Conventional and CT angiography in children: dosimetry and dose comparisons
- 858 Downloads
Tremendous advances have been made in imaging in children with both congenital and acquired heart disease. These include technical advances in cardiac catheterization and conventional angiography, especially with advancements in interventional procedures, as well as noninvasive imaging with MR and CT angiography. With rapid advances in multidetector CT (MDCT) technology, most recently 64-detector array systems (64-slice MDCT), have come a number of advantages over MR. However, both conventional and CT angiography impart radiation dose to children. Although the presence of radiation exposure to children has long been recognized, it is apparent that our ability to assess this dose, particularly in light of the rapid advancements, has been limited. Traditional methods of dosimetry for both conventional and CT angiography are somewhat cumbersome or involve a potential for substantial uncertainty. Recent developments in dosimetry, including metal oxide semiconductor field effect transistors (MOSFET) and the availability of anthropomorphic, tissue-equivalent phantoms have provided new opportunities for dosimetric assessments. Recent work with this technology in state-of-the-art cardiac angiography suites as well as with MDCT have offered direct comparisons of doses in infants and children undergoing diagnostic cardiac evaluation. It is with these dose data that assessment of risks, and ultimately the assessment of risk-benefit, can be better achieved.
KeywordsALARA CT radiation Cardiac angiography Infants and children
There have been tremendous advances in cardiac imaging in children during the last several decades. These include technical advances for both noninvasive imaging, such as multidetector array CT (MDCT) angiography , and conventional angiography during heart catheterization. These advances have accelerated recently, and are part of the basis for this ALARA conference. Although these advances provide a substantial diagnostic benefit, these modalities are not without risk. Because both CT angiography and conventional angiography require ionizing radiation, this is also a shared risk. Although the bioeffects of ionizing radiation include both deterministic and stochastic effects (discussed elsewhere in this issue), the doses used for CT angiography are well below those used for deterministic effects. This is not the same for conventional angiographic assessment, where skin burns have been reported after interventional procedures . The stochastic risk is in common for these two imaging modalities. Because of this, it is worth reviewing several facets of the radiation issue.
Therefore, the following material (1) briefly summarizes the rapid evolution of imaging techniques, (2) addresses whether the issue of potential radiation-induced cancer is still important, (3) discusses how traditional radiation dosimetry is problematic, (4) reviews some of the current investigations using contemporary dosimetric evaluation for both CT and conventional angiography, and (5) compares doses for both CT and conventional angiography.
Rapid evolution of imaging techniques
Is radiation exposure during diagnostic procedures still an important issue?
The issue of radiation dose from CT and potential development of cancer followed a series of articles in the American Journal of Roentgenology [4, 5, 6]. Although this was more than 5 years ago, the issue of potential bioeffects of low-level radiation continues to be a topical one. For example, the National Toxicology Program recently listed radiation, which includes discussion of medical radiation, as a known carcinogen . Even more recently, in June 2005, the National Academy of Sciences [Biological Effects of Ionizing Radiation (BEIR) VII Report] discussed this issue of low-level radiation and potential cancer development. Among comments were the following: ‘–the risk of cancer proceeds in a linear fashion at lower doses without a threshold and –the smallest dose has the potential to cause a small increased risk to humans–. Although there continues to be some debate about the risk, if any, of low-level radiation, the fact remains that a number of organizations continue to support the perspective of a potential connection. Obviously, this includes the Society for Pediatric Radiology, with this ALARA conference, but it also involves, notably, organizations such as the American Academy of Pediatrics, with a clinical report on CT and radiation ; the National Cancer Institute and the Society for Interventional Radiology (SIR), with a March 2005 advisory paper on radiation risks for patients and staff ; and the American College of Radiology Safety Committee, which is developing a White Paper on radiation risks and diagnostic imaging (David Kushner, M.D., personal communication).
In addition, the technical advances continue to, it seems, outpace our efforts at understanding the actual dose. This is particularly evident with gated cardiac examinations in children, as is discussed below.
Current dosimetry: what are the problems and solutions?
Some of the problems with MDCT include the following. First, the technology is powerful. Diagnostically, this provides us with rapid, high-quality examinations that have had marked advantages in children [1, 11]. But the word ‘powerful–also subsumes potentially higher radiation doses than were seen with conventional CT and older MDCT technology. For example, recent work with phantoms in the setting of 64-slice MDCT has shown that doses in the range of 50 mSv to nearly 120 mSv can be generated by maximizing those settings that provide radiation dose in children (Donald Frush, M.D., unpublished data). There is no provision (either from a regulatory standpoint or a manufacturer standpoint) that prevents this from occurring. Simply stated, the radiology community and manufacturers, as well as regulatory agencies, need to work toward a regulatory component of CT, such as warning indicators for certain high radiation dose examinations.
One other problem with MDCT is that the scientific community is lagging in terms of providing data on CT use. This includes outcomes for the newer MDCT as well as technical parameters for optimizing examination. These data lag behind both marketing pressures to sell the newest CT scanner and public perceptions and demands of the benefits of these types of scanners. This has been seen with screening examinations as well as gated cardiac CT studies in adults. The presumed validation of gated examinations in cardiac screening was recently promulgated by Oprah Winfrey . Although the outcome of this sort of promotion has not been scientifically rigorously tested, the influence on public opinion by such spokespersons and subsequent provocative trends in use are, in general, recognized.
In addition, and perhaps most important, dose data for both the newer MDCT examinations and the contemporary angiographic equipment are lacking. This is multifactorial but in large part a result of the fact that previous methods of dosimetric assessment have been cumbersome. For example, surface assessment such as dose area product (DAP) does not provide organ dose and effective dose equivalent data for determination of risks. Traditional thermoluminescence dosimetry techniques are essentially single-exposure and take a great deal of effort to load into (and remove from) the phantoms, including anthropomorphic phantoms. Essentially, they are a single-exposure device during a CT or angiographic examination. For a CT examination, the CT dose index (CTDI) and resultant dose length product (DLP) displayed on the console are problematic for MDCT [13, 14], and in general will misrepresent the true dose to the patient. In addition, the conversion factors used with the DLP to estimate the effective dose might not reflect the dose resulting from current CT technology. Other methods of dosimetric assessment such as those using Monte Carlo codes might have not been validated using contemporary CT equipment. What is needed, then, is an accurate and simple dosimetric technology. This would allow more versatile assessment of current cardiac imaging modalities and allow for dose comparisons. This, then, would help in establishing the risk–benefit balance.
The first stage in the use of this combination of dosimetry tools was validation. This was carried out using traditional TLD dosimetry and Monte Carlo modeling. In adult CT examinations, MOSFET technology was found to be accurate and reliable compared with these other two more conventional techniques during chest and abdomen MDCT examinations. The advantage of the MOSFET technology is that this is a real-time assessment. Data are automatically transmitted to a laptop computer. Multiple CT examinations (or angiographic ‘runs– can be obtained in a single session, a great advantage compared with essentially single examination TLD technology.
After this validation was completed, we evaluated the MOSFET dosimetry and phantom model with CT dosimetry estimation, namely the DLP estimation. The DLP is derived from the CTDI available on scanner consoles. Using a conversion factor, the DLP estimates the dose for the MDCT parameters selected. Again, this dose does not have anything to do with the individual patient but is derived from an acrylic phantom (two sizes only) exposed using the individual settings. As mentioned before, we have found that this DLP method is relatively inaccurate in providing doses. For example, in a series of comparisons between MOSFET technology and the DLP method in adult chest and abdomen scanning, there was up to 50% greater effective dose measured by MOSFET technology than estimated by the DLP method (Lynne Hurwitz, Durham, N.C., unpublished data).
In summary, current technology provides new opportunities for dosimetric assessment in diagnostic imaging. In particular, comparison of the effective dose for pediatric cardiac angiography and CT angiography is now possible.
Dose comparisons for MDCT and conventional angiography in children
This is divided into three sections: CT angiography, gated CT angiography, and conventional angiography. Much of what is discussed here has been information recently presented at national meetings or in the press. Summary data for this material are provided.
A gated cardiac CT provides additional challenges in children. The nature of this examination, traditionally with relatively high tube current and very low pitch, makes it a potentially very high-dose examination. Despite this fact, excellent anatomic information can be obtained and can obviate cardiac catheterization and associated risks. Data regarding doses for pediatric gated cardiac CT, however, are limited. In our laboratory we have found that the adult dose range is approximately 11 mSv to more than 20–5 mSv; however, published data in children are lacking. We looked at gated cardiac CT in children and found, depending on the setting used, that doses could range just over 7.0 mSv to more than 25 mSv (Caroline Hollingsworth, Durham, N.C., unpublished data). Although image quality in these phantom studies was not assessed, our experience has been that acceptable image quality can be obtained with examinations performed at the lower end of these doses (Fig. 1). With that said, it is reasonable to conclude that gated cardiac CT examinations will provide doses that are probably multiples of nongated CT angiographic effective dose equivalents in children.
In summary, the dose will depend on a number of factors, most obviously fluoroscopic time, and projection, use of fluoroscopy vs. cine angiography, and age and size of patient. Although it would have been possible to recreate an individual catheterization procedure, we found it more versatile to develop a dose calculator based on a dose rate (per second) depending on the projection for both cine and fluoroscopy. We found in the 1-year-old and 5-year-old that we were able to calculate effective dose equivalent for procedures. This applied to both prospective assessment for a procedure where the length of fluoroscopic and cine evaluation for the various projections could be estimated. In addition, this calculator allowed one to go back and determine a dose estimate knowing this information on a procedure that had already been performed. In general, CT angiographic (nongated) dosimetry was below conventional angiographic assessment (diagnostic only) for routine evaluation of structures such as the aorta for coarctation, or pulmonary arteries for pulmonary arteries stenosis. Again, the dose delivered will vary depending on the individual patient needs, and the physician performing the examination. Be that as it may, the calculator and the information provided from our dosimetry from MDCT offered a more versatile and accurate tool for assessment of the relative doses (for the 1- and 5-year-old children) and determination of potential risks of these procedures.
There has been a rapid evolution of imaging techniques. The traditional methods of dosimetry have substantial limitations for assessing radiation dose from evolving cardiac imaging modalities consisting of conventional angiography and CT angiography. However, use of MOSFET technology together with a set of pediatric anthropomorphic phantoms has resulted in an accurate and simple, extremely versatile method for radiation dose determination. In particular, this allowed us to better estimate doses for pediatric cardiac imaging during MDCT and conventional angiography. From these dosimetry tools, the effect of techniques such as decreasing tube current for gated CT examinations in small children will likely ensue, in the spirit of the ALARA principle.
- 3.Frush DP (2004) Computed tomography. Advance for Imaging and Oncology Administrators 14:13–0Google Scholar
- 7.Department of Health and Human Services (2006) 11th report on carcinogens. Public Health Service, National Toxicology Program. http://www.ntp.niehs.nih.gov/ntp/roc/toc11.html
- 8.National Academy of Sciences (2006) Health risks from exposure to low levels of ionizing radiation: BEIR VII Phase 2. National Academies Press, Washington DC. http://www.fermat.nap.edu/openbook/030909156X/html/7.html
- 9.Brody AS, Frush DP, Huda W, et al (2006) American Academy of Pediatrics clinical report: radiation risk to children from CT imaging. Pediatrics (in press)Google Scholar
- 10.National Cancer Institute (2006) Interventional fluoroscopy: reducing radiation risks for patients and staff. http://www.cancer.gov/cancertopics/interventionalfluoroscopy
- 12.The Oprah Winfrey Show (2006) http://www.oprah.com/health/yourbody/slide/numberone/slide_yourbody_numberone_102.jhtml