Histological changes caused by the prolonged placement of ureteral access sheaths: an experimental study in porcine model
- 94 Downloads
The objective is to evaluate the histological damage to the ureteral wall caused by the prolonged placement of an access sheath in porcine model. Six ureters from three female pigs were randomized into three groups. In each group, an UAS with different indwelling time was inserted. 9.5/11.5 Fr Flexor® ureteral access sheaths were inserted in both ureters with left indwelling for 30 min and right for 60 min. The ureteral access sheath was advanced up to the proximal ureter. No resistance was observed during the insertion. Ureters were harvested in immediately after the sheath placement process in one pig. The ureters of the remaining two pigs were removed at 1 and 2 weeks after the procedure, respectively. Histological examination took place also in these specimens. Ureters with an indwelling time of 30 min: histological examination of the ureter after immediate dissection revealed signs of acute inflammation at the distal ureter. The ureter dissected at 1 week showed minimal focal transmural inflammation along its length with minimal epithelial hyperplasia. The ureter dissected at 2 weeks, no signs of inflammation. Ureters with an indwelling time of 60 min: histological examination of the immediately dissected ureter revealed signs of acute inflammation at the distal ureter. At 1 week, chronic transmural inflammation was predominantly observed in the distal ureter. At 2 weeks, minimal transmural inflammation was observed. The use of UAS did not cause any severe histological damage on porcine ureters. Acute signs of inflammation gradually recovered within 2 weeks.
KeywordsAccess sheath Ureteroscopy Porcine Ureteral damage RIRS
Compliance with ethical standards
Conflict of interest
None of the authors has any conflict of interest.
No funding was received for this study.
All applicable international, national, and/or institutional guidelines for the care and use of animals were followed.
- 9.Schoenthaler M, Buchholz N, Farin E, Ather H, Bach C, Bach T, Denstedt JD, Fritsche HM, Grasso M, Hakenberg OW, Herwig R, Knoll T, Kuehhas FE, Liatsikos E, Liske P, Marberger M, Osther PJ, Santos JM, Sarica K, Seitz C, Straub M, Traxer O, Trinchieri A, Turney B, Miernik A (2014) The Post-Ureteroscopic Lesion Scale (PULS): a multicenter video-based evaluation of inter-rater reliability. World J Urol 32:1033–1040CrossRefPubMedGoogle Scholar
- 12.Traxer O, Wendt-Nordahl G, Sodha H, Rassweiler J, Meretyk S, Tefekli A, Coz F, de la Rosette JJ (2015) Differences in renal stone treatment and outcomes for patients treated either with or without the support of a ureteral access sheath: the Clinical Research Office of the Endourological Society Ureteroscopy Global Study. World J Urol 33:2137–2144CrossRefPubMedPubMedCentralGoogle Scholar