Radiation dose optimization in pediatric temporal bone computed tomography: influence of tube tension on image contrast and image quality
- First Online:
- 183 Downloads
The purpose of this experimental study was to investigate the effect of tube tension reduction on image contrast and image quality in pediatric temporal bone computed tomography (CT).
Seven lamb heads with infant-equivalent sizes were scanned repeatedly, using four tube tensions from 140 to 80 kV while the CT-Dose Index (CTDI) was held constant. Scanning was repeated with four CTDI values from 30 to 3 mGy. Image contrast was calculated for the middle ear as the Hounsfield unit (HU) difference between bone and air and for the inner ear as the HU difference between bone and fluid. The influence of tube tension on high-contrast detail delineation was evaluated using a phantom. The subjective image quality of eight middle and inner ear structures was assessed using a 4-point scale (scores 1–2 = insufficient; scores 3–4 = sufficient).
Middle and inner ear contrast showed a near linear increase with tube tension reduction (r = −0.94/−0.88) and was highest at 80 kV. Tube tension had no influence on spatial resolution. Subjective image quality analysis showed significantly better scoring at lower tube tensions, with highest image quality at 80 kV. However, image quality improvement was most relevant for low-dose scans.
Image contrast in the temporal bone is significantly higher at low tube tensions, leading to a better subjective image quality. Highest contrast and best quality were found at 80 kV. This image quality improvement might be utilized to further reduce the radiation dose in pediatric low-dose CT protocols.
KeywordsImage quality Temporal bone CT Tube tension Radiation exposure to patients Pediatric CT
- 10.Radiation risks and pediatric computed tomography (CT): a guide for health care providers National Cancer Institute Website. http://www.cancer.gov/cancertopics/causes/radiation/radiation-risks-pediatric-CT. Accessed 13th of July 2011
- 11.Goske MJ, Applegate KE, Boylan J, Butler PF, Callahan MJ, Coley BD, Farley S, Frush DP, Hernanz-Schulman M, Jaramillo D, Johnson ND, Kaste SC, Morrison G, Strauss KJ, Tuggle N (2008) The Image Gently campaign: working together to change practice. AJR Am J Roentgenol 190(2):273–274. doi:10.2214/AJR.07.3526 PubMedCrossRefGoogle Scholar
- 13.Szucs-Farkas Z, Semadeni M, Bensler S, Patak MA, von Allmen G, Vock P, Schindera ST (2009) Endoleak detection with CT angiography in an abdominal aortic aneurysm phantom: effect of tube energy, simulated patient size, and physical properties of endoleaks. Radiology 251(2):590–598. doi:10.1148/radiol.2512081687 PubMedCrossRefGoogle Scholar