Rapid measurement of perchlorate in polar ice cores down to sub-ng L−1 levels without pre-concentration
- 243 Downloads
An ion chromatography-electrospray ionization-tandem mass spectrometry (IC-ESI-MS/MS) method has been developed for rapid and accurate measurement of perchlorate in polar snow and ice core samples in which perchlorate concentrations are expected to be as low as 0.1 ng L−1. Separation of perchlorate from major inorganic species in snow is achieved with an ion chromatography system interfaced to an AB SCIEX triple quadrupole mass spectrometer operating in multiple reaction monitoring mode. Under optimized conditions, the limit of detection and lower limit of quantification without pre-concentration have been determined to be 0.1 and 0.3 ng L−1, respectively, with a linear dynamic range of 0.3–10.0 ng L−1 in routine measurement. These represent improvements over previously reported methods using similar analytical techniques. The improved method allows fast, accurate, and reproducible perchlorate quantification down to the sub-ng L−1 level and will facilitate perchlorate measurement in the study of natural perchlorate production with polar ice cores in which perchlorate concentrations are anticipated to vary in the low and sub-ng L−1 range. Initial measurements of perchlorate in ice core samples from central Greenland show that typical perchlorate concentrations in snow dated prior to the Industrial Revolution are about 0.8 ng L−1, while perchlorate concentrations are significantly higher in recent (post-1980) snow, suggesting that anthropogenic sources are a significant contributor to perchlorate in the current environment.
KeywordsIce core Ion chromatography-tandem mass spectrometry Perchlorate Pollution Ultra-trace analysis
Funding for this work was provided by US National Science Foundation (Award 1203533; Major Research Instrumentation Award 0922816 for the acquisition of AB SCIEX QTRAP 5500 tandem mass spectrometer). We would like to thank the South Dakota State University Campus Mass Spectrometry Facility and Linhong Jing for technical support and maintenance of the QTRAP 5500 mass spectrometer. We are grateful to Brian Logue, Robert Oda, Randy Jackson, Michael Stutelberg, Brendan Mitchell, and Erica Manandhar, of the Department of Chemistry and Biochemistry, South Dakota State University, for their assistance in this project.
Conflict of Interest
The authors declare that they have no conflict of interests.
- 3.National Research Council (2005) Health implications of perchlorate ingestionGoogle Scholar
- 15.Ericksen G (1983) The Chilean nitrate deposit. Am Sci 71:366–374Google Scholar
- 18.Buizert C, Gkinis V, Severinghaus JP, He F, Lecavalier BS, Kindler P, Leuenberger M, Carlson AE, Vinther B, Masson-Delmotte V, White JWC, Liu Z, Otto-Bliesner B, Brook EJ (2014) Greenland temperature response to climate forcing during the last deglaciation. Science 345:1177–1180CrossRefGoogle Scholar
- 21.McConnell JR (2002) A 250-year high-resolution record of Pb flux and crustal enrichment in central Greenland. Geophys Res LettGoogle Scholar
- 22.Hautman DP, Munch DJ, Eaton AD, Haghani AW (1999) US EPA Method 314.0Google Scholar
- 25.EPA (2008) Method 314.2, Determination of perchlorate in drinking water using two-dimensional ion chromatography with suppressed conductivity detection, EPA Document #: 815-B-08-001Google Scholar
- 27.Wagner HP, Pepich BV, Pohl C, Later D, Srinivasan K, Lin R, DeBorba B, Munch DJ (2007) Selective method for the analysis of perchlorate in drinking waters at nanogram per liter levels, using two-dimensional ion chromatography with suppressed conductivity detection. J Chromatogr A 1155:15–21CrossRefGoogle Scholar
- 29.EPA (2005) Method 331.0, Determination of perchlorate in drinking water by liquid chromatography electrospray ionization mass spectroscopy, EPA Document #: 815-R-05-007Google Scholar
- 30.EPA (2005) Method 332.0, Determination of perchlorate in drinking water by ion chromatography with suppressed conductivity and electrospray ionization mass spectrometry, EPA Document #: EPA/600/R-05/049Google Scholar
- 32.AB Sciex QTRAP 5500 System. http://www.absciex.com/products/mass-spectrometers/qtrap-lcms-systems/qtrap-5500-lcmsms-system. Accessed 25 Sep 2014
- 33.EPA (2012) Guidance for methods development and methods validation for the RCRA programGoogle Scholar