High-resolution spiral CT of the breast at very low dose: concept and feasibility considerations
- 968 Downloads
Mammography, today’s standard imaging approach, has deficits with respect to the superimposition of anatomical structures. Dedicated CT of the breast so far indicated that it can provide superior soft-tissue imaging, but that it still has significant limitations with respect to spatial resolution and dose. We have assessed novel dedicated breast CT technology.
Based on simulations and measurements we developed novel technology which uses direct-conversion CdTe material and photon-counting electronics with 100 μm detector element size for close to 100% dose efficiency. We assessed the potential for the imaging of microcalcifications of 100 to 200 μm diameter and soft-tissue lesions of 1 to 5 mm diameter by simulations at dose levels between 1 and 6 mGy.
Microcalcifications of 150 μm and soft-tissue lesions of 2 mm diameter were found to be clearly detectable at an average glandular dose of 3 mGy. Separate displays are required for high-resolution microcalcification and for low-resolution soft-tissue analysis. Total CT data acquisition time will be below 10 s.
Dedicated breast CT may eventually provide comprehensive diagnostic assessment of microcalcifications and soft-tissue structures at dose levels equivalent to or below those of two-view screening mammography.
KeywordsComputed tomography (CT) Breast Image quality Spatial resolution Dose efficiency
- 15.Kalender WA (2009) High-resolution CT of the breast: a proposal! Eur Radiol 19(Suppl 4):849–852Google Scholar
- 16.European Reference Organisation for Quality Assured Breast Screening and Diagnostic Services (EUREF) (2008) European guidelines for quality assurance in breast cancer screening and diagnosis, Fourth Edition. http://www.euref.org/index.php?option=com\_content\&view=article\&id=5\&Itemid=25. Accessed May 26, 2011
- 17.American College of Radiology (2008) ACR Practice guidline for performance of Screening and diagnostic mammography. http://www.acr.org/secondarymainmenucategories/quality_safety/guidelines/breast/screening_diagnostic.aspx. Accessed May 26, 2011
- 19.EC EURATOM 7th Framework Program “Dedicated CT of the female breast: Feasibility, optimization and comparison to competing imaging modalities”, Contract No. FP/213153 (2008–2010) PI: W. A. Kalender. http://www.imp.uni-erlangen.de/BreastCT/index.html. Accessed May 26, 2011
- 20.Deutsche Forschungsgemeinschaft Research Unit FOR 661 “Multimodal Imaging for Preclinical Research”, DFG Az. KA 1254/11-2 (2009–2012) PI: W. A. Kalender. http://www.imp.uni-erlangen.de/for661/index.htm. Accessed May 26, 2011
- 21.Hendrick RE, Pisano ED, Averbukh A et al (2010) Comparison of acquisition parameters and breast dose in digital mammogrphy and screen-film mammography in the American College of Radiology Imaging Network Digital Mammographic Imaging Screening Trial. Am J Radiology 194:362–369Google Scholar
- 23.Kalender WA (2011) Computed tomography. Publicis Corporate Publishing, ErlangenGoogle Scholar
- 27.International Commission on Radiation Units and Measurements (1989) ICRU report 44: tissue substitutes in radiation dosimetry and measurementGoogle Scholar
- 31.Bundesministerium für Bildung und Forschung (BMBF) Spitzencluster Medical Valley, Verbund Bildgebende Diagnostik: Leitprojekt Brust CT, Az. 01EX1002 PI: W. A. KalenderGoogle Scholar