MR-guided breast biopsy at 3T: diagnostic yield of large core needle biopsy compared with vacuum-assisted biopsy
- First Online:
- Cite this article as:
- Meeuwis, C., Veltman, J., van Hall, H.N. et al. Eur Radiol (2012) 22: 341. doi:10.1007/s00330-011-2272-6
- 699 Downloads
The purpose of this study was to evaluate two MR-guided biopsy techniques at 3 T, large core needle breast biopsy (LCNB) and vacuum-assisted breast biopsy (VAB) and to compare the diagnostic yield and rate of complications to determine the optimal biopsy technique at 3 T.
55 LCNB and 64 VAB were consecutively performed. Benign biopsy results were verified by retrospective correlation of histology, with pre-interventional, post-interventional MRI studies and follow-up and were classified as representative or non-representative. Time to follow-up was up to 2 years for the considered non-representative benign lesions. Statistical analysis was performed using the Chi-squared test.
LCNB was technically successful in 100% of patients (55/55) and VAB in 98% of patients (63/64). Histopathological analysis resulted in 45 (82%) benign, 3 (5%) high-risk and 7 (13%) malignant lesions for LCNB and 43 (67%) benign, 3 (5%) high-risk and 18 (28%) malignant lesions. Distribution was significantly different (p < 0.001), favouring VAB over LCNB.
Because of the substantially higher diagnostic yield and certainty of a benign diagnosis, VAB is the optimal biopsy technique at 3 T. LCNB should be considered when VAB is not feasible.
KeywordsBreast cancer Biopsy MRI High field Neoplasms
The use of Magnetic Resonance Imaging (MRI) for the detection and evaluation of breast lesions continues to increase. MRI of the breast has high sensitivity and lower specificity in the evaluation of breast lesions, with sensitivity of 90% and a specificity of 72% .
The increasing use of magnets with high field strength (3 Tesla [T] and over) in clinical practice is very suitable. With the high signal-to-noise ratio of these systems, appropriate acquisitions can be used to achieve high spatial and temporal resolution that allows depiction of lesion morphology and lesion enhancement with unprecedented accuracy [2, 3, 4]. Compared with the results of 1 T and 1.5 T MRI in the literature, 3 T may have an even higher sensitivity to breast cancer . Nevertheless, even at 3 T it is often not possible to separate malignant from benign findings purely on imaging characteristics alone. Accurate diagnosis of these lesions requires histological evaluation.
Magnetic resonance imaging of the breast reveals suspicious lesions occult at target ultrasound or mammography in about 44% of the cases [6, 7]. In these suspicious MRI breast lesions MRI-guided biopsy is the technique of choice. Moreover, MRI-guided biopsy is indicated in all lesions where an ultrasound or mammographically detected and biopsied lesion is possibly not the same lesion as that detected at MRI. The demand for MRI-guided breast interventions is growing owing to the increasing use of breast MRI in clinical practice.
Magnetic resonance-guided tissue sampling of suspicious MRI breast lesions, can be accomplished by fine-needle aspiration, wire localisation followed by surgical excision, large core-needle biopsy (LCNB), or by vacuum-assisted biopsy (VAB) [8, 9].
Magnetic resonance-guided LCNB using 14 G needles and VAB have both been described as suitable techniques for MRI-guided breast biopsy at 1.5 T [10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21]. At 3 T only MRI-guided LCNB has been evaluated [22, 23]. Moreover, no studies directly compare the diagnostic yield of LCNB with VAB at any field strength. In this retrospective study we evaluated both methods at 3 T and compared the diagnostic yield and rate of complications to determine the optimal biopsy technique at 3 T.
Materials and methods
Indication for MRI examination in 147 patients
High risk screening
–Positive family history or BRCA positive
–History of high risk lesion or cancer
–Questionable lesion mammography/ultrasound
–Positive axillary node, unknown primary
Staging of known breast cancer
The level of suspicion was reported on a scale of 0–6, identical to that in the lesion-assessment categories used in the breast imaging reporting and data system (BI-RADS) . Numeric categories were the following: 0, needs additional imaging evaluation; 1, normal; 2, benign; 3, probably benign; 4, suspicious; 5, highly suggestive of malignancy; 6 proven malignancy. Classification was based on lesion morphology and enhancement kinetics . For lesions interpreted as probably benign, suspicious or highly suggestive of malignancy on MRI, correlative ultrasound was performed to determine if the lesion was ultrasonographically evident and, thereby amenable to tissue sampling under ultrasound guidance. All patients with a BIRADS 3, 4 or 5 lesions that were ultrasonographically occult were scheduled for a MRI-guided breast biopsy. In 80% of the biopsied cases there was a mass lesion, in 13% of the biopsied cases there was non mass like enhancement and in 7% of the cases there was a focus.
All MRI-guided breast biopsies were performed in a clinical setting. The study has been carried out in the Netherlands in accordance with the applicable rules concerning the review of research ethics committees and informed consent.
MRI-guided biopsy procedure and equipment
All biopsies were performed on a Siemens Magnetom Trio system, using a dedicated four-channel open breast coil (Invivo Interventional Instruments, Wurzburg, Germany), with an add-on device for needle positioning (Noras, Germany).
Field of view (mm2)
340 × 170
Effective slice thickness (mm)
Number of acquisitions
1 before and 3 after contrast medium
Number of slices
Acquired voxel size (mm3)
0.9 × 0.9 × 1
Temporal resolution (s)
MRI-guided large core needle biopsy
MRI-guided vacuum-assisted biopsy
A coaxial sheath was placed using a 9-gauge or 10-gauge titanium stylet in the appropriate grid location. The stylet was replaced by a sterile plastic MRI-visible obturator. If a FLASH 3D proved a good correlation, the plastic obturator was replaced by one of the respective VAB devices and up to 12 biopsies were taken.
Data collection and statistic analysis
All biopsies were performed by three experienced MRI-breast radiologists who had a lot of experience with MRI-guided LCNB and less with MRI-guided VAB. All three radiologist had more than 3 years experience with MRI-guided LCNB and had done more than 50 procedures each. MRI-guided VAB was only done at 1.5 T for 1 year and all three radiologist had done about 10 procedures each, before they started to do the same procedure at 3 T.
It is possible to get the needle on the right spot
The biopsy can be safely performed according to the performing physician
The lesion was removed completely or partially according to the post-VAB imaging.
We judged VAB not possible, if the risk of skin tears, severe bleeding, or severe pain in the patient was high. Complications, such as pain and arterial bleeds were registered and were in this study not considered as technical unsuccessful.
After histopathological analysis, the lesions were classified in three categories: benign, high-risk or malignant. Malignant lesions and high-risk lesions were surgically removed. In the case of a benign lesion, the images of the biopsy procedure were re-evaluated to determine whether the tissue samples were representative and at least 6-month follow-up MRI was recommended. Total time to follow-up was 2 years in case of unchanged lesions.
Statistical analysis was done with SPSS 12.0 and using the Mann–Whitney U test. Significance was assumed at a level of p < 0.05.
In our patient series, LCNB was technically successful in 100% of patients (55/55) and VAB was technically successful in 98% of patients (63/64) (n.s.).
According to the post-interventional imaging 1 of the 72 VAB procedures was unsuccessful and immediate rebiopsy was performed. Assessment was considered impaired, but successful in 1 LCNB procedure and 3 VAB procedures (n.s.). Motion artefacts occurred in 1 patient at LCNB. Histological analysis of the lesion revealed normal breast tissue. After more than 1 year of follow-up, no signs of malignancy were found.
The lesion size ranged from 3 to 28 mm with a median lesion size of 12 mm, 50% were smaller than 1 cm (28 LCNB and 32 VAB) and 8% of all the lesions were smaller than 5 mm (foci) and were all biopsied with MRI-guided VAB.
LCNB (n = 55)
VAB (n = 64)
Normal breast tissue
In the LCNB group histopathological analysis of the biopsy samples resulted in benign lesions in 45/55 (82%), in high-risk lesions in 3/55 (5%) and in malignant lesions in 7/55 (13%). In the VAB group this resulted in 43/64 (67%) benign lesions, 3/64 (5%) and 18/64 (28%) malignant lesions (Fig. 5).
We evaluated LCNB and VAB using a 3 T clinical MR system and compared the diagnostic yield and rate of complications to determine the optimal biopsy technique at 3 T.
Published data on the use of MRI-guided biopsy at 1.5 T reported a technical success rate of 95% to 100% and cancer yields ranging from 20% to 56% for LCNB [10, 11, 15], and a technical success rate of 96% to 100% and cancer yields ranging from 24% to 40% for VAB [16, 17, 18, 19, 20, 21]. These numbers are comparable to the accepted rates reported in mammography- and ultrasound-guided procedures . In our patient series, in the LCNB group 13% were found to be malignant after biopsy and 28% in the VAB group. LCNB was technically successful in 100% of patients (55/55) and VAB was technically successful in 98% of patients (63/64), but more importantly VAB was significantly more often representative than LCNB.
Magnetic resonance-guided LCNB using 14 G needles at 1.5 T have been reported by several groups [10, 11, 12, 13, 14, 15]. Kuhl et al.  reported the largest series of MRI-guided LCNB in 59 patients with 78 lesions. One carcinoma was missed, and histological underestimations were not observed. Chen et al.  reported 34 successful LCNB in 35 lesions in 29 women and 6% (2/35) were underestimated carcinoma. Two lesions with atypical ductal hyperplasia were upgraded to malignancy after surgery. In a multicentre European trial Perlet et al.  reported successful VAB in 517 (96%) of 538 lesions. Histological underestimation was observed in 8 patients (2%). An upgrade was noted in 5 patients who had a diagnosis of ADH, and in 3 patients who had a diagnosis of DCIS.
The biopsies we performed, 20% of the LCNB were considered unrepresentative biopsies and were excluded from this study. After chirurgical excision biopsy 7% of carcinomas were missed and histological underestimation was present in 6%. VAB showed 4 unrepresentative biopsies, 2 missed carcinomas and histological underestimation was not observed. Eight of the 14 non-representative LCNB were smaller than 1 cm (57%). Therefore, LCNB biopsy at 3 T is not recommended for lesions smaller than 10 mm.
Two of our VAB procedures were converted to LCNB because of the location of the lesion (too close to the skin and the other lesion was too close to the chest wall), which is in part attributable to the more rigid biopsy technique with the grid system. Using angulation of the needle will probably allow VAB in a number of patients in whom the technique first failed. In another patient scheduled for LCNB a 14 G needle could not enter the breast because of the solid consistency of the breast. Finally a successful biopsy was performed with an 18 G needle.
The chance of detecting malignancy is very much dependent on the à priori chance of the biopsied lesion being malignant. It is important to realise that it is essential to compare the results of the pathological analysis of the biopsy specimen with the lesion seen at MRI. A diagnosis of fibrosis with chronic inflammation is usually acceptable in lesions with no mass effect classified as BI-RADS 3 to BI-RADS 4a. However, this diagnosis is absolutely insufficient in lesions classified as BI-RADS 5, as well as in any lesion presenting as a clear mass.
For BIRADS III (probably benign) lesions, at mammography a 6-month short-term follow-up is recommended, because the frequency of carcinoma in these lesions is low, ranging from 0.5% to 2% [26, 27, 28, 29]. Published results show a wide range of cancer yields (0.6% to 10%) among probably benign MRI lesions [30, 31, 32, 33, 34, 35]. There are a few indications for which mammographic stereotactic biopsy of BIRADS III lesions may be considered, such as when follow-up is compromised or not available, in the instance of planned pregnancy, or for patient anxiety [27, 36, 37, 38]. Eighteen percent (41 lesions) of all the MRI-guided biopsied lesions were classified as BIRADS III breast lesions, because a lot of women nowadays do not except a 6 month follow-up and prefer a biopsy directly. Three percent of these lesions were found malignant at histopathologic analysis.
Magnetic resonance-guided LCNB using 14 G needles and VAB have both been described as suitable techniques for MRI-guided breast biopsy at 1.5 T [8, 9], although several potential limitations of LCNB have been described at 1.5 T or lower . Compared with 1.5 T, at 3 T an extra potential limitation of the LCNB technique has been described by Peters et al. . Susceptibility artefacts increase with increasing field strength. This can cause a substantial signal void around the 14 G needle when performing MRI-guided LCNB at 3 T.
The certainty of the histopathological diagnosis increases with an increase in the amount of tissue extracted, especially in a situation where there is no real-time feedback of the needle position during biopsy, as is the case in MRI-guided breast biopsy. In lesions with no mass effect LCNB may not sample enough material to ascertain diagnosis. In mass lesions the non-rigid breast tissue may allow a small solid lesion to move during biopsy, causing the needle to pass right beside the lesion, rather than through its core.
Our study has several limitations. First of all in our study population most patients with a representative benign lesion had a 6 month follow-up instead of a 24 months follow up which is necessary to make them eligible to be considered as truly benign.
An important limitation of this study is the fact that the two biopsy techniques were compared in a sequential and not in a parallel way. Lots of covariates could influence the results of the two study arms, the most important being the fact that experience was much higher for the second biopsy technique, resulting in a much steeper learning curve and possibly better results. Also the fact that two different vacuum assisted biopsy devices have been used is another potential limitation as the study subgroup becomes heterogeneous.
In our study up to 12 VAB were taken instead of >24 core (11-Gauge) specimens, which is recommended in the latest European consensus meeting .
Another very important limitation is the highly heterogeneous study population. The fact that patients were referred from 22 different hospitals for MRI-guided biopsy and the consecutively study population is a major limitation. Although all breast MRI’s from the different hospitals, had a second reading from a MRI-breast radiologist in our institution, before the biopsy and in doubtful cases an additional diagnostic breast MRI was made in our institution, it is unthinkable that equal clinical guidelines for such referrals apply for all hospitals or that experience levels match for all those different radiologists.
Compared with the results of 1 T and 1.5 T MRI in the literature, 3 T may have higher sensitivity in the detection of breast cancer . Nevertheless the specificity is comparable with the results of 1.5 T . Demand for adequate MRI-guided tissue sampling remains, also at high field strength. Our study shows that MRI-guided VAB should be used for all MRI-guided breast biopsies at 3 T because of the substantially higher diagnostic yield and increased certainty of a benign diagnosis, compared with LCNB. LCNB should be considered when VAB is unavailable or not feasible.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.