Shear-wave elastography of the ulnar collateral ligament of the elbow in healthy volunteers: a pilot study
- 68 Downloads
To estimate the intra-observer repeatability of shear wave elastography in the UCL of the elbow, and to compare shear wave velocities between dominant and non-dominant arms.
Materials and methods
Twenty elbows in ten healthy volunteers were evaluated [five males, five females; mean age, 31.8 ± 10.3 years]. Shear wave velocity was measured on three separate days during the span of 1 week utilizing a linear 18-MHz transducer. Elastograms were obtained until ten ROIs were drawn, not drawing more than two ROIs on any elastogram. Elastograms were considered diagnostic if any portion of the UCL was colored in and free of boundary artifacts. Median velocity and interquartile range were recorded. A result was considered reliable if the IQR/median ratio of the ten measurements was < 0.3.
IQR/median was < 0.3 in 88% of sessions, although in 28% of sessions fewer than 60% of elastograms were diagnostic. The ICC was 0.05 (95% CI; − 0.18–0.36; poor). Repeatability coefficient (95% limits of agreement) was 1.95 m/s (95% CI; 1.61–2.37 m/s). Mean velocity in dominant arms was 5.14 ± 0.53 m/s and 5.24 ± 0.39 m/s in non-dominant (p = 0.558).
Mean shear wave velocity was similar between dominant and non-dominant arms. Although repeatability was poor as assessed by ICC, the repeatability coefficient may be a more useful indicator of clinical utility once shear wave velocities in diseased ligaments are explored. Future studies should therefore evaluate velocities in diseased ligaments and develop techniques to improve elastogram quality.
KeywordsUlnar collateral ligament Shear wave Elastography Elbow
The authors thank Domenica Delgado for her assistance throughout this project and Dr. Jett R. Brady for his advice in study conception. We also thank Dr. Todd Erpelding, PhD, and Canon Medical Systems, USA, Inc. for their system support.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflicts of interest.
- 16.Payne C, Watt P, Cercignani M, Webborn N. Reproducibility of shear wave elastography measures of the Achilles tendon. Skelet Radiol. 2018;47(6):779–84.Google Scholar
- 21.Atanda A Jr, Averill LW, Wallace M, Niiler TA, Nazarian LN, Ciccotti MG. Factors related to increased ulnar collateral ligament thickness on stress sonography of the elbow in asymptomatic youth and adolescent baseball pitchers. Am J Sports Med. 2016;44(12):3179–87.Google Scholar
- 22.Wada KW, Watanabe M, Tomita K, Takashima M, Inagaki K. Ultrasound elastographic assessment of the medial collateral ligament in. The elbow joints of baseball players. Showa University. J Med Sci. 2014;26(4):301–7.Google Scholar
- 36.Song P, Mellema DC, Sheedy SP, Meixner DD, Karshen RM, Urban MW, et al. Performance of 2-dimensional ultrasound shear wave elastography in liver fibrosis detection using magnetic resonance elastography as the reference standard: a pilot study. J Ultrasound Med. 2016;35(2):401–12.CrossRefGoogle Scholar
- 38.Sadeghi S, Rothenberger S, Akbarian D, Cortes DH. Effect of frequency and focal depth of push pulses on acoustic intensity, mechanical index, and shear wave amplitude for elastography imaging. SM J Biomed Eng. 2017;3(1):1015.Google Scholar