C0 and GORALS12L0. This project was further supported by the Howard Holderness Distinguished Medical Scholar grant. Hardware support was also provided by Carestream Health Inc. (Rochester, NY).
Compliance with ethical standards
The scientific guarantor of this publication is Dr. Yueh Lee MD, PhD.
Conflict of interest
The authors of this manuscript declare relationships with the following companies:
Drs. Lee, Zhou and Lu are co-inventors of the stationary chest tomosynthesis imaging system evaluated in this study. Dr. Zhou has equity ownership and serves on the board of directors of Xintek, Inc., to which the technologies used or evaluated in this article have been or will be licensed. Dr. Lu has equity ownership in Xintek, Inc. All of these relationships are under management by the University of North Carolina's COI committees.
Statistics and biometry
Mrs. Agathe Ceppe, MS, kindly provided statistical advice for this manuscript.
Marsico Lung Institute, The University of North Carolina at Chapel Hill, Chapel Hill, NC 27599
Written informed consent was obtained from all subjects (patients) in this study.
Institutional Review Board approval was obtained.
• performed at one institution
- 1.Sly PD, Gangell CL, Chen L et al (2013) Risk factors for bronchiectasis in children with cystic fibrosis. N Engl J Med 368(21):1963–1970Google Scholar
- 3.(2016) Patient Registry Annual Data Report. Cyst Fibros Found Patient Regist. Available via https://www.cff.org/Research/Researcher-Resources/Patient-Registry/2016-Patient-Registry-Annual-Data-Report.pdf
- 4.Dournes G, Menut F, Macey J et al (2016) Lung morphology assessment of cystic fibrosis using MRI with ultra-short echo time at submillimeter spatial resolution. Eur Radiol 26(11):3811–3820Google Scholar
- 6.Dobbins JT 3rd, McAdams HP, Godfrey DJ, Li CM (2008) Digital tomosynthesis of the chest. J Thorac Imaging 23(2):86–92Google Scholar
- 7.Terzi A, Bertolaccini L, Viti A et al (2013) Lung cancer detection with digital chest tomosynthesis: baseline results from the observational study SOS. J Thorac Oncol 8(6):685–692Google Scholar
- 9.Vult von Steyern KV, Björkman-Burtscher I, Geijer M (2013) Radiography, tomosynthesis, CT and MRI in the evaluation of pulmonary cystic fibrosis: an untangling review of the multitude of scoring systems. Insights Imaging 4(6):787–798Google Scholar
- 10.Vult von Steyern KV, Björkman-Burtscher I, Geijer M (2012) Tomosynthesis in pulmonary cystic fibrosis with comparison to radiography and computed tomography: a pictorial review. Insights Imaging 3(1):81–89Google Scholar
- 11.Vult von Steyern KV, Björkman-Burtscher I, Weber L, Höglund P, Geijer M (2014) Effective dose from chest tomosynthesis in children. Radiat Prot Dosimetry 158(3):290–298Google Scholar
- 12.Lee YZ, Burk LM, Wang KH et al (2011) Prospective respiratory gated carbon nanotube micro computed tomography. Acad Radiol 18(5):588–593Google Scholar
- 13.Shan J, Tucker AW, Lee YZ et al (2015) Stationary chest tomosynthesis using a carbon nanotube x-ray source array: a feasibility study. Phys Med Biol 60(1):–81Google Scholar
- 14.Hartman AE, Shan J, Wu G et al (2016) Initial clinical evaluation of stationary digital chest tomosynthesis. Proc. SPIE 9783, Medical Imaging: Physics of Medical Imaging, 978366. Available from: https://doi.org/10.1117/12.2216850
- 15.Wu G, Inscoe C, Calliste J, Lee YZ, Zhou O, Jianping L (2015) Adapted fan-beam volume reconstruction for stationary digital breast tomosynthesis. Proceedings Volume 9412, Medical Imaging: Physics of Medical Imaging, 94123J. Available from: https://doi.org/10.1117/12.2081931
- 18.Choo JY, Lee KY, Yu A et al (2016) A comparison of digital tomosynthesis and chest radiography in evaluating airway lesions using computed tomography as a reference. Eur Radiol 26(9):3147–3154Google Scholar
- 19.Yamada Y, Jinzaki M, Hashimoto M et al (2013) Tomosynthesis for the early detection of pulmonary emphysema: diagnostic performance compared with chest radiography, using multidetector computed tomography as reference. Eur Radiol 23(8):2118–2126Google Scholar
- 23.(1991) 1990 Recommendations of the International Commission on Radiological Protection. Ann ICRP 21(1–3):1–201. Available from: http://www.icrp.org/publication.asp?id=icrp%20publication%2060
- 24.Wiest PW, Locken JA, Heintz PH, Mettler FA Jr (2002) CT scanning: a major source of radiation exposure. Semin Ultrasound CT MR 23(5):402–410Google Scholar
- 34.Beydon N, Amsallem F, Bellet M et al (2002) Pulmonary function tests in preschool children with cystic fibrosis. Am J Respir Crit Care Med 166(8):1099–1104Google Scholar
- 36.Vikgren J, Zachrisson S, Svalkvist A et al (2008) Comparison of chest tomosynthesis and chest radiography for detection of pulmonary nodules: human observer study of clinical cases. Radiology 249(3):1034–1041Google Scholar
- 37.Shan J, Burk L, Wu G et al (2015) Prospective gated chest tomosynthesis using CNT X-ray source array. Proceedings Volume 9412, Medical Imaging: Physics of Medical Imaging, 941242. Available from: https://doi.org/10.1117/12.2081701
- 38.Gomi T, Nakajima M, Fujiwara H et al (2012) Comparison between chest digital tomosynthesis and CT as a screening method to detect artificial pulmonary nodules: a phantom study. Br J Radiol 85(1017):e622–e629Google Scholar
- 40.Chan HP, Goodsitt MM, Helvie MA et al (2014) Digital breast tomosynthesis: observer performance of clustered microcalcification detection on breast phantom images acquired with an experimental system using variable scan angles, angular increments, and number of projection views. Radiology 273(3):675–685Google Scholar