Ionomer and protein size analysis by analytical ultracentrifugation and electrospray scanning mobility particle sizer
- 247 Downloads
By combining analytical ultracentrifugation (AUC) in liquid phase and scanning mobility particle sizer (SMPS) in the gas phase, additional information on the particle size and morphology has been obtained for rigid particles. In this paper, we transfer this concept to soft particles, allowing us to analyze the size and molar mass of the short side chain perfluorosulfonic acid ionomer Aquivion® in a dilute aqueous suspension. The determination of the primary size and exact molar mass of this class of polymers is challenging since they are optically transparent and due to the formation of different aggregate structures depending on the concentration and solvent properties. First, validation of AUC and SMPS measurements was carried out using the well-defined biopolymers bovine serum albumin (BSA) and lysozyme (LYZ) to confirm the reliability of the results of the two unique and independent classifying methods. Then, the ionomer Aquivion® was studied using both techniques. From the mean molar mass of 185 ± 14 kDa obtained by AUC, a mean hydrodynamic diameter of 7.6 ± 0.5 nm was calculated. The particle size obtained from SMPS (7.1 nm) agrees very well with the results from AUC showing that the molecule was transferred into the gas phase without significantly changing its structure. In conclusion, the Aquivion® is molecularly dispersed in the used aqueous buffer solution without any aggregate formation in the investigated concentration range (< 2 g l−1).
KeywordsScanning mobility particle sizer Analytical ultracentrifugation Multiwavelength detector Combined analysis
The authors acknowledge HI ERN and Bavaria (Grant-no. DBF01253) as well as Greenerity GmbH for financial support of this work. The authors further acknowledge the funding of the Deutsche Forschungsgemeinschaft (DFG) through the Cluster of Excellence “Engineering of Advanced Materials” as well as DFG Project PE 427/28-2. TT acknowledges the fellowship from Alexander von Humboldt foundation. SEW acknowledges the travel grant from ARBRE-MOBIEU and COST for the 23. International Analytical Ultracentrifugation Workshop and Symposium.
- Bhattacharyya S, Maciejewska P, Börger L et al (2006) Development of a fast fiber based UV-vis multiwavelength detector for an ultracentrifuge. In: Wandrey C, Cölfen H (eds) Analytical ultracentrifugation, VIII edn. Springer, New York, p 9Google Scholar
- Cauchy A-L (1832) Mémoire sur la rectification des courbes et la quadrature des surfaces courbes. Oxford University, OxfordGoogle Scholar
- Demeler B (2005) UltraScan: a comprehensive data analysis software package for analytical ultracentrifugation experiments. In: Scott DJ, Harding SE, Rowe AJ (eds) Analytical ultracentrifugation: techniques and methods. Royal Society of Chemistry, London, p 210Google Scholar
- Fasman G (1976) Handbook of biochemistry and molecular biology: proteins, 3rd edn. CRC Press, ClevelandGoogle Scholar
- Grot W (1986) Nafion as a separator in electrolyte cells in Nafion product bulletin. DuPont Co., WilmingtonGoogle Scholar
- Kratky O, Leopold H, Stabinger H (1973) The determination of the partial specific volume of proteins by the mechanical oscillator technique in part D: enzyme structure. Elsevier 27:98Google Scholar
- Liu W-H, Yu T-Y, Yu T et al (2007) Static light scattering and transmission microscopy study of dilute Nafion solutions. e-Polymers 109:1Google Scholar
- Mächtle W, Börger L (2006) Analytical ultracentrifugation of polymers and nanoparticles. Springer, Berlin HeidelbergGoogle Scholar
- Uchiyama S, Arisaka F, Stafford WF, Laue T (eds) (2016) Analytical ultracentrifugation. Instrumentation software, and applications. Springer, TokyoGoogle Scholar
- van Holde K, Johnson W, Ho P (2006) Principles of physical biochemistry, 2nd edn. Pearson/Prentice Hall, Upper Saddle RiverGoogle Scholar