High survival in young patients using a second generation uncemented total hip replacement
- 1.6k Downloads
Whilst excellent long-term results with contemporary uncemented stems have been reported for total hip arthroplasty in young patients, the survival rates for the whole reconstruction are often compromised by high failure rates on the acetabular site due to peri-acetabular osteolysis and accelerated wear.
In patients 60 years old or younger, we retrospectively reviewed the results at a minimum of ten years of 89 consecutive uncemented total hip replacements in 88 patients using the press-fit Fitmore acetabular component in combination with the CLS Spotorno stem and a 28-mm Metasul metal-on-metal articulation or a 28-mm alumina ceramic on conventional polyethylene bearing. The mean age at the time of surgery was 49 years (range, 25–60). The mean clinical and radiological follow-up was 12 years (range, 10–15).
Six patients (six hips) died and two patients (two hips) were lost to follow-up. Five hips were revised: one for deep infection, one for peri-prosthetic femoral fracture, and one for aseptic stem loosening. In two hips an isolated revision of the acetabular liner was performed (one for recurrent dislocation and one for unexplained pain). No revision was performed for accelerated wear, osteolysis or aseptic loosening of the acetabular shell. We could not detect peri-acetabular osteolysis visible on plain radiographs in those hips evaluated radiographically. The Kaplan-Meier survival with revision for any reason as the endpoint was 94% (95% confidence interval, 86–97) at 12 years.
The survival rates and the radiological outcomes with this implant combination in this young and active patient group are encouraging when compared to the results reported for other uncemented cups in this age group.
KeywordsFemoral Component Heterotopic Ossification UHMWPE Aseptic Loosening Acetabular Component
One of the most challenging problems in total hip arthroplasty (THA) remains the management of the young patient. Good and excellent long-term survival rates for contemporary uncemented femoral components have been reported [1, 2, 3, 4], even in young patients . There is evidence in the current literature that the acetabular site is critical in uncemented THA, especially in young patients [6, 7, 8]. Despite encouraging long-term fixation of the metallic shell , long-term survival of uncemented cups is frequently compromised by excessive liner wear and peri-acetabular osteolysis [6, 7, 10], particularly in young and active patients [7, 9, 11, 12].
We report the results at 12 years of an independent series using a contemporary uncemented modular titanium press-fit cup in combination with a long-term proven contemporary uncemented femoral component [2, 5] in a cohort of young and active patients.
Patients and methods
Patient demographics and distribution of hips
Age, mean (range)
Hips, no. (%)
Body mass index (kg/m²), mean (range)
Number of hips (%)
A straight collarless double-tapered titanium stem (CLS Spotorno stem; Zimmer, Winterthur, Switzerland) was implanted without cement in all patients with use of the press-fit technique described by Spotorno et al. .
The bearing surfaces used were either a 28-mm Metasul (Zimmer, Winterthur, Switzerland), forged, high-carbide ([C], 0.2–0.25%) CoCr alloy metal-on-metal (63 hips) or a 28-mm Al2O3 ceramic head (Biolox forte; CeramTec, Plochingen, Germany) articulating with conventional UHMW-Polyethylene (Sulene; Zimmer, Winterthur, Switzerland) ceramic-on-polyethylene (25 hips). This conventional UHMW-PE is made from compression-molded, calcium stearate free GUR 1020 resin and sterilized by gamma irradiation at a dose of minimum 25 kGy in a nitrogen environment. In one hip, a 22-mm metal-on-polyethylene articulation was used.
Either a modified Watson-Jones or a transgluteal Bauer approach to the hip was used with the patient in the supine position. Postoperatively, partial weight-bearing for six weeks was recommended. Routine prophylaxis for the prevention of heterotopic ossification was diclofenac 50 mg twice per day for three weeks. Intravenous third-generation cephalosporin (1.5 g cefuroxime) was administered peri-operatively. Anticoagulation therapy consisted of low-molecular weight heparin (enoxaparin) administered subcutaneously the day before surgery and continued for six weeks postoperatively.
For radiographic evaluation, standard pelvis anteroposterior (AP) and lateral radiographs of the hip were taken. The inclination, anteversion , horizontal and vertical positions of the acetabular component were determined, taking the inter-teardrop line as a fixed landmark, as described by Massin et al. . The distribution of radiolucent gaps or osteolysis at the acetabular bone-prosthesis interface was recorded using the three zones described by DeLee and Charnley . Osteolysis was defined as a lucent zone devoid of trabecular bone and usually with a sclerotic border and classified as minor osteolysis (≤1.5 cm²) or major osteolysis (>1.5 cm²) [8, 12, 19].
Radiolucent lines or osteolysis at the femoral bone–prosthesis interface was evaluated using the zones described by Gruen et al. . Femoral osteopenia due to stress shielding and femoral component fixation were graded according to the criteria described by Engh et al. . Heterotopic ossification was graded according to Brooker et al. .
The reconstructed femoral offset was measured as the distance from the centre of rotation of the femoral head to the long axis of the femoral shaft. The native femoral offset was measured on the disease-free contra-lateral hip in patients without contralateral hip replacement or on pre-operative radiographs.
In hips with ceramic-on-polyethylene articulation, linear wear was measured by one blinded observer on standard anteroposterior supine digital radiographs (5-MPix resolution) determining femoral head penetration in the single image technique using the Roman V 1.70 software (Institute of Orthopaedics, Oswestry, UK) as described by Geerdink et al. . Since the Fitmore cup has an eccentric biradial design with a hemispherical periphery, the technique had to be adapted by finding the circles that best fit the hemispherical periphery of the cup to accurately determine the cup centre and the prosthetic femoral head. For the interobserver repeatability, the mean difference between the two wear measurements was −0.01 mm/year (95% CI, −0.1 to 0.08), and for the intra-observer repeatability the mean difference was 0.02 mm (95% CI, −0.06 to 0.09).
Kaplan-Meier survivorship analyses were performed with the use of several end points: revision for aseptic loosening, revision for any reason, and a re-operation on the hip for any cause (e.g., loosening of either one or both of the components, osteolysis, liner exchange, infection, peri-prosthetic fracture, or dislocation) were used as the endpoints. Revision was defined as “an operation that involved removal and/or replacement of one or more components of a joint replacement”; re-operation means all forms of surgery after total hip arthroplasty. Acetabular revision was further subdivided into “major”, involving removal and/or replacement of the acetabular metal shell, and “minor”, which did not involve the removal and/or replacement of the shell, such as an exchange of liner. Differences in survival rates between different groups of patients were tested for statistical significance using the two-sided log-rank test. Continuous variables were compared with the use of a two-sided Student’s t- test. We considered p-values of <0.05 to be significant. SPSS® version 17.0 (SPSS Inc, Chicago, IL) and Graphpad Prism® version 5.0 (Graphpad Software, San Diego, CA) were used to record and analyse the collected data.
Five out of 89 total hip arthroplasties (6%) have been revised. Mean time to revision was 8.5 years (range, 5.8–9.9 years). In two hips (2%) an isolated revision of the modular acetabular liner was performed; in one hip, the metal-on-metal bearing was changed to a metal-on-polyethylene articulation for unexplained pain after 8.2 years. In another hip, the bearing was revised for recurrent late dislocation after 9.9 years. No revision was performed for accelerated wear. In two hips (2%), the femoral and the acetabular component were revised; in one hip, the stem and the shell were removed for deep infection 9.4 years after the initial operation. Another hip had a femoral revision for peri-prosthetic femoral fracture after 9.2 years; the acetabular component was revised at the time of stem revision although it was well fixed intra-operatively.
An isolated revision of the femoral component was performed in one hip (1%) for aseptic loosening after 5.8 years. In this case, the femoral component had been primarily undersized and subsided subsequently. There was no revision for aseptic loosening of the acetabular shell.
One hip with a ceramic-on-polyethylene bearing showed significant wear of about 50% of the liner; in this case liner exchange was recommended, but the patient refused the operative revision. We recommended annual reassessments to observe progression of wear and development of osteolysis. There were no other impending revisions or re-operations at the time of most recent follow-up.
One hip required surgical resection for symptomatic heterotopic ossification 7.5 years after initial operation. Another patient, who already had a liner revision for recurrent dislocation at 9.9 years (see “Revisions”), suffered a peri-prosthetic femoral fracture after trauma and was treated with open reduction and internal fixation at 11.7 years; all components were well fixed despite fracture and not revised in this case.
Intra-operative complications occurred in six hips (7%) and were all related to a proximal femoral fracture during insertion of the femoral component. Four fractures were stabilized with cable wires only, and in one hip a longitudinal split of the proximal femur was treated with cable wires and a stem revision (cemented stem); this hip was also included in the study, but radiological evaluation and survivorship analysis was performed only for the acetabular component.
Radiographical examination showed that no acetabular or femoral components were loose at a minimum of ten years, and all appeared radiologically to have solid osseous ingrowth. We could not detect peri-acetabular osteolysis radiographically in any hip.
Radiolucent lines of less than 2 mm were present around the acetabular component in two of 64 hips (3% of the hips with radiological review), involving DeLee-Charnley zone III in two hips and DeLee-Charnley zone I in one hip. We did not find peri-acetabular radiolucent lines greater than 2 mm in any hip.
The mean initial cup inclination angle was 45.7° (range, 35.6–61.3°). The mean inclination angle at the time of final review was 45.9° (range, 35.0–59.8°). No significant change was found in cup inclination (paired Student t-test, p = 0.64) and no acetabular component showed horizontal or vertical migration of more than 3 mm when comparing early and late postoperative radiographs. The mean cup anteversion angle was 12.7° (range, 0–34.6°). The mean linear wear rate in the subgroup of hips with ceramic-on-polyethylene articulation (n = 25) was 0.13 ± 0.06 mm/year (range, 0.03–0.24).
Radiolucent lines around the femoral component were observed in 12 of 64 hips (18% of the hips with radiological review) and were limited to the proximal femur (Gruen zones 1 [9 hips], 2 [1 hip], 6 [2 hips] and 7 [9 hips]). No hip had distal cortical hypertrophy, third or fourth-degree stress-shielding with atrophy of the proximal part of the femur, pedestal formation at the tip of the prosthesis with bridging of the complete femoral canal, or subsidence >5 mm comparing early and late postoperative radiographs. Heterotopic ossification was found in 17 of 64 hips (27%) and was graded according to Brooker as grade I (11 hips), II (4 hips) and III (2 hips).
The mean native hip offset was 36.6 mm (range, 20.4–59.1) and the mean reconstructed femoral offset was 34.4 mm (range, 18.5–53.3). There was a statistically significant difference between native and reconstructed femoral offset (paired Student t-test, p = 0.02).
The HHS and the Devane activity score improved from a mean of 45 points preoperatively to a mean of 91 points at last follow-up (paired Student t-test, p < 0.001) and from 2.6 to 3.2 points (paired Student t-test, p = 0.01), respectively. At the latest follow-up, one patient (one hip; 1% of the unrevised living cohort) had a poor result (HHS < 70 points). In this case, an intra-operative fracture of the greater trochanter was fixed with tension band wiring. At last follow-up, all components were well fixed, but the patient complained about pain in the hip (VAS 6/10; HHS 46 points). The mean HHS of the 12 patients who had no radiological follow-up was 95 (range, 75–100) points.
Since the introduction of total hip arthroplasty in the late 1960s, the management of younger patients has been challenging and outcomes in this group of patients are controversial. Uncemented cups were introduced in the 1980s to overcome the problems of high long-term failure rates due to osteolysis and aseptic loosening as seen with cemented acetabular reconstruction, especially in young patients, despite improved cementing techniques. They are widely used in THA all over the world. While the success of uncemented acetabular components in terms of fixation is well established, problems with polyethylene wear and osteolysis often compromise the long-term survival. In their analysis based on the Norwegian Arthroplasty Register, Hallan et al. showed high revision rates for polyethylene wear, osteolysis and aseptic loosening leading to a marked decline in survivorship after ten years, while most devices performed well up to seven years. The authors estimated the 12-year survival for metal-backed acetabular components with conventional polyethylene between 74% and 90% when acetabular revision for any reason was used as the endpoint. They concluded that none of the components used in their study had satisfactory long-term results . Similar results were documented in several non-register reports.
The problems of osteolysis and accelerated polyethylene wear become most evident in the young and active patient [8, 9]. In general, there is relatively little information on the survival of uncemented acetabular components in young patients in the second decade. Published reports with more than ten years of follow-up consistently show unacceptably high rates of peri-acetabular osteolysis and of re-operation for problems related to the acetabular side. In a study from the Finnish Arthroplasty Register, in patients under 55 years of age, only the Harris-Galante II cup showed a survival rate >80% at ten years with cup revision for any reason (including liner exchange) as the endpoint (87%); at 13 years survival declined to 78% . Utting et al. reviewed the results of 70 hemispherical Harris-Galante I cups in patients under the age of 50 years. At 14 years they found a cumulative survival of 84%, with acetabular revision for any reason as the endpoint; survival with impending revision as the endpoint was 55% at 14 years. Of the hips radiographically reviewed, 19% showed acetabular osteolysis . Duffy et al. reported the results of 84 Harris-Galante I cups in patients 50 years or younger. At ten years cumulative survival with acetabular revision for any reason was 84.5% and osteolysis was found in 20% of the unrevised hips . In a series by Della Valle et al., very acceptable results at a minimum follow-up time of 20 years were achieved with use of the Harris-Galante I acetabular component, but osteolysis was observed in more than 30% of patients who were younger than 50 years . Recently, Kim et al. reported the results of a contemporary uncemented THA using the Duraloc acetabular component in combination with the uncemented Profile stem and a 22-mm CoCr head articulating with conventional ram-extruded 415 GUR polyethylene in a consecutive series of 532 hips in patients younger than 50 years old. Cumulative survival of the acetabular component at 15 years was 82% and peri-acetabular osteolysis was found in 20% of all hips .
Different factors might explain the high survival and low prevalence of peri-acetabular osteolysis in our study. First, remarkably low rates of peri-acetabular osteolysis and revision for any reason have been reported using the second generation 28-mm metal-on-metal Metasul articulation , also in young patients , though Holloway et al. reported 14% peri-acetabular osteolysis in a small cohort of 22 hips . Problems with liner disassociation  and metal hypersensitivity/ALVAL  have been reported, whereas metallosis and pseudotumours seem not to be an issue with this 28-mm polyethylene-backed bearing surface. These complications have been reported with large-head metal-on-metal articulation . However, we have made a change in our clinical practice away from metal-on-metal to ceramic-on-HXLPE articulations, as the local and systemic long-term effects of metal-on-metal THA are still not fully understood .
Second, we found a comparatively low linear wear rate of 0.13 mm/year using a 28-mm alumina-ceramic on UHMWPE bearing in our patients with a mean age of 49 years. Our results compare well to those reported for 28-mm alumina-ceramic heads on UHMWPE in a retrieval study by Kim et al. (0.11 mm/y) when uncemented metal-backed acetabular components were used in a cohort with a mean age of 45 years . Although different wear rates were found in hip simulator studies, clinical and retrieval studies suggest that the penetration/wear rate is higher with metal-on-UHMWPE than ceramic-on-UHMWPE bearing . Considering this, we believe ceramic-on-UHMWPE/HXLPE articulations are a reasonable option for THA also in young and active patients.
Third, the acetabular component used in this study has a relatively thin metal back (2.5-mm thickness in all sizes) whereas most modular uncemented cups available have a thicker metal shell. Using the Trilogy shell (thickness of the metal back from 2.7 mm to 4.6 mm dependent on size) a marked decrease in peri-acetabular cancellous bone density up to 34% within one year after implantation was described . A thin metal back with lower relative rigidity might allow for a higher stress transfer to cancellous peri-acetabular bone and therefore might reduce stress shielding. It has been hypothesized that a compromised cancellous bone structure plays an important role in the development of osteolytic lesions . The flattened polar region optimizes press-fit and load transfer in the periphery of the cup, which might enhance osseointegration and sealing of the periphery of the acetabular component and limit the access of wear particles to the retro-acetabular bone. A relatively low retro-acetabular stress shielding and good peripheral osseointegration could contribute to a low rate of periacetabular osteolysis, although we have no data to support this contention.
The importance of the present report is that it represents the minimum ten-year results using contemporary uncemented implants, which are still in clinical use in a consecutive series of young and active patients with only a few patients lost to follow-up. No acetabular revision was performed for accelerated wear, osteolysis or aseptic loosening at the time of the last follow-up. We did not detect peri-acetabular osteolysis visible on plain radiographs in those hips evaluated radiographically, and all components appeared to have solid osseous ingrowth. Aseptic loosening was negligible as a failure mode in this series; we found only one aseptic stem loosening due to a primary undersized implant and no aseptic acetabular loosening in any hip.
There are important limitations of this study. First, the use of two different types of bearings makes the group heterogeneous. The study was not sufficiently powered to compare the performance of the two articulations used. Another limitation is the evaluation of osteolysis using radiographs, as the incidence of osteolysis might be underestimated on radiographs compared to evaluation by computed tomography or magnetic resonance imaging . In addition, there is a potential bias on the radiographic results as 12 hips could not be evaluated radiographically.
The survival rates and the radiological outcome after primary uncemented THA in our young and active patient group with a mean age of 49 years are encouraging into the early second decade. Our findings confirm the results of previous reports [36, 37] and compare well with those reported for the other press-fit acetabular components based on the thin cup design, like the Allofit press-fit cup  and the Morscher press-fit cup . Peri-acetabular osteolysis and liner wear was not a major problem in this cohort at 12 years, which is noticeable when compared to the results reported for other uncemented cups in this age group. Longer follow-up is necessary to monitor if development of peri-acetabular osteolysis is just delayed into the late second decade.
Conflict of interest
M.R.S. was supported by the non-commercial research fund of the “Deutsche Arthrose-Hilfe e.V.”. In addition, benefits have been directed to a research fund by Zimmer Inc.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.
- 8.Kim YH, Kim JS, Park JW, Joo JH (2011) Periacetabular osteolysis is the problem in contemporary total hip arthroplasty in young patients. J Arthroplasty. doi: 10.1016/j.arth.2011.03.022
- 9.Della Valle CJ, Mesko NW, Quigley L, Rosenberg AG, Jacobs JJ, Galante JO (2009) Primary total hip arthroplasty with a porous-coated acetabular component. A concise follow-up, at a minimum of twenty years, of previous reports. J Bone Joint Surg Am 91(5):1130–1135Google Scholar
- 23.Geerdink CH, Grimm B, Vencken W, Heyligers IC, Tonino AJ (2008) The determination of linear and angular penetration of the femoral head into the acetabular component as an assessment of wear in total hip replacement: a comparison of four computer-assisted methods. J Bone Joint Surg Br 90(7):839–846PubMedCrossRefGoogle Scholar
- 28.Malik A, Dorr LD, Long WT (2009) Impingement as a mechanism of dissociation of a metasul metal-on-metal liner. J Arthroplasty 24(2):323.e13–16Google Scholar