A novel four-dimensional analytical approach for analysis of complex samples
- 884 Downloads
A two-dimensional LC (2D-LC) method, based on the work of Erni and Frei in 1978, was developed and coupled to an ion mobility-high-resolution mass spectrometer (IM-MS), which enabled the separation of complex samples in four dimensions (2D-LC, ion mobility spectrometry (IMS), and mass spectrometry (MS)). This approach works as a continuous multiheart-cutting LC system, using a long modulation time of 4 min, which allows the complete transfer of most of the first - dimension peaks to the second - dimension column without fractionation, in comparison to comprehensive two-dimensional liquid chromatography. Hence, each compound delivers only one peak in the second dimension, which simplifies the data handling even when ion mobility spectrometry as a third and mass spectrometry as a fourth dimension are introduced. The analysis of a plant extract from Ginkgo biloba shows the separation power of this four-dimensional separation method with a calculated total peak capacity of more than 8700. Furthermore, the advantage of ion mobility for characterizing unknown compounds by their collision cross section (CCS) and accurate mass in a non-target approach is shown for different matrices like plant extracts and coffee.
Keywords2D-LC CCS Ginkgo biloba IM-qTOF-MS Ion mobility LC+LC
We are thankful to Agilent for the third Infinity pump system and Phenomenex for the HPLC columns.
Compliance with ethical standards
Conflicts of interest
The authors declare that they have no conflict of interest.
- 8.Marriott PJ, Wu Z, Schoenmakers P. Nomenclature and conventions in comprehensive multidimensional chromatography—an update. LCGC Eur. (2012);25Google Scholar
- 9.Hu L, Chen X, Kong L, Su X, Ye M, Zou H. Improved performance of comprehensive two-dimensional HPLC separation of traditional Chinese medicines by using a silica monolithic column and normalization of peak heights. J Chromatogr A. 2005;1092:191–8. doi: 10.1016/j.chroma.2005.06.066.CrossRefGoogle Scholar
- 14.Pringle SD, Giles K, Wildgoose JL, Williams JP, Slade SE, Thalassinos K, et al. An investigation of the mobility separation of some peptide and protein ions using a new hybrid quadrupole/travelling wave IMS/oa-ToF instrument. Int J Mass Spectrom. 2007;261:1–12. doi: 10.1016/j.ijms.2006.07.021.CrossRefGoogle Scholar
- 16.Campuzano I, Bush MF, Robinson CV, Beaumont C, Richardson K, Kim H, et al. Structural characterization of drug-like compounds by ion mobility mass spectrometry: comparison of theoretical and experimentally derived nitrogen collision cross sections. Anal Chem. 2012;84:1026–33. doi: 10.1021/ac202625t.CrossRefGoogle Scholar
- 24.May JC, Goodwin CR, Lareau NM, Leaptrot KL, Morris CB, Kurulugama RT, et al. Conformational ordering of biomolecules in the gas phase: nitrogen collision cross sections measured on a prototype high resolution drift tube ion mobility-mass spectrometer. Anal Chem. 2014;86:2107–16. doi: 10.1021/ac4038448.CrossRefGoogle Scholar
- 26.Malkar A, Devenport NA, Martin HJ, Patel P, Turner MA, Watson P, et al. Metabolic profiling of human saliva before and after induced physiological stress by ultra-high performance liquid chromatography-ion mobility-mass spectrometry. Metabolomics. 2013;9:1192–201. doi: 10.1007/s11306-013-0541-x.CrossRefGoogle Scholar
- 30.Stoll DR, Talus ES, Harmes DC, Zhang K Evaluation of detection sensitivity in comprehensive two-dimensional liquid chromatography separations of an active pharmaceutical ingredient and its degradants. Anal Bioanal Chem; 2014. 265–277. doi: 10.1007/s00216-014-8036-9.Google Scholar
- 37.Taylor L, Tichy S. Ion mobility technology for LC/MS reveal greater detail. http://www.agilent.com/cs/library/eseminars/public/Ion Separation - Innovations in Mobility Techniques.pdf. 2013 Accessed 22 Jan 2016.