Spontaneous Solitaire™ AB Thrombectomy Stent Detachment During Stroke Treatment
- 391 Downloads
Spontaneous Solitaire™ stent retriever detachment is a rarely defined entity seen during stroke treatment, which can result in a disastrous clinical outcome if it cannot be solved within a critical stroke treatment time window. Two solutions to this problem are presented in the literature. The first is to leave the stent in place and apply angioplasty to the detached stent, while the second involves surgically removing the stent from the location at which it detached. Here, we present a case of inadvertent stent detachment during stroke treatment for a middle cerebral artery M1 occlusion resulting in progressive thrombosis. The detached stent was removed endovascularly by another Solitaire stent, resulting in the recanalization of the occluded middle cerebral artery.
KeywordsStroke Thrombectomy Angiography
Conflict of interest
The authors Süha Akpınar and Güliz Yılmaz state that there is no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from the patient included in the study.
- 2.Kang DH, Park J, Hwang YH, Kim YS (2013) Inadvertent self-detachment of Solitaire AB stent during the mechanical thrombectomy for recanalization of acute ischemic stroke: lessons learned from the removal of stent via surgical embolectomy. J Korean Neurosurg Soc 53:360–363CrossRefPubMedCentralPubMedGoogle Scholar
- 3.Mordasini P, Zubler C, Schroth G, Gralla J (2012) Thrombectomy for acute ischemic stroke treatment: a review. EJMINT Invited Review: 1238000077, Epub 21 Sep 2012Google Scholar
- 5.U.S. Food and Drug Administration (2012) 510(k) Premarket Notification Database. Solitaire FR Revascularization Device. Summary of Safety and Effectiveness. No. K113455. Rockville, MD: FDA. 2 March 2012. http://www.accessdata.fda.gov/SCRIPTS/cdrh/devicesatfda/index.cfm?db=pmn&id=K113455 (accessed 7 Apr 2013)
- 6.Covidien ev3 Neurovascular, Inc (2012) Instructions for Use Solitaire FR Revascularization Device [supplied with the device in the product package]. ev3 Neurovascular, Inc. 70648–001 Rev. 02/12Google Scholar