A fast cardiac gamma camera with dynamic SPECT capabilities: design, system validation and future potential
- 3.2k Downloads
The goal of this study is to present the Discovery NM 530c (DNM), a cardiac SPECT camera, interfacing multi-pinhole collimators with solid-state modules, aiming at slashing acquisition time without jeopardizing quality. DNM resembles PET since it enables 3-D SPECT without detector motion. We further envision how these novel capabilities may help with current and future challenges of cardiac imaging.
DNM sensitivity, spatial resolution (SR) and energy resolution (ER), count rate response, cardiac uniformity and cardiac defect contrast were measured and compared to a dedicated cardiac, dual-head standard SPECT (S-SPECT) camera.
DNM sensitivity was more than threefold higher while SR was notably better. Significantly, SR was the same for 99mTc and 201Tl. ER was improved on DNM and allowed good separation of 99mTc and 123I spectral peaks. Count rate remained linear on DNM up to 612 kcps, while S-SPECT showed severe dead time limitations. Phantom studies revealed comparable uniformity and defect contrast, notwithstanding significantly shorter acquisition time for the DNM. First patient images, including dynamic SPECT, are also presented.
DNM is raising the bar for expedition and upgrade of practice. It features high sensitivity as well as improved SR, temporal resolution and ER. It enables reduction of acquisition time and fast protocols. Importantly, it is potentially capable of dynamic 3-D acquisition. The new technology is potentially upgradeable and may become a milestone in the evolution of nuclear cardiology as it assumes its key role in molecular imaging of the heart.
KeywordsCardiac gamma camera Single photon emission computed tomography Myocardial perfusion scintigraphy Discovery NM 530c
The technique of SPECT for myocardial perfusion scintigraphy (MPS) is the most common, non-invasive imaging tool for risk stratification of patients with coronary artery disease (CAD) [1, 2, 3, 4, 5]. Millions of studies are performed each year, greatly exceeding other competing modalities , such as PET or MRI, in spite of some advantages offered by them. The growth of SPECT results from the vast extent of CAD worldwide and its associated morbidity and mortality, and enormous social and economic burden. This situation has created an urgent need for an efficient, non-invasive tool to risk-stratify patients and thus to direct and monitor therapy. The need has perpetuated a continuous effort to upgrade SPECT MPS [7, 8, 9, 10].
Notwithstanding the growth and value of current standard SPECT (S-SPECT) practice, it is limited and needs to be transformed in order to remain a truly effective tool. Of foremost concern are the costs of the technique. Expenditures for health care in the USA have doubled over the last 10 years [11, 12] and cardiovascular disease accounts for 20% of these total costs. The growth rates of imaging costs continue to outpace other medical services  and stress imaging has grown 6% per year from 1993 to 2001 . While, in theory, cutting procedure time could have reduced costs, attempts to slash acquisition time are limited with current SPECT technology. This is due mainly to the low efficiency of the classic Anger gamma camera and its parallel-hole collimation, and additionally to the limited spatial resolution (SR) and energy resolution (ER) of the sodium iodide crystal and the process of light scintillation. Measurements of gamma photon position and energy are derived from the many signals produced by a large array of photomultiplier tubes. The individual signals are statistically poor, and this limits both the SR and ER. Limited ER leads to contamination of the image by scattered gamma photons, leading to further deterioration of SR and target to non-target contrast ratios. Marginal ER often leads to instability of uniformity corrections and degradation of the detector’s uniformity of response. The collection of resulting errors may be propagated during attenuation-corrected SPECT reconstruction .
A more fundamental limitation of conventional MPS is the relative nature of its normalized output images. This occurs because the interpretation model relies, sometimes erroneously, on having at least one myocardial region with normal perfusion. In CAD that involves all three major arteries; the extent and severity of ischaemia will be reduced because of the image normalization, and often only the most severely underperfused areas will be detected [16, 17, 18, 19]. Sometimes, in “balanced ischaemia” and in left main coronary stenosis, MPS may totally miss patients who have a very high-risk coronary disease. Since the extent of perfusion abnormality is critical in deciding which patients would benefit from coronary revascularization, such underestimation of the extent of disease is a major limitation of S-SPECT that needs to be overcome [20, 21, 22]. Several innovative attempts to measure myocardial blood flow (MBF) and coronary flow reserve (CFR) with S-SPECT technology have been carried out, but have not gained wide clinical acceptance [23, 24, 25, 26, 27]. This situation reflects both the recognition of the need for such measurements as well as the technical difficulties and limitations involved. Since MBF and CFR can be measured by PET [28, 29], a large body of respected researchers has been predicting the emergence of PET as the primary MPS tool for a considerable time [30, 31, 32, 33, 34]. Interestingly, such adoption of PET for MPS has not yet occurred despite the abundance of PET/CT scanners in recent years. The expressed customer preference may be related to the ease of use and availability of both SPECT radiopharmaceuticals and equipment.
In DNM, the detectors and pinholes are all focused on the heart organ in order to maximize the imaging sensitivity there. The miniature pixels and miniature size of the CZT modules allows the packing and arrangement of multiple detectors in DNM with sufficient proximity and coverage to ensure that 3-D sampling of the heart provided by stationary acquisition is sufficient for tomographic reconstruction. Unlike traditional pinhole collimation that uses magnification, in DNM imaging, minification is used (Fig. 2b). In this case system resolution is preserved because of the small pixels and high intrinsic resolution the CZT technology [40, 41]. This matching allows a reduction in detector size without loss of resolution. The savings in detector area allows the economical construction of multiple detectors, each with a full view of the heart. Since each detector is stationary, it has the full acquisition time to acquire the image. Even a few minutes of acquisition time is more than enough to outperform the sensitivity of a rotating S-SPECT camera, where typically only a few seconds are allocated to each projection. Thus, DNM is an inherently tomographic technology, which similar to PET is potentially capable of truly dynamic tomographic study that is required for quantitative MPS.
The aim of this manuscript is to describe the technological concepts underlying the DNM design, to measure and compare DNM features to state-of-the-art S-SPECT technology, and to forecast how those improved capabilities will promote new significant changes in clinical practice and may influence the evolution of nuclear cardiology.
Materials and methods
The DNM gantry is similar to that of an S-SPECT gamma camera, yet its detector assembly is radically new. The detector assembly comprises a multi-pinhole collimator block, where each pinhole illuminates a solid-state pixelated, gamma ray detector made of CZT (Fig. 2). Thus, each pinhole and detector constitutes a complete miniature gamma camera.
In CZT, an absorbed gamma ray is converted directly to an electric charge. In S-SPECT with scintillation crystals, an absorbed gamma ray is first converted to ultraviolet light and only then to an electric charge in an “indirect” process. The indirect process is inefficient and is then followed by a high amplification step in the photomultiplier tube, resulting in limitation to the overall accuracy of the measured energy and position of the incident gamma rays. In CZT these limits are overcome by a “direct” signal of electron hole pairs that is ∼50 times bigger than the signal being amplified in the S-SPECT system. Thus, the ER is dramatically improved by the statistics of the collected signal. A known limitation of the ER in CZT results from depth of interaction effects in the detector material; however, this effect is measurably reduced in pinhole collimation geometry because of the oblique angles of incidence which reduce both the mean depth of interaction and the variability of these depths .
The intrinsic SR of the CZT technology is determined by the combination of the pixel size where the gamma photons are absorbed and the system geometry. The CZT pixels are physically 2.46 × 2.46 mm in size and have negligible crosstalk due to the direct conversion to an electric charge. In contrast conventional S-SPECT pixels do share signals because of the spreading of light between them and this crosstalk degrades their effective pixel size. The optimized small pixel size of DNM allows a significant minification without loss of resolution compared to conventional systems. Overall, the DNM system detector contains 19 detector heads of 8 × 8 cm. The detector heads are directed towards the heart so as to sample signals from the heart and the region around it. The target volume for high quality imaging in which the patient heart is positioned is approximately a sphere of 19 cm diameter. Once the patient is positioned on the imaging table, the position of the gantry is adjusted so that the heart is close to the centre of the field of view (FOV). An automatic algorithm can guide the operator to the best positioning.
Importantly, the intrinsic SR is independent of photon energy. In particular, for isotopes with low energy photons like 201Tl where intrinsic SR is a limiting factor for S-SPECT, the SR may be greatly improved and become similar to 99mTc. Another source for SR degradation in S-SPECT is septal penetration and scatter in the parallel-hole collimator by high energy photons emitted by 123I and 201Tl radiopharmaceuticals or from isotopic impurities. The well-shielded pinhole collimator may be free of these processes and thus may provide equivalent resolution to that of the 99mTc. Each CZT pixel is an independent detector provided with its own electronic processing unit using miniaturized attached electronics (Fig. 2d). Therefore the DNM may be capable of linear detection response even at very high count rates. This is not possible with S-SPECT where only two independent detector electronic chains exist, one for each head of the camera, and therefore detector processing dead time results in significant event losses and pile-ups, at high count rates. Since the DNM gantry does not move during acquisition and since all detectors are acquiring data from the same source simultaneously, the measured projections are temporally consistent and the reconstructed data may be termed “inherent and instantaneously tomographic”. In this respect DNM imaging resembles PET imaging that features temporal consistency of data acquired into projection views that may be beneficial for applications requiring dynamic information. The DNM is thus designed to acquire all types of cardiac SPECT procedures including multigated tomography and dynamic images. In addition to “on-the-fly” reconstruction the DNM saves each event of the acquired data in a list-mode format including its energy, spatial information, time of occurrence and gated ECG trigger, for later specialized analysis.
For DNM, dedicated image reconstruction was developed. SPECT reconstruction is based on an implementation of a 3-D iterative Bayesian reconstruction algorithm  combining accurate modelling of DNM geometrical and physical characteristics with optimized statistical techniques. An adaptation of one-step-late Green’s prior  was used to control smoothness of the reconstruction; 30–50 iterations of the algorithm were used for reconstruction of phantom and clinical data. Post-reconstruction, a Butterworth filter with a cut-off frequency of 0.37 cycles/cm and power of 7 was applied. Fine-tuning of the reconstruction algorithm parameters, optimized separately per type of acquisition (e.g. low-dose, high-dose), was based on the optimal trade-off between reconstructed uniformity of healthy myocardium and contrast of perfusion defects, using extensive phantom studies .
In the following experiments, whenever it was relevant, we have compared imaging capabilities of the DNM system to a state-of-the-art, cardiac-dedicated, dual-head S-SPECT camera (Ventri, GE Healthcare).
The count sensitivity of the two cameras (DNM and S-SPECT) for 99mTc with energy window set at 140.5 keV ± 10% was compared using the cardiac insert of an anthropomorphic torso phantom (Data Spectrum), in air without any scatter material. The patient volume that is fully seen by all detectors is denoted the quality FOV (QFOV) For the DNM two different phantom locations were tested: (1) at the centre of the QFOV and (2) the point closest to the camera body within the QFOV. The same phantom was used for sequential sensitivity measurements on both DNM and S-SPECT systems. The measurements were corrected for decay. Sensitivity was calculated as counts/s per MBq.
Spectra of 99mTc, 201Tl and 123I were measured using a method similar to the “intrinsic ER” test defined in NEMA NU-1 2001 standard section 2.2, on both DNM and S-SPECT detectors without collimation, using a point source. ER was calculated by measuring full-width at half-maximum (FWHM) as percentage of peak position. ER is important for scatter rejection from image data as well as intrinsic uniformity and calibration stability. In addition, energy spectra of combined 99mTc and 123I were measured to evaluate simultaneous 99mTc/123I dual isotope acquisition capabilities of both systems.
Collimator penetration in 123I imaging
A 123I point source of 88.8 MBq activity located at a distance of 15 cm from the collimator was imaged in planar mode on an S-SPECT system and in one of the DNM collimated detectors to visualize respective collimator penetration.
SPECT resolution (with scatter)
SPECT resolution with scatter was measured using three 99mTc point sources (1.5 × 1.5 × 1.5 mm) placed according to NEMA in a cylindrical phantom filled with water. The phantom centre was placed 15 cm away from the external surface of the collimator. The two-dimensional (X and Y) activity profiles were plotted and FWHM calculated for each of the three point sources; the results of the central, radial and tangential resolutions as defined by NEMA protocol are reported. The following reconstruction parameters were used for DNM: reconstruction type: maximum likelihood expectation maximization (MLEM), number of iterations: 250, reconstruction voxel size: 2 × 2 × 2 mm, and for S-SPECT: reconstruction type: filtered backprojection (FBP), reconstruction voxel size: 1.6 × 1.6 × 1.6 mm. No filters were used for both modalities. Additional similar measurements were performed for 201Tl point sources. Note that while the S-SPECT result is fully compliant with NEMA protocols, FBP is not possible with the DNM and therefore the result is not compliant with the NEMA protocol.
Count rate performance
The count rate performances of the DNM and of the S-SPECT were measured to assess the count rate linearity, especially at high rates. Good count rate performance is crucial for quantitative dynamic SPECT applications, such as CFR measurements performed on data acquired during the passage of an injected bolus through the heart chambers. Since CFR and left ventricular ejection fraction (LVEF) measurements both require very high temporal resolution, of the order of 1 s or even faster, very high count rates are mandatory for a good image. However, high count rates often cause a non-linear camera response that significantly reduces accuracy. Therefore both high camera sensitivity and count rate linearity are required.
For S-SPECT, measurements were based on NEMA NU-1 guidelines and were performed accordingly: without collimation and using a decaying 99mTc source. Since multi-pinhole collimation is an integral part of the DNM system, the measurements were performed with collimation. The DNM measurements were performed using the multiple calibrated source method: 12 syringes with increasing activity from 199.8 to 1,994.3 MBq (53.9 mCi) of 99mTc were sequentially placed at the centre of the QFOV. The counts measurement was performed on S-SPECT using an acquisition mode that was dedicated for high count rates. The DNM has only one standard acquisition mode for all count rates. The energy window was the same for both systems, i.e. 140.5 keV ± 10%. The DNM was configured to stop the acquisition when the count rate exceeds a predetermined electronic throughput computer limit.
Phantom studies of 99mTc myocardial perfusion SPECT
Series of anthropomorphic torso phantom (Data Spectrum) acquisitions were performed in order to evaluate image quality in multiple imaging scenarios relevant to clinical practice. The main clinical task for MPS is detection and characterization of perfusion defects; thus useful images should achieve uniform uptake within normally perfused myocardium and delineate myocardial perfusion defects in pathological situations. To capture this, we have compared S-SPECT and DNM performance by measuring the trade-off between uniformity of normal myocardium and contrast of perfusion defects for manufacturer-recommended acquisition parameters with a standard camera and short acquisition times intended for clinical use of DNM.
Phantom preparation and acquisition
A total of 370 MBq of 99mTc-MIBI was injected in a resting patient’s left arm while lying on the camera table. A list-mode dynamic acquisition was started simultaneously with the injection and continued for 5 min, allowing offline division of the data into various time intervals. The first 90 s were segmented to intervals of 3 s per step. Attenuation correction was performed using a map obtained from a dedicated CT imaging device. The data of each time interval were reconstructed separately resulting in 30 attenuation-corrected volumes each consisting of a set of transaxial slices. In addition, a volume containing data from the entire 5-min acquisition was generated and a region of interest (ROI) was drawn inside the left ventricular cavity. This ROI was copied to the other 30 volumes and the counts within it were measured and a left ventricular cavity time-activity curve was generated.
Comparison of DNM and S-SPECT sensitivity. S-SPECT sensitivity, being independent of location, is measured only once
Scan phantom position
Sensitivity (counts/s per MBq)
Sensitivity (DNM)/sensitivity (S-SPECT)
DNM QFOV centre
DNM max. rate in QFOV
ER calculated on the energy spectra curve as percentage of FWHM from peak position energy and its corresponding keV value. ER is better on DNM for all three isotopes
ER DNM, % (keV)
ER S-SPECT, % (keV)
Simultaneous dual isotope 99mTc/123I energy spectrum
SPECT resolution (with scatter)
Central, tangential and radial resolution calculated according to NEMA NU-1 for both DNM and S-SPECT for point sources filled with 99mTc or 201Tl
Count rate performance
Phantom studies of 99mTc myocardial perfusion SPECT
Results of uniformity and contrast measurements for phantom experiments simulating low-dose rest part of MPS acquisition, including lateral and septal wall defect inserts. Each measurement represents a series of ten phantom acquisitions performed sequentially on DNM and S-SPECT (Ventri). Overall, notwithstanding the significantly shorter acquisition time of DNM, uniformity and defect contrast trade-off are comparable to S-SPECT
Male phantom configuration mean (SD)
Female phantom configuration mean (SD)
Lat EC, %
Sep EC, %
Improved image characteristics
The enhanced ER demonstrated in DNM improves scatter rejection and thus contributes to image contrast. It can also facilitate the separate imaging of simultaneously acquired multiple different isotopes. In particular, the improved separation between the energy peaks of 99mTc and 123I and the lack of 123I septal penetration for the pinhole collimators (Figs. 5 and 6) are expected to lead to a variety of novel 99mTc/123I dual isotope protocols. Another important feature of pixelated CZT imaging, which is advantageous for dual isotope protocols, is the common SR for all energies and isotopes. Thus, the SR for 201Tl is the same as 99mTc (Table 3 and Fig. 7). Lower 201Tl resolution in S-SPECT has been a source of decreased contrast on 201Tl images and also a source of mismatch in the comparison of stress/rest left ventricular sizes in current 99mTc/201Tl dual isotope protocols.
The DNM technology has other practical advantages; since each detector pixel has independent electronic signal readout and processing, it is therefore capable of accurate count measurements even at very high count rates, where S-SPECT cameras with their single large detector crystals fail (Fig. 8). We denoted this as “high rate count linearity”. Critically, since the gantry of DNM does not move during acquisition and since all detectors are therefore acquiring the data from the source simultaneously, the projection images recorded are therefore all consistent in the time domain. That is, patient motions and isotope distributions are the same for each and every projection image submitted to the reconstruction module. This aspect of the data resembles PET imaging and its advantages; we denote it as “inherently tomographic imaging”. Importantly, data acquired in a short time interval is also reconstructable, allowing even further reduction of the acquisition time when necessary, e.g. in emergency scenarios. Since images can be reconstructed on the fly, acquisition can be terminated at a preset count density or when the data becomes sufficient for diagnosis. Because of its inherent tomographic features, the DNM is capable of dynamic tomographic sampling with high temporal resolution of the left ventricular cavity and the myocardium during the initial passage of an injected bolus through the heart chambers (Fig. 11). These data are the classic input for compartmental modelling aimed at absolute CFR measurements.
Besides reducing acquisition time, the increased sensitivity of the DNM may alternatively enable a cost-saving reduction in radiopharmaceutical dosage and a corresponding reduction in patient radiation exposure, all options that contribute to flexibility and cost-effectiveness in the nuclear cardiology clinic. In addition, faster patient procedures allow more economical usage of the total dose that is prepared in advance, especially in the setup of a 1-day rest/stress protocol using 99mTc-based MPS tracers. This logistic advantage is very relevant today, due to the shortage of 99mTc generators in northern America . The short acquisition times and the small footprint may enable use near the coronary care unit itself, employing long-standing protocols [48, 49] for MPS in scenarios of acute coronary syndrome. Alternatively, scientists and clinicians concerned with the best image quality may utilize the DNM sensitivity to acquire higher statistics still in a relatively short time. For example, more precise gated acquisition with more and hence shorter gate intervals may allow the algorithmic creation of de-blurred perfusion images free of both respiratory motion and cardiac motion artefacts [50, 51]. Such de-blurring is an essential preliminary step before any hardware improvement of SR can become effective. In addition, an increased number of gates may be used for new applications that are related to myocardial electrical conduction and contractility that are currently being developed . Patient motion itself and its consequential artefacts may be reduced because of increased patient comfort and cooperation during the short acquisitions. Furthermore the inherent tomographic imaging may potentially enable 3-D patient motion correction including rotation, while S-SPECT methods can usually directly account only for translations. In addition, S-SPECT artefacts resulting from temporal inconsistency, such as the “upward creep” artefact, which is created when the heart gradually moves up during the S-SPECT camera rotation [53, 54], may be less of a problem with the temporally consistent DNM projections.
Potential limitations of the DNM should also be considered; a major theoretical concern is incomplete sampling of data. This may occur since the DNM collimation is concentrated on the heart vicinity and may unevenly sample extracardiac activity. Our phantom studies that included various degrees of liver activity, as well as a recent multi-centre clinical trial, did not reveal explicit problems, so far. However, this potential theoretical limitation should be addressed by special studies designed to evaluate the actual effect, and especially for optimizing the accuracy of quantitative dynamic measurements. Another limitation of our study relates to the difficulty in comparing such radically different technologies, especially when standardized quality control procedures, like NEMA, do not exist for DNM. In particular, the SR measurements (Table 3) had to be done using different reconstruction algorithms, FBP for S-SPECT and MLEM for DNM. Since measurements were performed on point sources, they may not ideally reflect the actual resolution that would be achieved in real-world clinical scenarios.
Expedited new protocols
In addition to shorter acquisition times, the DNM may allow fast MPS protocols with back-to-back stress/rest gated SPECT acquisitions, significantly shorter than 1 h, while occupying the camera for no more than 15 min total. There are at least three different ways. The first possibility is using stress/rest subtraction protocols. Such protocols shall begin with a stress acquisition. If stress data are abnormal, another dose is injected while the patient is resting within the camera, followed by another acquisition 3 min later and then image subtraction. Better subtraction is expected with DNM short acquisition because of the reduction in patient motion and myocardial shift. The 3-min wait for the second acquisition is similar to the time needed for patient positioning for a second separate acquisition in S-SPECT. Moreover, this time may be used for dynamic gated acquisition of the injected bolus.
A second fast protocol that may be facilitated by DNM is simultaneous dual isotope imaging, for example 99mTc and 201Tl following stress and rest injection, respectively. Such protocols have been evolving for S-SPECT [59, 60, 61] and may now reach maturity, because of the available DNM detector performance.
A third fast MPS protocol is the reconsideration of tracers that are rapidly washed from the myocardium because of the high sensitivity and simultaneous views data. A very interesting tracer is 99mTc-boronic acid teboroxime (BATO), an FDA-approved tracer that is close to Saperstein’s  definition of an ideal myocardial perfusion tracer. BATO’s extraction fraction is higher than those of 201Tl or 99mTc tracers and its myocardial uptake remains a near-linear function of flow even at high flow rates [63, 64]. Its initial myocardial uptake as well as its segmental washout rate correlate well with coronary blood flow, providing two different targets for quantitative measurements [23, 65]. Moreover, the high myocardial extraction and rapid biological clearance allow for very short stress/rest protocols . The differential washout of BATO may produce filling of stress perfusion defect within a few minutes of early delayed images in a high percentage of patients, potentially eliminating the need for a second rest injection . In spite of BATO’s unique qualities, it never gained wide usage. Acquisition and tomographic reconstruction of rapidly changing myocardial distribution with rotating S-SPECT technology is prone to inconsistency errors. Another issue was high concentration in the liver . These barriers led to the disappearance of BATO from mainstream nuclear cardiology practice . A recent simulation study  showed that a stationary SPECT camera might suit dynamic BATO studies better than a rotational S-SPECT camera. Indeed, with DNM, using adenosine or regadenoson  stress, with BATO injected when the patient is inside the camera, a high count rate stress/redistribution study may be completed in less than 10 min. If both perfusion and function are normal or if perfusion is abnormal but redistribution is noted in all defects, a separate rest BATO injection may be omitted. Otherwise, 15 min later and onward, a separate rest injection followed by a 5-min acquisition may be performed, completing a back-to-back stress/rest study in just about 30 min and occupying the camera for about 15 min.
Hybrid non-invasive CAD diagnosis: “the best of both worlds”
Currently, accurate structural imaging of CAD can be obtained non-invasively with multidetector computed X-ray tomography (MDCT) . However, when the ability of MDCT to detect ischaemia was tested against MPS, several studies [31, 73, 74] have found that while MDCT is consistently associated with outstanding negative predictive values (usually >90%) it provides only mediocre positive predictive values (29–44%). The ultimate value of a CAD imaging modality does depend on its ability to optimally assign the best treatment strategy to patients, especially differentiating between medical therapy and invasive revascularization . The ability of MDCT to meet this challenge is problematic, given that even invasive diagnostic angiography, the gold standard of coronary anatomy, is suboptimal for the task [76, 77]. Since DNM is also a small footprint camera and its short acquisition time resonates well with MDCT acquisition time, their optional usage in concert may help to efficiently arrive at a comprehensive diagnosis. This hybrid design concept (Fig. 1) is a continuation of “the best of both worlds” idea that underlined the creation of the Hawkeye—the archetypical, first clinical SPECT/PET/CT . In the context of CAD, such hybridization may evolve into a new type of diagnostic cardiac lab, where the decision to revascularize is made based on non-invasive procedures that are physically, temporally and conceptually separated from the invasive action.
Dynamic SPECT and the comeback of “first pass”
DNM high sensitivity combined with its dynamic SPECT capabilities may cause renewed interest in radionuclide angiocardiography, traditionally known as “first pass” (FP) , however, in a new tomographic form. The use of FP, one of the first breakthroughs in the history of nuclear medicine, has almost vanished in spite of its many advantages. FP acquisition time is less than 1 min, while it provides information on the volume and motion of each heart chamber and a measurement of its ejection fraction (EF). In particular, it gives more accurate information on the right ventricle than available with equilibrium gated angiocardiography (MUGA scan) . FP can be repeated immediately, using tracers that rapidly disappear from the blood pool or ultra-short-lived radionuclides such as 191mIr , for imaging multiple interventions. FP has been shown to provide important complementary information to SPECT MPS . Nevertheless, its usage in the planar mode had declined for a variety of reasons. In order to reduce the errors that resulted from planar background subtraction, each cardiac chamber had its own imaging view and only a few seconds of optimal time window for acquisition. Therefore the information was based on a few cardiac cycles only. A “good bolus” was required for reliable data and was often non-obtainable in patients with damaged veins due to chemotherapy, and in patients with reduced cardiac function. Other serious problems were caused by the count rate limitations of the Anger camera. On the one hand, the count rate from a large dose could exceed the linear range of the counting electronics, yet, on the other hand, basic gated frames were typically acquired in approximately 0.1 s and had very few signal counts which caused errors in the detection of the cardiac wall motion. As described earlier the DNM camera has the technological assets to overcome those technical limitations. In addition, since DNM is tomographic, background should not be a major problem and acquisition can be continued for unlimited cardiac cycles and a “good bolus” may not be essential.
Evolutionary perspective: the potential for increased resolution
The future of DNM may be the most intriguing question since the design can be optimized in a variety of directions. Since nuclear cardiology is concerned today mainly with imaging of relatively large “cold spots”, of the order of 1 cm or more, the current optimization achieves significant reduction in acquisition time with conservative improvements in SR (Tables 1 and 3). In addition, currently, any major improvement of SR would be masked by cardiac and respiratory motions. However, the imaging entitlement of the new technology could equally be used to achieve very high SR if needed and from an evolutionary perspective, resolution improvement should be given high priority in the years ahead. In fact, DNM core technology was developed in the microSPECT environment that has recently achieved a resolution of 400 μm, while maintaining sufficient sensitivity . Thus, the nuclear cardiology community may witness in the near future an evolving line of cameras with continuously improving SR. The prospect of using gated MPS and/or MDCT data for improved resolution recovery or for SPECT motion tracking and de-blurring seems to be a critical next step [50, 51]. Beyond that obligatory step, an increase in resolution to allow differentiation between subendocardial—and subepicardial—myocardium may benefit classic MPS. An increase in ultimate resolution would also enable a more accurate CFR measurement by allowing a reduction of spillover effects from the blood pool into the myocardium and vice versa, as both effects are contaminating the respective time-activity curves.
We also envision that in the near future increased resolution will be a key factor as nuclear cardiology matures within molecular imaging. While the importance of PET imaging will continue, unlike PET, SPECT is capable of simultaneous multiple isotope imaging and additionally, in theory, a better SR may ultimately be achieved with SPECT. Since identifying individual patients with vulnerable plaques at high risk for rupture is a central challenge, various studies including first clinical trials have been using different tracers for ”hot spot” plaque imaging studies . Other research targets include imaging of left ventricular remodelling, angiogenesis, apoptosis and hypoxia, gene expression, and stem cell therapy, to delineate a few. Thus we can predict a steady evolution of nuclear cardiology beyond the assessment of myocardial perfusion, towards the characterization of molecular events. Since DNM technology and recent innovative camera designs  have great potential for further improvement, they may turn out to be the first evolutionary step in linking molecular biology science and clinical cardiology.
The DNM, an ultra-fast cardiac SPECT camera, was created to answer the call of current evolutionary challenges in nuclear cardiology. It features high sensitivity as well as improved SR, temporal resolution and ER. It enables reduction of acquisition time and fast protocols. Importantly, it is inherently tomographic imaging and features high count rate linearity and therefore is potentially capable of dynamic 3-D acquisition. The DNM is designed to optionally work with MDCT, together comprising a system of comprehensive, non-invasive, functional and anatomical imaging for the management of CAD.
This article is dedicated to the memory of Bruce Hasegawa; many of the ideas contained herein were inspired by his near and far presence.
Conflicts of interest
The first author (MB) was consulting GE Healthcare during the DNM project. All the other authors are employees of GE.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.
- 2.Hachamovitch R, Berman DS, Kiat H, Cohen I, Friedman JD, Shaw LJ. Value of stress myocardial perfusion single photon emission computed tomography in patients with normal resting electrocardiograms: an evaluation of incremental prognostic value and cost-effectiveness. Circulation 2002;105:823–9.CrossRefPubMedGoogle Scholar
- 6.US Imaging Market Guide, Arlington Medical Resources, Malvern (PA) (2007).Google Scholar
- 11.Pear R. Health spending exceeded record $2 trillion in 2006 New York Times. Health. 2008 Jan 8Google Scholar
- 13.Report to the Congress: Medicare payment policy, Medicare Payment Advisory Commission, Washington, DC 2007.Google Scholar
- 15.Case J, Cullom S, Bateman T. Myocardial perfusion single-photon emission computed tomography attenuation correction. In: Iskandrian AE, Verani MS, editors. Nuclear cardiac imaging: principles and applications. 3rd ed. New York: Oxford University; 2003. p. 106–20.Google Scholar
- 18.Lima RSL, Watson DD, Goode AR, Siadaty MS, Ragosta M, Beller GA, et al. Incremental value of combined perfusion and function over perfusion alone by gated SPECT myocardial perfusion imaging for detection of severe three-vessel coronary artery disease. J Am Coll Cardiol 2003;42:64–70.CrossRefPubMedGoogle Scholar
- 19.Ragosta M, Bishop AH, Lipson LC, Watson DD, Gimple LW, Sarembock IJ, et al. Comparison between angiography and fractional flow reserve versus single-photon emission computed tomographic myocardial perfusion imaging for determining lesion significance in patients with multivessel coronary disease. Am J Cardiol 2007;99:896–902.CrossRefPubMedGoogle Scholar
- 23.Smith AM, Gullberg GT, Christian PE. Experimental verification of technetium 99m-labeled teboroxime kinetic parameters in the myocardium with dynamic single-photon emission computed tomography: reproducibility, correlation to flow, and susceptibility to extravascular contamination. J Nucl Cardiol 1996;3(2):130–42.CrossRefPubMedGoogle Scholar
- 24.Di Bella EV, Ross SG, Kadrmas DJ, Khare HS, Christian PE, McJames S, et al. Compartmental modeling of technetium-99m-labeled teboroxime with dynamic single-photon emission computed tomography: comparison with static thallium-201 in a canine model. Invest Radiol 2001;36(3):178–85.CrossRefPubMedGoogle Scholar
- 28.Yoshinaga K, Katoh C, Noriyasu K, Iwado Y, Furuyama H, Ito Y, et al. Reduction of coronary flow reserve in areas with and without ischemia on stress perfusion imaging in patients with coronary artery disease: a study using oxygen 15-labeled water PET. J Nucl Cardiol 2003;10:275–83.CrossRefPubMedGoogle Scholar
- 40.Blevis I, Tsukerman L, Volokh L, Hugg J, Jansen F, Bouhnik JP. CZT gamma camera with pinhole collimator: detector efficiency. Abstract: Nuclear Science Symposium 2008. IEEE 19–25 Oct 2008Google Scholar
- 41.Blevis I, Tsukerman L, Volokh L, Hugg J, Jansen F, Bouhnik JP. CZT gamma camera with pinhole collimator: spectral measurements. Abstract: Nuclear Science Symposium 2008. IEEE 19–25 Oct 2008.Google Scholar
- 44.Volokh L, Lahat C, Binyamin E, Blevis I. Myocardial perfusion imaging with an ultra-fast cardiac SPECT camera—a phantom study. Nuclear Science Symposium Conference Record, 2008. NSS '08. IEEE 19–25 Oct 2008. p. 4636–4639.Google Scholar
- 45.Cerqueira MD, Weissman NJ, Dilsizian V, Jacobs AK, Kaul S, Laskey WK, et al. Standardized myocardial segmentation and nomenclature for tomographic imaging of the heart: a statement for healthcare professionals from the Cardiac Imaging Committee of the Council on Clinical Cardiology of the American Heart Association. Circulation 2002;105(4):539–42.CrossRefPubMedGoogle Scholar
- 57.Santiago JF, Heiba SI, Jana S, Mirzaitehrane M, Dede F, Abdel-Dayem HM. Transient ischemic stunning of the myocardium in stress thallium-201 gated SPET myocardial perfusion imaging: segmental analysis of myocardial perfusion, wall motion and wall thickening changes. Eur J Nucl Med Mol Imaging 2002;29(8):979–83.CrossRefPubMedGoogle Scholar
- 67.Phillips DJ, Henneman RA, Merhige ME. Rapid diagnosis of coronary disease using dipyridamole and teboroxime washout imaging [abstract]. J Am Coll Cardiol 1994;23:255.Google Scholar
- 72.Leber W, Knez A, von Ziegler F, Becker A, Nikolaou K, Paul S, et al. Quantification of obstructive and nonobstructive coronary lesions by 64-slice computed tomography: a comparative study with quantitative coronary angiography and intravascular ultrasound. J Am Coll Cardiol 2005;46:147–54.CrossRefPubMedGoogle Scholar
- 77.Boden WE, O’Rourke RA, Teo KK, Maron DJ, Hartigan PM, Sedlis SP, et al. Impact of optimal medical therapy with or without percutaneous coronary intervention on long-term cardiovascular end points in patients with stable coronary artery disease (from the COURAGE Trial). Am J Cardiol 2009;104(1):1–4.CrossRefPubMedGoogle Scholar
Open AccessThis is an open access article distributed under the terms of the Creative Commons Attribution Noncommercial License (https://creativecommons.org/licenses/by-nc/2.0), which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.