Frequent Embolization in Peripheral Angioplasty: Detection with an Embolism Protection Device (AngioGuard) and Electron Microscopy
- 93 Downloads
Purpose: To evaluate the deliverability and protection capabilities of an embolism protection filter in angioplasty of peripheral arteries. Methods: The Angioguard emboli capture guidewire system was applied in 11 patients with femoropopliteal lesions (6 stenoses, 3 occlusions, 2 controls). Data on lesion crossing, flow deceleration and macroembolization were recorded. Filter membranes were evaluated with scanning electron microscopy (SEM). Results: System delivery was successful in all patients. Primary lesion crossing was feasible in four of six stenoses; predilatation was required in two of six. Marked flow deceleration was recorded in six patients. Emboli next to the filter were detected in each patient with concentric plaques, but could not reliably be removed with the filter. Downstream macroembolization was also present in all patients with concentric stenoses, but in none with chronic occlusion. None of the patients had clinical signs of ischemia. SEM analysis demonstrated only small particles on control group filters and non-obliterating fibrinous conglomerates on filters used in chronic occlusion. Substantial obliteration was seen on several filters used in stenotic lesions. Conclusion: Microembolization of fibrin aggregates is a common incident in balloon angioplasty of femoropopliteal stenoses. Macroembolization occurred more frequently than previously reported. The use of embolism protection filters aided in the detection but not in the removal of larger emboli.
KeywordsAngioplasty, peripheral vessels Embolization Embolization protection
We thank Cordis (Haan, Germany; a Johnson & Johnson Company, Warren, NJ, USA) for providing the AngioGuard devices. We are grateful to Michael Bretzke for assistance with the study.
Presented at CIRSE annual meeting, September 22–26, 2001, Gothenburg, Sweden.
- 2.Müller-Hülsbeck, S. 2002Cerebral protection devices for treatment of carotid stenoses: Where do we stand today?Fortschr Röntgenstr174945954Google Scholar
- 6.McDermott, JC, Crummy, AB 1994Complications of angioplasty.Semin Interv Radiol11145149Google Scholar
- 12.Abraham, P, L’hoste, P, Pasco-Papon, A, Saumet, JL 1996Asymptomatic peripheral embolism during balloon angioplasty.AJR Am J Roentgenol16612301231Google Scholar
- 13.Zanella, FE, Berkefeld, J 2000Protected carotid stenting.Z Kardiol894752Google Scholar
- 14.Martin, JB, Pache, JC, Treggiari-Venzi, M, Murphy, KJ, Gailloud, P, Puget, E, Pizzolato, G, Sugiu, K, Guimaraens, L, Théron, J, Rüfenacht, DA 2001Role of distal balloon protection technique in the prevention of cerebral embolic events during carotid stent placement.Stroke32479484PubMedGoogle Scholar
- 16.Manninen, HI, Räsänen, HAT, Vanninen, RL, Vainio, P, Hippeläinen, M, Kosma, VM 1999Stent placement versus percutaneous transluminal angioplasty of human carotid arteries in cadavers in situ: Distal embolization and findings at intravascular US, MR imaging, and histopathologic analysis.Radiology212483492PubMedGoogle Scholar