Compact devices for generation of reference trace VOC mixtures: a new concept in assuring quality at chemical and biochemical laboratories
Volatile organic compounds (VOCs) in gas mixtures at trace level (nmol/mol) are routinely measured by chemical and biochemical laboratories as climate indicators, indoor air quality pollutants from building materials emissions, contaminants in food and beverages, and biomarkers in body fluids (blood, urine, breath) of occupational exposure or human diseases. Current analytical instruments used for measurements are gas chromatographs equipped with various injector and detector configurations. The assurance of measurement quality is done by using a huge amount of certified liquid VOC standard solutions (or gaseous VOC standard cylinders) with multiple dilutions to reach the required trace level. This causes high standard uncertainty in instrument calibrations, high cost, and high consumption of analysis and laboratory personal time. In this paper, we present the implementation of portable generators producing VOC gas standards at trace level for automatic and direct calibration of VOC detectors employed in various contexts, removing the need for preparation of matrix calibration standards in cylinders. Two compact devices in-house developed by two national metrology institutes—the Istituto Nazionale di Ricerca Metrologica (INRIM) and the Federal Institute of Metrology (METAS)—are here used to dynamically generate reference gas mixtures in an SI traceable way. The two devices are based on different technologies: diffusion and permeation, for INRIM and METAS, respectively. A metrological characterization is given and the practical implementation at chemical and biochemical laboratories is discussed.
KeywordsReference VOC gas mixtures Dynamic generation Metrological traceability GC calibration Chemical laboratory Biochemical laboratory
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
- 1.Barrie L, Braathen G. Highlights from the Global Atmosphere Watch Programme; 2017.Google Scholar
- 2.Penkett S, Barrie L. WMO/GAW Expert Workshop on Global Long-term Measurements of Volatile Organic Compounds, WMO TD No. 1373. WMO; 2007.Google Scholar
- 4.European Metrology Research Programme. ENV56 KEY-VOCs research project 2014–2017. n.d. http://www.key-vocs.eu/.
- 6.Ashley DL, Bonin MA, Cardinali FL, McCraw JM, Wooten JV. Blood concentrations of volatile organic compounds in a nonoccupationally exposed US population and in groups with suspected exposure. Clin Chem. 1994;40:1401–4.Google Scholar
- 12.Taipale R, Ruuskanen TM, Rinne J, Kajos MK, Hakola H, Pohja T, et al. Technical note: quantitative long-term measurements of VOC concentrations by PTR-MS—measurement, calibration, and volume mixing ratio calculation methods. Atmos Chem Phys. 2008;8:6681–98. https://doi.org/10.5194/acp-8-6681-2008.CrossRefGoogle Scholar
- 19.Horn W, Richter M, Nohr M, Wilke O, Jann O. Application of a novel reference material in an international round robin test on material emissions testing. Indoor Air. 2017; https://doi.org/10.1111/ina.12421.
- 20.Richter M, Mull B, Horn W, Brödner D, Mölders N, Renner M. Reproducibly emitting reference material on thermoplastic polyurethane basis for quality assurance/quality control of emission test chamber measurements. Build Environ. 2017;122:230–6. https://doi.org/10.1016/J.BUILDENV.2017.06.005.CrossRefGoogle Scholar
- 21.Haider SI, Asif ES. Quality control training manual: comprehensive training guide for API, finished pharmaceutical and biotechnologies laboratories. Boca Raton: CRC Press; 2011.Google Scholar
- 24.Nelson GO. Gas mixtures: preparation and control. Lewis Publishers; 1992.Google Scholar
- 25.International Organization of Legal Metrology. Evidential breath analyzers: Ethylometers. OIML R 126 2012;126.Google Scholar
- 28.Sassi G, Demichelis A, Sassi MP. Air flow rate thermal control system at low pressure drop. Flow Meas Instrum. 2014;35:44–7. https://doi.org/10.1016/j.flowmeasinst.2013.12.001.CrossRefGoogle Scholar
- 29.Demichelis A, Sassi G, Sassi MP. A handy method for reproducible and stable measurements of VOC at trace level in air. Springer International Publishing; 2014, p. 109–13. doi: https://doi.org/10.1007/978-3-319-00684-0_21.
- 30.Demichelis A, Sassi G, Sassi MP. Metrological performances of mass flow controllers for dynamic gas dilution. Accred Qual Assur. 2013:181–6. https://doi.org/10.1007/s00769-013-0974-y.
- 33.European Metrology Research Programme, ENV01. MACPoll Metrology of Chemical Pollutants in Air. 2011–2014 n.d. http://www.macpoll.eu/.
- 34.Pascale C, Guillevic M, Ackermann A, Leuenberger D, Niederhauser BC. Two generators to produce SI-traceable reference gas mixtures for reactive compounds at atmospheric levels. Meas Sci Technol. 2017; https://doi.org/10.1088/1361-6501/aa870c.
- 35.CMOS. Technology Working Principle and its Applications n.d. https://www.elprocus.com/cmos-working-principle-and-applications/. Accessed 30 Sept 2017.
- 39.International Organization for Standardization (ISO). ISO 6145-10:2002—Gas analysis—preparation of calibration gas mixtures using dynamic volumetric methods—part 10: permeation method. 2002.Google Scholar
- 40.International Organization for Standardization (ISO). ISO 6145-8:2005—Gas analysis—preparation of calibration gas mixtures using dynamic volumetric methods—part 8: diffusion method. 2005.Google Scholar
- 41.Stan H-J. Pesticide residue analysis in foodstuffs applying capillary gas chromatography with mass spectrometric detection: state-of-the-art use of modified DFG-multimethod S19 and automated data evaluation. J Chromatogr A. 2000;892:347–77. https://doi.org/10.1016/S0021-9673(00)00308-3.CrossRefGoogle Scholar