Using ICP-qMS to trace the uptake of nanoscale titanium dioxide by microalgae–potential disadvantages of vegetable reference material
- 316 Downloads
As nanoscale materials have gained in economic importance over recent years, concerns about accumulation in the environment and, consequently, analysis of nanoparticles in biological material have increasingly become the focus of scientific research. A nanomaterial used in a wide range of food, consumer and household products is titanium dioxide (nTiO2). Monitoring of nTiO2 via determination of elemental titanium (Ti) can be very challenging because of a variety of possible interferences. This work describes problems during the development of a quantification method for titanium dioxide (TiO2) using inductively coupled plasma-quadrupole mass spectrometry (ICP-qMS). To evaluate the analytical method, certified vegetable reference material NCS DC 73349 was used. Interestingly, measurements of NCS DC 73349 seemed to result in acceptable recovery values—however, this was without considering interferences or conceivable differences in the natural isotopic abundance of the certified titanium calibration solution and NCS DC 73349. Actually, recoveries were lower than initially assumed. The potential interferences causing augmented recovery could be attributed to the presence of the elements sulfur (S) and phosphorus (P), which were able to form oxide ions and nitrogen-interfering species. The effect of such interfering cluster ions could be prevented by dry ashing as a sample preparation step, to evaporate S and P, before digestion with aqua regia in a high-pressure asher (HPA). Final practicability of the analysis method was proved by monitoring the uptake of nTiO2 by the microalgae Scenedesmus acutus in an environmental exposure study.
KeywordsNanoscale titanium dioxide Algae Scenedesmus acutus Matrix interferences High-pressure asher (HPA) Inductively coupled plasma-quadrupole mass spectrometer (ICP-qMS)
We would like to thank Evonik Industries for providing Aeroxide TiO2 P25 and the OECD for participation in the OECD sponsorship programme.
- 1.Steinbach C, Förster A Knowledge base nanomaterials, DaNa, Frankfurt am Main. URL: http://www.nanoobjects.info/cms/lang/en/Wissensbasis/Titandioxid
- 14.Packer AP, Larivière D, Li C, Chen M, Fawcett A, Nielsen K, Mattson K, Chatt A, Scriver C, Erhardt LS (2007) Validation of an inductively coupled plasma mass spectrometry (ICP-MS) method for the determination of cerium, strontium, and titanium in ceramic materials used in radiological dispersal devices (RDDs). Anal Chim Acta 588:166–172CrossRefGoogle Scholar
- 17.Organisation for Economic Cooperation and Development (OECD) (2011) Guidelines for the testing of chemicals. Freshwater Alga and Cyanobacteria, Growth Inhibition Test (Guideline 201, Annex 5 corrected: 28 July 2011). OECD, ParisGoogle Scholar
- 18.Organisation for Economic Cooperation and Development (OECD) (2008) Guidelines for the testing of chemicals. Daphnia magna reproduction test (Guideline 211, adopted 3st October 2008). OECD, ParisGoogle Scholar
- 19.DIN 32645 (2008) Chemical analysis - Decision limit, detection limit and determination limit under repeatability conditions - Terms, methods, evaluation. Deutsches Institut für Normung. Beuth Verlag, BerlinGoogle Scholar
- 21.Vaughan M-A, Baines AD, Templeton DM (1991) Multielement analysis of biological samples by inductively coupled plasma-mass spectrometry. II. Rapid survey method for profiling trace elements in body fluids. Clin Chem 37:210–215Google Scholar
- 24.Marshall J, Franks J (1990) Multielement analysis and reduction of spectral interferences using electrothermal vaporization inductively coupled plasma-mass spectrometry. At Spectrosc 11:177–186Google Scholar