Feasibility of one-eighth time gated myocardial perfusion SPECT functional imaging using IQ-SPECT
- 495 Downloads
IQ-SPECT, an add-on to general purpose cameras based on multifocal collimation, can reduce myocardial perfusion imaging (MPI) acquisition times to one-fourth that of standard procedures (to 12 s/view). In a phantom study, a reduction of the acquisition time to one-eighth of the standard time (to 6 s/view) was demonstrated as feasible. It remains unclear whether such a reduction could be extended to clinical practice.
Fifty patients with suspected or diagnosed CAD underwent a 2-day stress–rest 99mTc-sestamibi MPI protocol. Two consecutive SPECT acquisitions (6 and 12 s/view) were performed. Electrocardiogram-gated images were reconstructed with and without attenuation correction (AC). Polar maps were generated and visually scored by two blinded observers for image quality and perfusion in 17 segments. Global and regional summed stress score (SSS), summed rest score (SRS) and summed difference score (SDS) were determined. Left ventricular volumes and ejection fraction were calculated based on automated contour detection.
Image quality was scored higher with the 12 s/view acquisition, both with and without AC. Summed scores were statistically comparable between the 6 s/view and the 12 s/view acquisition, both globally and in individual coronary territories (e.g. in images with AC, SSS were 6.6 ± 8.3 and 6.2 ± 8.2 with 6 s and 12 s/view, respectively, p = 0.10; SRS were 3.9 ± 5.6 and 3.5 ± 5.3, respectively, p = 0.19; and SDS were 2.8 ± 5.7 and 2.6 ± 5.7, respectively, p = 0.59). Both acquisitions allowed MPI-based diagnosis of CAD in 25 of the 50 patients (with AC). Calculated end-diastolic volume (EDV) and end-systolic volume (ESV) were modestly higher with the 6 s/view acquisition than with the 12 s/view acquisition (EDV +4.8 ml at rest and +3.7 ml after stress, p = 0.003; ESV +4.1 ml at rest and +2.6 ml after stress, p = 0.01), whereas the ejection fraction did not differ (−1.2 % at rest, p = 0.20, and −0.9 % after stress, p = 0.27).
Image quality and LV functional parameters obtained with a one-eighth acquisition time were statistically comparable to the previously validated one-fourth time protocol using IQ-SPECT. Shorter acquisition times without loss of diagnostic accuracy provide improved patient comfort and streamlined departmental efficiency.
KeywordsCoronary artery disease 99mTc-sestamibi SPECT/CT IQ SPECT Low-count acquisition Myocardial perfusion imaging
Compliance with ethical standards
Dr. Caobelli is supported by a fellowship grant from Mallinckrodt Pharma.
Dr. Bengel receives research grants from Mallinckrodt Pharma and Siemens, and speaker honoraria from Bayer, Siemens, GE Healthcare and Mallinckrodt.
Conflicts of interest
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the principles of the 1964 Declaration of Helsinki and its later amendments or comparable ethical standards.
For this type of study, formal consent is not required.
- 5.Pirich C, Keinrath P, Barth G, Rendl G, Rettenbacher L, Rodrigues M. Diagnostic accuracy and functional parameters of myocardial perfusion scintigraphy using accelerated cardiac acquisition with IQ SPECT technique in comparison to conventional imaging. Q J Nucl Med Mol Imaging. 2014. In press.Google Scholar
- 10.Rajaram R, Bhattacharya M, Ding X, Malmin R, Rempel TD, Vija AH, Zeintl J. Tomographic performance characteristics of the IQ.SPECT system. Nuclear Science Symposium and Medical Imaging Conference (NSS/MIC), 2011 IEEE. p. 2451–56. doi: 10.1109/NSSMIC.2011.6152666
- 16.Vija H, Malmin R, Yahil A, Zeintl J, Bhattacharya M, Rempel TD, et al. A method for improving the efficiency of myocardial perfusion imaging using conventional SPECT and SPECT/CT imaging systems. Nuclear Science Symposium Conference (NSS/MIC), 2010 IEEE. p. 3433–37. doi: 10.1109/NSSMIC.2010.5874444
- 17.de Jong AN, Akkermans JBM, Van Asperen EA, Snoijs CHP. Left ventricular ejection fraction in gated myocardial perfusion, a comparison of SPECT and IQ-SPECT (OP847). Eur J Nucl Med Mol Imaging. 2014;41 Suppl 2:S151–705.Google Scholar
- 18.Kaleva E, Hippeläinen E, Sippo-Tujunen I, Sipilä O, Pitkonen M. Dynamic heart phantom ejection fractions with IQ-SPECT system vs. LEHR collimators (OP674). Eur J Nucl Med Mol Imaging. 2014;41 Suppl 2:S151–705.Google Scholar
- 19.Snoijs CHP, Akkermans JBM, de Jong AN. A phantom study comparing left ventricular ejection fraction values from myocardial perfusion LEHR-SPECT and IQ-SPECT (OP217). Eur J Nucl Med Mol Imaging. 2014;41 Suppl 2:S151–705.Google Scholar
- 21.Takahashi T, Tanaka H, Kozono N, Tanakamaru Y, Idei N, Ohashi N, et al. Characteristics of images of angiographically proven normal coronary arteries acquired by adenosine-stress thallium-201 myocardial perfusion SPECT/CT-IQ SPECT with CT attenuation correction changed stepwise. Ann Nucl Med. 2015;29:256–67CrossRefPubMedGoogle Scholar