Measurement of human serum unconjugated estriol without derivatization using liquid chromatography-tandem mass spectrometry candidate reference method and compared with two immunoassays
A candidate reference measurement procedure (RMP) for measurement of unconjugated estriol in human serum has been developed and validated. The proposed method is highly reliable and uses isotope dilution coupled with liquid chromatography-tandem mass spectrometry (ID-LC-MS/MS) and requires no derivatization. An appropriate amount of serum was accurately weighed and spiked with an isotopically labeled internal standard. Unconjugated estriol and its internal standard were extracted from serum matrix using liquid-liquid extraction prior to reversed-phase LC-MS/MS. Calibrator bracketing was used to give higher specificity and accuracy for assigning serum level. The accuracy of the candidate RMP was validated by split-sample comparison to established RMPs. The lowest limit of detection (LLoD) and lowest limit of quantification (LLoQ) for developed RMP was estimated to be 0.14 nmol/L and 0.35 nmol/L, respectively. Both intra- and inter-assay imprecisions were ≤2.19% at 1.39, 17.34 and 69.35 nmol/L, respectively. Recoveries were 98.54% to 100.34% and linear response ranged from 0.35 to 173.38 nmol/L. No interference was observed. Biases were 5.6% and 2.8% against the targets of RELA2015A (3.87 nmol/L) and RELA2015B (40.62 nmol/L), respectively. Moreover, the candidate RMP was successfully applied to measure level of unconjugated estriol in serum samples of pregnant women (n = 3) and compared with two immunoassays in clinical laboratory. Our developed method is simple, accurate, and can be used as a candidate RMP to determine total unconjugated estriol level in human serum. Further improvement of certain immunoassays in accuracy and precision is needed.
KeywordsEstriol Reference method Tandem mass spectrometry Immunoassay
liquid chromatography/tandem mass spectrometry
multiple reaction monitoring
reference measurement procedure
Joint Committee for Traceability in Laboratory Medicine
Reference Laboratories in Laboratory Medicine
International Federation of Clinical Chemistry and Laboratory Medicine
The authors thank Prof. Zhiguo Wang for sharing information on the uE3 results of the PT event 2016 conducted by China National Center for Clinical Laboratory.
This work was supported by the National Natural Science Foundation of China (81572088), the Natural Science Foundation of Guangdong Province (2015A030313340), the Specific Research Fund for TCM Science and Technology of Guangdong Provincial Hospital of Chinese Medicine (YN2014ZHR204 and YN2016QJ15), and Administration of Traditional Chinese Medicine Bureau of Guangdong Province (20172063).
Compliance with ethical standards
Conflicts of interest
All authors declare that they have no conflict of interest.
The research involved human participants and/or animals
The research involves serum samples taken from pregnant women and this study is approved by Ethics Committee of Guangdong Provincial Hospital of Chinese Medicine (the other name: Second Affiliated Hospital of Guangzhou University of Chinese Medicine). The number of this Review is: B2016-159-01.
Ethics committee approved the exemption of informed consent.
- 1.Pepe GJ, Albrecht ED. Actions of placental and fetal adrenal steroid hormones in primate pregnancy. Endocr Rev. 1995;16(5):608–48.Google Scholar
- 2.Cutolo M, Sulli A, Seriolo B, Accardo S, Masi AT. Estrogens, the immune response and auto-immunity. Clin Exp Rheumatol. 1995;13(2):217–26.Google Scholar
- 4.Fishman J, Brown JB, Hellman L, Zumoff B, Gallagher TF. Estrogen metabolism in normal and pregnant women. J Biol Chem. 1962;237(237):1489–94.Google Scholar
- 13.Reynolds T, John R. Comparison of assay kits for unconjugated estriol shows that expressing results as multiples of the median causes unacceptable variation in calculated risk factors for syndrome. Clin Chem. 1992;38(9):1888–93.Google Scholar
- 17.College of Pathologists (CAP) Ligand (Special) Y-B 2016 Participant Summary.Google Scholar
- 18.Siekmann L. Determination of steroid hormones by the use of isotope dilution-mass spec-trometry: a definitive method in clinical chemistry. J Steroid Biochem. 1979;11(1):117–23.Google Scholar
- 19.Gust M, Vulliet E, Giroud B, Garnier F, Couturier S, Garric J, et al. Development, validation, and comparison of LC-MS/MS and RIA methods for quantification of vertebrates-like sex steroids in prosobranch mollusks. J Chromatogr B. 2010;878(19):1487–92.Google Scholar
- 21.Zhang Q, Han L, Wang J, Lin H, Ke P, Zhuang J, et al. Simultaneous quantitation of endogenous estrone, 17beta-estradiol, and estriol in human serum by isotope-dilution liquid chromatography-tandem mass spectrometry for clinical laboratory applications. Anal Bioanal Chem. 2017;409(10):2627–38.CrossRefGoogle Scholar
- 28.International Organization for Standardization (ISO). In vitro diagnostic medical devices: measurement of quantities in samples of biological origin–requirements for content and presentation of reference measurement procedures. Geneva: ISO; 2009. ISO15193.Google Scholar
- 29.International Organization for Standardization (ISO). In vitro diagnostic medical devices: metrological traceability of values assigned to calibrators and control materials. Geneva: ISO; 2003. ISO 17511.Google Scholar
- 30.Botelho JC, Shacklady C, Cooper HC, Tai SS, Van Uytfanghe KV, Thienpont LM, Vesper HW (2013) Isotope-dilution liquid chromatography-tandem mass spectrometry candidate reference method for total testosterone in human serum. Clin Chem 59(2):372–380Google Scholar
- 32.Clinical and Laboratory Standards Institute (CLSI). Liquid Chromatography-Mass Spectrometry Methods; Approved Guideline. William C: CLSI, 2014. CLSI document C62-AGoogle Scholar
- 33.International Organization for Standardization (ISO)/IEC Guide 98-3:2008. Uncertainty of measurement –Part 3: Guide to the expression of uncertainty in measurement (GUM:1995). Geneva: ISO; 2008.Google Scholar
- 34.Clinical and Laboratory Standards Institute (CLSI). Preliminary evaluation of quantitative clinical laboratory measurement procedures; Approved guideline -3rd edition. Wayne: CLSI, 2014. CLSI document EP10-A3-AMD.Google Scholar
- 35.International Federation of Clinical Chemistry and Laboratory Medicine (IFCC). RELA-IFCC external quality assessment scheme for Reference Laboratories in Laboratory Medicine. Available at: http://www.dgkl-rfb.de:81/4Daction/g_search_RELA/RELA2015. Accessed Jan 2017.
- 36.International Federation of Clinical Chemistry and Laboratory Medicine (IFCC). RELA-IFCC external quality assessment scheme for Reference Laboratories in Laboratory Medicine. Available at: http://www.dgkl-rfb.de:81/4Daction/g_search_RELA/RELA2014. Accessed Sep 2015.
- 37.Clinical and Laboratory Standards Institute (CLSI). Procedures for the handling and pro- cessing of blood specimens for common laboratory tests; Approved guideline, 4th edition. Wayne: CLSI, 2010. CLSI document H18-A4.Google Scholar
- 38.Jung PG, Kim B, Park SR, So HY, Shi LH, Kim Y. Determination of serum cortisol using isotope dilution–liquid chromatography–mass spectrometry as a candidate reference method. Anal Bioanal Chem. 2004;380(5–6):782–8.Google Scholar
- 39.Sargent M, Harrington C, Harte R. Guidelines for achieving high accuracy in isotope dilution mass spectrometry (IDMS). LGC Limited: Royal Society of Chemistry, 2002.Google Scholar
- 40.Botelho JC, Ribera A, Cooper HC, Vesper HW. Evaluation of an Isotope Dilution HPLC Tandem Mass Spectrometry Candidate Reference Measurement Procedure for Total 17-β Estradiol in Human Serum. Anal. Chem.,2016; 88 (22):11123–9.Google Scholar
- 41.Kushnir MM, Komaromy-Hiller G, Shushan B, Urry FM, Roberts WL. Analysis of dicar- boxylic acids by tandem mass spectrometry. High-throughput quantitative measurement of methylmalonic acid in serum, plasma, and urine. Clin Chem. 2001; 47(11): 1993–2002.Google Scholar
- 42.Stöckl D, Sluss PM, Thienpont LM. Specifications for trueness and precision of a reference measurement system for serum/plasma 25-hydroxyvitamin D analysis. Clin Chim Acta. 2009;408(1):8–13.Google Scholar
- 43.Fraser CG, Kallner A, Kenny D, Petersen PH. Introduction: Strategies to set global quality specifications in laboratory medicine. Scand J Clin Lab Invest. 1999; 59(7):477–8.Google Scholar
- 44.LM. Thienpont. Quality specifications for reference methods. Scand J Clin Lab Invest 1999; 59:535–538.Google Scholar
- 45.StoÈckl D, Reinauer H. Development of criteria for the evaluation of reference method values. Scand J Clin Lab Invest 1993; 53 Suppl 212:16–8.Google Scholar
- 46.Clinical and Laboratory Standards Institute (CLSI). Measurement procedure comparison and bias estimation using patient samples; Approved guideline, 3rd edition. Wayne: CLSI, 2013. CLSI document EP09-A3.Google Scholar