An interactive teaching device simulating intussusception reduction
- 800 Downloads
Intussusception is relatively uncommon, occurring in 0.5 to 2.3 cases per 1,000 live births in the USA. Radiology residents, therefore, have few opportunities to participate in intussusception reduction during training, and practicing radiologists encounter it infrequently. Training is essential, as successful reduction avoids surgery. The judgment involved in reducing an intussusception is best gained with experience. We developed a training device that simulates fluoroscopic intussusception reduction with air. The device consists of a doll that contains a cylinder with similar stress and strain characteristics to the human colon. The trainee pumps air into the cylinder through a rectal tube using a standard hand-held air reduction pump. A sensor measures the pressure within the chamber and transmits readings to a computer, which displays images from actual intussusception reductions based on the pressure maintained within the device. A random component in the software gives the user a new experience each time and models uncertainties in the actual reduction process, including perforation. This intussusception reduction simulator can enhance resident education, giving residents the opportunity to practice this technique before employing it on a real patient. The simulator can also help practicing radiologists become more comfortable with intussusception air reduction.
KeywordsIntussusception Reduction Teaching model Education Simulation Children
Intussusception is an important cause of acute abdomen in infancy and childhood. The incidence of intussusception is estimated to be 0.5–2.3 cases per 1,000 live births in the USA [1, 2]. However, it is likely that the incidence is actually higher, as some cases resolve spontaneously before being confirmed either radiographically or surgically . The relative rarity of this condition limits radiology residents’ opportunity to gain supervised experience in the non-surgical reduction of intussusception. A recent survey found that 22% of senior radiology residents have had no experience in reducing intussusception, while 21% have been involved with only one case during their entire residency . However, Meyer et al.  demonstrated that a radiologist must be involved in at least four air intussusception reductions to become skillful with this technique.
Nonsurgical reduction of intussusception is the preferred method of treatment, and both air and liquid reduction have been shown to be safe and effective . Hence, it falls squarely on the radiologist to be familiar with and proficient at performing this procedure so that patients avoid unnecessary surgery. Since the judgment, confidence and skill involved in reducing intussusception are best gained with experience, radiology residents—and any other practicing radiologists who might be asked to treat a patient’s intussusception—will benefit from the training tool we describe here to supplement their caseload. Simulation tools such as this intussusception reduction simulator constitute a major revolution in medical education .
This was a collaborative effort among radiologists and both student and faculty engineers at the UC Davis College of Engineering and was reviewed and approved by our institution’s IRB. We developed an interactive teaching device to simulate reducing an intussusception with air under the guidance of an instructor. During the simulation, the instructor controls the ease of maintaining pressure within the device and thus the speed of the reduction, while coaching the trainee. The interaction between the instructor and the trainee is essential to the learning process.
Help the trainee recognize the progression and reduction of intussusception.
Simulate complications such as perforation that the trainee should detect immediately.
Monitor the fluoroscopy time with the intention of improving technique to lower radiation dose.
Allow the instructor to control the trainee’s success at maintaining adequate pressure, so that the trainee develops awareness of the importance of the rectal “seal.”
Create a compact, portable device that can be easily assembled and that has a user-friendly interface for both trainee and instructor.
The trainee using this simulator device can usually reduce the intussusception in about 5 min if he or she maintains pressure of approximately 120 mmHg, and sometimes succeeds with lower pressures. If he or she does not maintain sufficient pressure, the intussusception often fails to reduce and can regress. If the trainee delivers too much pressure, the chance of perforation increases. The simulation progresses much more quickly than a real-life intussusception reduction in order to limit instructor and trainee fatigue. In addition, because it is critical that a trainee learn to recognize pneumoperitoneum, the probability of perforation is higher than in real life. This gives the trainee opportunities to discuss and simulate management of tension pneumoperitoneum. The simulation progresses quickly and with lower pressures in the earlier stages of reduction, as is common during actual procedures.
Videos of a trainee and instructor reducing an intussusception and of the trainee encountering a perforation are available as electronic supplementary materials 1 and 2.
George M. Retan described fluoroscopically guided hydrostatic reduction of intussusception in 1927, making it one of the first interventional radiological procedures . It was subsequently popularized by Ravitch and McCune  in their experimental study in 1948. Since then, fluoroscopically guided reduction has been considered the initial management of choice for most uncomplicated cases of childhood intussusception. Although US has been used effectively to guide the reduction of intussusception [14, 15], fluoroscopic monitoring is preferred among general and pediatric radiologists in the United States. A survey performed among pediatric radiologists in European teaching hospitals showed that more than 90% of respondents preferred fluoroscopically guided reduction .
Both air and liquid are commonly employed for intussusception reduction, but there is a growing trend among pediatric radiologists to shift from hydrostatic to pneumatic reduction [17, 18]. Air reduction is a quick and clean technique that is easy to learn  and is associated with smaller perforations with less fecal peritoneal contamination than liquid enemas . The filling speed with air is also about seven times faster than fluid, facilitating a higher force of reduction . In comparison to fluoroscopic hydrostatic reduction, pneumatic reduction requires less radiation exposure because of shorter fluoroscopic times and lower kilovolt peak and mA settings [5, 21]. Because of these advantages coupled with its greater mechanical simplicity, we chose to develop a model that would use the pneumatic technique.
We attempted to simulate experiences that are commonly encountered in intussusception reduction. (1) Air leak occurs when the instructor releases a valve in the tubing. The trainee must recognize that the intussusceptum has stopped reducing or has even regressed and then consider what could be causing the insufflator to cease transmitting pressure. The trainee should carefully assess for pneumoperitoneum, as a sudden pressure drop can occur when the bowel perforates. The trainee might also note a sudden inability to maintain previous pressure levels. (2) The program allows reduction to occur at various speeds, sometimes requiring the trainee to apply pressure for sustained periods without appreciable change in intussusceptum position. Although reduction success correlates with pressure, the use of a random number generator simulates the inherent uncertainty in the process of reducing an intussusception. (3) Some of the images are difficult to interpret because of poor contrast and overlapping air-filled bowel loops, as well as differing orientation depending on whether the patient is supine or prone. The trainee must quickly decipher the images and decide whether reduction is progressing or complete. (4) With excessive pressure—and sometimes with pressures less than 120 mmHg—perforation can occur; the trainee must instantly recognize this complication and discuss treatment of tension pneumoperitoneum. There is no specific trigger for perforation, however, as even at pressures as high as 300 mmHg perforation occurs in only 5% of patients . Because it is important to recognize this complication, pneumoperitoneum is more common in this simulation than has been reported in actual cases . As in a real intussusception reduction, the trainee must multitask during this procedure by simultaneously controlling the insufflator, monitoring the pressure, ensuring a tight grip on the buttocks, delivering radiation, interpreting fluoroscopic images and springing into action if perforation occurs.
One intrinsic weakness of this device arises from the fact that the statistical likelihood of progression or regression of reduction at multiple specific pressures is apparently not established, with success reported at different pressures that vary among researchers [5, 8, 9, 10, 11, 21]. Therefore the values we have chosen are estimates, based on empirical testing of the device, our own experience and literature review. If there is sufficient interest in this device, we will improve it by correlating the patient’s clinical presentation with the ease of intussusception reduction. When using this device, the trainee experiences a shorter version of what can be a very lengthy procedure, as we found that both trainees and instructors fatigued when the reduction proceeded at rates often encountered in routine clinical practice. When faced with a real intussusception reduction, the trainee might therefore think that his or her attempt at reduction is unsuccessful if it does not progress as rapidly as it does in this simulator; the instructor must make this clear during the simulation. In addition, the image showing perforation is from a different patient from that in the initial radiographs, making it easy for the resident to recognize that this complication has occurred; however, we have no pneumoperitoneum image occurring during an intussusception reduction in our practice to date. In order to better simulate the actual reduction experience, the device could include a recording of a crying child. Finally, the experience could have been enhanced by allowing the trainee to deliver continuous fluoroscopy (so that they could use excessive radiation more easily, increasing the opportunity for instruction about appropriate radiation delivery), but this proved to be technically challenging. If there is widespread interest in this device, we will work with a commercial partner to make it available and also develop an instructor’s manual delineating essential feedback and other points the instructor should emphasize.
Although air reduction has many advantages over liquid, many practicing radiologists were trained with liquid reduction and are uncomfortable with air. This device can help these practitioners transition to using the air reduction technique, enabling them to offer their patients the advantages that air reduction provides. Air reduction is considered easy to learn, but it is difficult to learn in a vacuum.
In conclusion, most radiology residents and general radiologists have limited hands-on experience in reducing intussusception. Although some children with intussusception are treated at pediatric hospitals—and thus by radiologists with a great deal of experience with reduction—many patients present to smaller community facilities, and the success of their reduction depends on a radiologist who may never have performed this procedure, either during training or afterward. Failed reduction in the community often necessitates transfer to a tertiary care facility, and the resulting delay can diminish the likelihood that a repeat attempt at reduction will succeed. Since judgment and confidence are best gained and refined by experience, this simulator helps compensate for the rarity of intussusception by simulating important components of air reduction. Residents, fellows and even nervous attending radiologists may, after practicing pneumatic reduction with this device, develop the necessary skills and confidence to successfully reduce their next case of intussusception.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.
(MPG 26844 kb)
(MPG 72906 kb)
Open AccessThis is an open access article distributed under the terms of the Creative Commons Attribution Noncommercial License (https://creativecommons.org/licenses/by-nc/2.0), which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.