High spatial resolution free-breathing 3D late gadolinium enhancement cardiac magnetic resonance imaging in ischaemic and non-ischaemic cardiomyopathy: quantitative assessment of scar mass and image quality
- 576 Downloads
To compare breath-hold (BH) with navigated free-breathing (FB) 3D late gadolinium enhancement cardiac MRI (LGE-CMR)
Materials and methods
Fifty-one patients were retrospectively included (34 ischaemic cardiomyopathy, 14 non-ischaemic cardiomyopathy, three discarded). BH and FB 3D phase sensitive inversion recovery sequences were performed at 3T. FB datasets were reformatted into normal resolution (FB-NR, 1.46x1.46x10mm) and high resolution (FB-HR, isotropic 0.91-mm voxels). Scar mass, scar edge sharpness (SES), SNR and CNR were compared using paired-samples t-test, Pearson correlation and Bland-Altman analysis.
Scar mass was similar in BH and FB-NR (mean ± SD: 15.5±18.0 g vs. 15.5±16.9 g, p=0.997), with good correlation (r=0.953), and no bias (mean difference ± SD: 0.00±5.47 g). FB-NR significantly overestimated scar mass compared with FB-HR (15.5±16.9 g vs 14.4±15.6 g; p=0.007). FB-NR and FB-HR correlated well (r=0.988), but Bland-Altman demonstrated systematic bias (1.15±2.84 g). SES was similar in BH and FB-NR (p=0.947), but significantly higher in FB-HR than FB-NR (p<0.01). SNR and CNR were lower in BH than FB-NR (p<0.01), and lower in FB-HR than FB-NR (p<0.01).
Navigated free-breathing 3D LGE-CMR allows reliable scar mass quantification comparable to breath-hold. During free-breathing, spatial resolution can be increased resulting in improved sharpness and reduced scar mass.
• Navigated free-breathing 3D late gadolinium enhancement is reliable for myocardial scar quantification.
• High-resolution 3D late gadolinium enhancement increases scar sharpness
• Ischaemic and non-ischaemic cardiomyopathy patients can be imaged using free-breathing LGE CMR.
KeywordsLate gadolinium enhancement, MRI High spatial resolution Myocardial infarction Non-ischaemic cardiomyopathy Free-breathing
Body mass index
High spatial resolution free-breathing
Normal spatial resolution free-breathing
Field of view
Full width half maximum
Late gadolinium enhancement cardiac magnetic resonance imaging
Magnetic resonance imaging
Phase-sensitive inversion recovery
Scar edge sharpness
Late gadolinium enhancement (LGE) imaging is an established cardiovascular magnetic resonance (CMR) technique to assess myocardial viability . LGE-CMR is commonly used in the diagnosis of ischaemic cardiomyopathy (ICM) and to guide revascularisation therapy. In addition, LGE-CMR has the ability to detect partially infarcted myocardial tissue, known as grey zone, which is believed to be a substrate for arrhythmia . Grey zone mass quantification using LGE-CMR has been shown to predict post-myocardial infarction mortality . In non-ischaemic cardiomyopathy (NICM), a specific LGE pattern allows for a differential diagnosis . More recently, scar mass quantification has been shown to predict adverse cardiovascular outcome in various non-ischaemic cardiomyopathies [4, 5, 6]. The increasing range of clinical applications warrants optimisation of current LGE-CMR protocols.
Standard LGE-CMR is performed during multiple breath-holds in several pre-defined two-dimensional (2D) orientations with an inversion recovery (IR) gradient echo sequence with the inversion time manually chosen to null the signal of healthy myocardium . Three-dimensional (3D) LGE-CMR is desirable because it allows (1) imaging of the whole heart without slice gaps, (2) imaging with high SNR and CNR, and (3) performing post-acquisition reformatting in any desired plane [7, 8]. However, 3D LGE-CMR is hampered by long breath-hold duration (>20 s) , thereby compromising its robustness in a clinical setting that includes vulnerable patients . To overcome these limitations, 3D LGE-CMR can be performed during free-breathing, for example with respiratory gating based on diaphragmatic pencil-beam navigation . The free-breathing approach allows to apply a phase sensitive inversion recovery sequence (PSIR) in combination with high spatial resolution (submillimeter voxel size). The PSIR sequence triggers inversions every other heartbeat with an additional proton density-weighted reference image acquired on alternate heartbeats. As a result, PSIR reconstructed images are less sensitive to wrong inversion time and heart rate variability as compared with standard inversion recovery . In theory, high spatial resolution LGE-CMR should result in better defined scar borders (specified as scar edge sharpness, SES), which could diminish partial volume averaging that contributes to scar and grey zone mass overestimation [11, 12]. To date, breath-hold 3D LGE-CMR has not been compared with free-breathing 3D LGE-CMR with matched spatial resolution. Furthermore, high spatial resolution free-breathing 3D LGE-CMR has not been compared with normal spatial resolution free-breathing 3D LGE-CMR in a clinical setting.
Therefore, the purpose of this study was to compare scar mass, grey zone mass, SES, and image quality of late gadolinium enhancement cardiac MRI using multiple breath-hold 3D phase sensitive inversion recovery imaging with matched normal spatial resolution navigated free-breathing 3D LGE-CMR (FB-NR), and to compare FB-NR with high spatial resolution navigated free-breathing 3D LGE-CMR (FB-HR) at 3.0 Tesla in ischaemic and non-ischaemic cardiomyopathy patients.
Navigated free-breathing LGE-CMR validation against standard breath-hold LGE-CMR was performed within the frame of clinical CMR protocol development. This retrospective study was considered a chart review. Therefore, written informed consent was waived by the hospital institutional review board.
From May 2012 until November 2013, breath-hold and free-breathing sequences were acquired in patients that were referred for LGE-CMR, if there was enough time in the clinical scanning slot. Of the 54 patients in which both BH and FB were performed, there were two incomplete free-breathing datasets (one acquisition was stopped due to allergic contrast reaction; one was stopped because of very low navigator efficiency). In one patient there was no LGE detectable. The remaining 51 patients were included (38 male; 13 female; mean age ± standard deviation: 59.9 years ± 11.4 years).
All CMR examinations were performed in our institution on a 3T unit (Ingenia, Philips Healthcare, Best, The Netherlands) with the patient in supine position. The body coil was used for transmission, and a 16-element anterior and 12-element posterior phased-array coil were used for signal reception. Following a standard clinical CMR protocol including 2D cine imaging in all cardiac axis views, the gadolinium-based contrast agent gadoterate meglumine (Dotarem, Guerbet, Villepinte, France) was administered intravenously at a dose of 0.15 mmol/kg. Ten minutes after contrast administration, a 2D gradient-echo T1 weighted sequence was used (i.e. Look-Locker) to visually determine the optimal inversion time (TI) of healthy myocardium. Approximately 10–15 min post-contrast, a whole heart high spatial resolution 3D gradient echo (T1 fast field echo) PSIR sequence was acquired during free breathing with diaphragmatic pencil-beam navigation. Image parameters were as follows: repetition time (TR) 4.15 ms; echo time (TE) 2.02 ms; TI was set at the null point plus 50 ms and ranged from 250 to 400 ms; flip angle (FA) 10°; field of view (FOV) 350 × 350 mm; matrix size 208 × 208; acquired pixel size 1.68 × 1.68 mm; reconstructed pixel size 0.91 × 0.91 mm; 71 transverse slices with 3.4 mm thickness and slice gap -1.7 mm; sensitivity encoding (SENSE) factor 3. The 2D pencil beam respiratory navigator was planned on the right hemidiaphragm. The acceptance window was set at 5 mm with a constant correction of 0.6. Saturation bands were planned on the navigator, vertebrae and thoracic subcutaneous adipose tissue to suppress residual motion artefacts. Acquisition time was 3 min 4 s assuming 100 % navigator efficiency and heart rate of 60 beats per minute (bpm). Directly after free-breathing LGE-CMR, BH was acquired using a 3D gradient echo PSIR sequence in short-axis slice orientation in two equal stacks during two breath-holds of 10–20 s duration. Subsequently, left two-chamber and four-chamber cardiac views were acquired using the same sequence. Typical scan parameters were: TR 4.31 ms; TE 2.09 ms; TI 250–400 ms (same as free-breathing); flip angle 10°; FOV 350 × 350 mm; matrix size 188 × 125 mm; acquired pixel size 1.86 × 2.8 mm; reconstructed pixel size 1.46 × 1.46 mm; 24 slices with 10 mm thickness and – 5 mm slice gap; SENSE factor 3.
MASS research software (Department of Radiology, Leiden University Medical Centre, Leiden, The Netherlands) was used for multiplanar reformatting and image analysis. BH data were analysed after offline fusion of data of the two breath-hold 3D stacks. Free-breathing 3D datasets were reformatted into two multiplanar reconstructions with identical geometry as BH: one with normal spatial resolution (FB-NR: 1.46 × 1.46 × 10 mm), matching BH, and one with high spatial resolution isotropic voxel reconstruction (FB-HR: 0.91 × 0.91 × 0.91 mm). (See Online Supplementary Material for a description of reconstruction method.) Images were analysed in random order. First, endocardial and epicardial contours were drawn manually. Second, regions with maximally hyperenhanced and hypoenhanced healthy myocardium were identified visually and a small region of interest was drawn. Third, myocardial scar was identified automatically using the validated full-width half maximum (FWHM) algorithm [2, 13, 14]. For this algorithm, maximal signal intensity (SImax) was defined as the mean signal intensity of the manually annotated hyperenhanced region, and minimal signal intensity (SImin) was defined as the mean signal intensity of the manually annotated hypoenhanced region. Myocardial scar was defined as voxels with signal intensity higher than 50 % (SImin + 0.5 × (SImax – SImin). Grey zone (only reported for ICM patients) was defined as voxels with signal intensity between 35 % (SImin + 0.35 × (SImax – SImin) and 50 % (SImin + 0.5 × (SImax – SImin). Fourth, manual adjustments to the automatically determined areas of scar were made only in cases of evident misinterpretation by the algorithm, for example in case of microvascular obstruction. Finally, scar and grey zone masses were calculated (in grams: specific weight of scar was set at 1.05 g/ml) using an in-house developed MATLAB script (Version R2012A, Natick, MA, USA).
Data are presented as mean ± standard deviation. BH sequence was compared to FB-NR for validation purposes. Subsequently, FB-NR was compared with FB-HR. Outcome measures were compared using paired-samples t-test. Agreement of scar mass was compared using Pearson’s correlation coefficient and Bland Altman analysis. A p value of less than 0.05 was considered to indicate statistical significance. Statistical analyses were performed using SPSS software (version 20, SPSS, Chicago, IL, USA).
Scar mass, all patients (n=48)
Scar mass was the same in BH versus FB-NR (mean ± SD: 15.5 g ± 18.0 g vs. 15.5 g ± 16.9 g, p = 0.997). Moreover, scar mass showed good correlation (r = 0.953) without systematic bias in Bland Altman analysis (mean difference ± SD: 0.00 g ± 5.47 g), as shown in the Online Supplementary Material. Limits of agreement were between −10.94 g and +10.93 g.
FB-NR yielded higher scar mass as compared with FB-HR (15.5 g ± 16.9 g vs. 14.4 g ± 15.6 g; p = 0.007). Although the data correlated well (r = 0.988), Bland Altman analysis demonstrated a systematic positive bias of + 1.15 g for FB-NR versus FB-HR (mean difference ± SD: + 1.15 g ± 2.84 g), as shown in the Online Supplementary Material. Limits of agreement were between –4.42 g and +6.72 g.
Scar and grey zone mass ICM patients (n=34)
In the ICM subgroup, scar mass was 18.9 g ± 20.2 g in BH versus 19.3 g ± 18.7 g in FB-NR, which was not significantly different (p = 0.670). In parallel with scar mass in all patients, FB-NR overestimated scar mass as compared with FB-HR in the ICM subgroup (19.3 g ± 18.7 g vs. 17.9 g ± 17.2 g, p = 0.01).
Grey zone was not significantly different between BH versus FB-NR (6.5 g ± 5.4 g vs. 6.0 g ± 6.0 g, p=0.128), and between FB-NR vs. FB-HR (6.0 g ± 6.0 g vs. 6.2 g ± 6.0 g, p=0.329).
Scar mass NICM patients (n=14)
In NICM patients scar mass was similar in BH vs. FB-NR (7.3 ± 6.1 g vs. 6.3 g ± 4.3 g, p=0.312). The difference in scar mass between FB-NR and FB-HR was not significant in this subgroup (6.3 g ± 4.3 g vs. 5.9 g ± 4.8 g, p=0.457).
As shown in the Online Supplementary Material, SNR and CNR were significantly lower in BH as compared with FB-NR (207.6 ± 85.8 vs. 1052.2 ± 877.2, p < 0.01 and 221.9 ± 70.9 vs. 864.9 ± 692.5, respectively). As a result of higher spatial resolution, SNR and CNR decreased significantly in FB-HR as compared with FB-NR (258.3 ± 94.9 vs. 1052.2 ± 877.2, p < 0.01 and 232.4 ± 91.5 vs. 864.9 ± 692.5, p < 0.01, respectively). In the ICM subgroup, SNR and CNR were lower in BH as compared with FB-NR (224.7 ± 87.3 vs. 1125.0 ± 1010.9, p < 0.01 and 240.4 ± 66.8 vs. 935.1 ± 789.1, p < 0.01, respectively). In FB-HR, SNR and CNR were lower as compared with FB-NR (272.9 ± 93.4 vs. 1125.0 ± 1010.9, p < 0.01 and 250.6 ± 89.1 vs. 935.1 ± 789.1, p< 0.01, respectively). Also in the NICM subgroup, SNR and CNR were lower in BH as compared with FB-NR (165.9 ± 67.9 vs. 875.5 ± 374.4, p < 0.01 and 177.1 ± 61.4 vs. 694.3 ± 330.3, p < 0.01, respectively), and lower in FB-HR as compared to FB-NR (222.8 ± 92.1 vs. 875.5 ± 374.4, p < 0.01 and 188.2 ± 84.5 vs. 694.3 ± 330.3, p < 0.01, respectively).
This study demonstrates that, as compared with multiple breath-hold 3D LGE-CMR, navigated free-breathing 3D LGE-CMR with matched spatial resolution allows reliable scar mass quantification with good image quality. Additionally, the free-breathing technique allows to image with high spatial resolution, thereby improving scar edge sharpness and reducing scar mass. This study involves ischaemic and non-ischaemic cardiomyopathy patients in a clinical setting which emphasises that the proposed sequence could be considered an alternative for breath-hold LGE-CMR at 3T systems.
Traditionally, LGE-CMR has been performed during multiple breath-holds. Breath-holds frequently cause slice misregistration, and repeated breath-holding can be challenging for vulnerable cardiomyopathy patients . This study shows that free-breathing 3D LGE-CMR yields higher SNR and CNR, as compared with breath-hold LGE-CMR with matched spatial resolution. Importantly, FB-NR allows reliable scar and grey zone mass quantification for ICM and NICM patients in a clinical setting on a 3 Tesla MR scanner. The real-life clinical setting also included vulnerable patients: population comprised 10 patients over 70 years of age, and several patients with heart failure and/or large transmural myocardial scar. Subgroup analysis of patients under and over 60 years of age revealed that there was no difference regarding image quality parameters between young and old patients (data not shown). Piehler et al. have found comparable results of motion-corrected free-breathing 2D PSIR sequence compared with a breath-hold 2D PSIR sequence with matched spatial resolution at 1.5 Tesla . This study extends their findings to the 3D-PSIR acquisition technique at 3 Tesla. Schultz et al. have compared breath-hold 3D-PSIR with 3D-IR LGE-CMR at 3T showing superior image quality of the former sequence . To our knowledge, this study is the first to compare a breath-hold 3D-PSIR sequence with a free-breathing 3D-PSIR sequence, with matched spatial resolution.
Because breath-holds can be challenging for patients, spatial resolution in breath-hold LGE-CMR is generally in the range of 5–10 mm in the through-plane direction. By relieving patients from repeated breath-holding, navigated free-breathing 3D LGE-CMR allows to image the whole heart without slice gaps with higher spatial resolution (especially in the through-plane direction). Our data show that, in FB-HR images, the scar border is better delineated (i.e. less blurry), as compared with FB-NR, supported by a 22.7 % relative increase in SES. Despite the increase in spatial resolution, SNR and CNR remained within the range of BH images. While others have shown subjective improvement of image sharpness of a high spatial resolution free-breathing 3D sequence against a normal spatial resolution breath-hold 2D sequence at 1.5 Tesla , our study is the first to objectively quantify SES in LGE-CMR. As shown in Fig. 6, we propose that increased SES in FB-HR results in smaller scar mass by reducing partial volume effects. Jablonowski et al. have shown that normal spatial resolution LGE-CMR overestimates scar mass, as compared with a high spatial resolution ex vivo sequence . As such, it may be assumed that FB-HR reduces scar mass overestimation. Aforementioned study by Peters et al. found no difference in scar mass between a high spatial resolution (0.62 × 0.62 × 2.5 mm) free-breathing 3D inversion recovery sequence and a normal spatial resolution (1.2 × 1.2 × 8 mm) breath-hold 2D inversion recovery sequence in 14 ICM patients . The discrepancy between our results and theirs might be explained by our larger difference in spatial resolution and different cohort size. Furthermore, we used two reconstructions from the same 3D dataset instead of two separately acquired sequences. By doing so, other variables than voxel size in itself are eliminated. Scar mass was not significantly smaller in FB-HR than FB-NR in the NICM subgroup analysis. This lack of difference is possibly related to the small sample size, and the fact that the FWHM method has lower reproducibility in NICM as compared with ICM .
Contrary to scar mass, grey zone mass did not differ between FB-NR and FB-HR. As illustrated in Fig. 6, the difference between the signal intensity decay of FB-NR and FB-HR diminishes at the transition area of scar to grey zone. As a result, the difference in grey zone mass is reduced to a non-significant level. It is important to note that the region characterised as grey zone probably shifts towards the endocardium in FB-HR. This suggest that FB-HR might localize grey zone more precisely. Schelbert et al. have performed ex vivo LGE-CMR with spatial resolutions ranging from the cellular level to approximately the level of our FB-HR reconstruction . They found a positive linear correlation between grey zone mass and voxel size, explained by a reduction in partial volume averaging at the border of definite scar and healthy myocardium. However, whether their results in the ex vivo rat heart can be extrapolated to in vivo human clinical resolution images remains uncertain. Another discrepancy with our study is the method used for grey zone quantification. The FWHM method used in our study has been shown to have higher reproducibility as compared with the standard deviation threshold method used by Schelbert et al. [2, 13].
From a clinical perspective, navigated free-breathing 3D LGE-CMR has the advantage to relieve patients from repeated breath-holding and generate images with superior image quality. In theory, high spatial resolution LGE-CMR could enable more accurate visualisation of specific patterns of (very) small scar lesions, especially critical in the case of NICM. In ICM, sharper delineation of scar might better define degree of transmurality, or define the location of grey zone area, for example to improve pre-procedural guidance of ablation therapy in arrhythmia patients. Furthermore, in a research setting, scar mass quantification is often applied in (non-) ICM to correlate with adverse cardiovascular outcome. This study shows that results from LGE-CMR protocols with varying spatial resolutions should be interpreted with caution. Also, application of a high spatial resolution free-breathing 3D LGE-CMR might increase power of clinical studies that include LGE-CMR. Finally, the longer scan duration of high spatial resolution free-breathing 3D LGE-CMR must be taken into account. The scan duration of 9–10 min might be challenging for the most vulnerable or critically ill patients. In such settings, voxel size might be reduced. Alternatively, the number of slices could be reduced to include only the left ventricle instead of the whole heart. It needs to be stressed that a single free-breathing 3D LGE dataset allows reformatting in any desired plane and scan duration should be compared with acquisition of multiple 2D views.
Our study has several limitations. First, the reference standard of scar mass quantification is histopathological staining which was not possible to perform in this study. Second, the non-random order of sequences could have compromised SNR and CNR in breath-hold LGE-CMR (which was always performed after free-breathing LGE-CMR) due to contrast wash-out. However, both sequences were performed with PSIR compensating for suboptimal TI selection due to wash out [18, 19]. In our view, the four- to fivefold increase of SNR and CNR could not be explained by the scan order. Another limitation is the retrospective character of the study which did not allow evaluating the robustness of the proposed sequence.
In a clinical setting including ischaemic and non-ischaemic cardiomyopathy patients, normal spatial resolution navigated free-breathing 3D LGE-CMR allows reliable scar mass quantification with higher SNR and CNR as compared with multiple breath-hold 3D LGE-CMR. High spatial resolution free-breathing 3D LGE-CMR has improved scar edge sharpness and reduced scar mass, as compared with normal spatial resolution free-breathing 3D LGE-CMR. High spatial resolution free-breathing 3D LGE-CMR might improve scar mass quantification and diagnostic performance, whilst relieving the patient from repeated breath-holding. In addition, free-breathing 3D LGE-CMR allows post-acquisition reformatting in any desired imaging plane.
We would to thank Gerrit Kracht for editing the figures.
The authors state that this work has not received any funding.
Compliance with ethical standards
The scientific guarantor of this publication is Prof H.J. Lamb.
Conflict of interest
The authors of this manuscript declare no relationships with any companies whose products or services may be related to the subject matter of the article.
Statistics and biometry
No complex statistical methods were necessary for this paper.
Written informed consent was waived by the Institutional Review Board.
Institutional Review Board approval was not required because the institution has stated that informed consent was waived because all scans were performed within clinical work-up.
• performed at one institution
- 2.Roes SD, Borleffs CJ, van der Geest RJ et al (2009) Infarct tissue heterogeneity assessed with contrast-enhanced MRI predicts spontaneous ventricular arrhythmia in patients with ischemic cardiomyopathy and implantable cardioverter-defibrillator. Circ Cardiovasc Imaging 2(3):183–190CrossRefPubMedGoogle Scholar
- 5.Gao P, Yee R, Gula L et al (2012) Prediction of arrhythmic events in ischemic and dilated cardiomyopathy patients referred for implantable cardiac defibrillator: evaluation of multiple scar quantification measures for late gadolinium enhancement magnetic resonance imaging. Circ Cardiovasc Imaging 5(4):448–456CrossRefPubMedGoogle Scholar
- 8.Kido T, Kido T, Nakamura M et al (2014) Three-dimensional phase-sensitive inversion recovery sequencing in the evaluation of left ventricular myocardial scars in ischemic and non-ischemic cardiomyopathy: comparison to three-dimensional inversion recovery sequencing. Eur J Radiol 83(12):2159–2166CrossRefPubMedGoogle Scholar
- 17.Peters DC, Appelbaum EA, Nezafat R et al (2009) Left ventricular infarct size, peri-infarct zone, and papillary scar measurements: A comparison of high-resolution 3D and conventional 2D late gadolinium enhancement cardiac MR. J Magn Reson Imaging 30(4):794–800CrossRefPubMedPubMedCentralGoogle Scholar
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.