Performance verification of a photogrammetric scanning system for micro-parts using a three-dimensional artifact: adjustment and calibration
- 17 Downloads
Performance verification is a fundamental issue to assure the traceability of a measurement instrument. This issue is very important for non-contact 3D scanning systems, also for the limited number of existing standards. There are many factors affecting the process in a photogrammetric scanning system, and they have to be considered during a performance verification. In this context, it is crucial to completely define the camera model and its spatial locations and orientations during the scan, estimating the intrinsic and extrinsic parameters. There are two main proposals in this paper. Firstly, authors investigated on how the reliability of the adjustment procedure could be improved, adopting a more complex geometry of the reference object, using a three-dimensional one instead of a bi-dimensional pattern of targets. Secondly, an approach to a calibration procedure for photogrammetric scanning system has been drawn up and applied, using the same artifact used for the adjustment. The ISO 15530-3 2011 standard was adopted for the uncertainty assessment of the results.
KeywordsMicro-scanning Calibration Adjustment Photogrammetry 3D image processing Uncertainty
Unable to display preview. Download preview PDF.
The authors would like to thank Ph.D. Marta Pesce for her valuable contribution to the first concept of this paper.
- 3.Qin Y, Brockett A, Ma Y et al (2009) Micro-manufacturing: research, technology outcomes and development issues. Int J Adv Manuf Technol 47(9–12):821–837Google Scholar
- 16.Percoco G, Modica F, Fanelli S (2016) Image analysis for 3D micro-features: a new hybrid measurement method. Precis Eng 48:123–132. https://doi.org/10.1016/j.precisioneng.2016.11.012 CrossRefGoogle Scholar
- 21.Mendricky R (2016) Determination of measurement accuracy of optical 3D scanners. 1565–1572. https://doi.org/10.17973/MMSJ.2016
- 22.Bernal C, De Agustina B, Marín MM, Camacho AM (2014) Accuracy analysis of fringe projection systems based on blue light technology. Key Eng Mater 615:9–14. https://doi.org/10.4028/www.scientific.net/KEM.615.9 CrossRefGoogle Scholar
- 26.Brown DC (1971) Close-range camera calibration. Photogramm Eng 37:855–866Google Scholar
- 29.Sims-Waterhouse D, Bointon P, Piano S, Leach RK (2017) Experimental comparison of photogrammetry for additive manufactured parts with and without laser speckle projection. 103290W. https://doi.org/10.1117/12.2269507
- 30.Lavecchia F, Guerra MG, Galantucci LM (2017) The influence of software algorithms on photogrammetric micro-feature measurement’s uncertainty. Int J Adv Manuf Technol 1–15. https://doi.org/10.1007/s00170-017-0786-z, 93
- 31.JCGM (Joint Committee for Guides in Metrology)200:2008- International Vocabulary of Metrology – Basic and General Concepts and Associated Terms (VIM), 2008Google Scholar
- 32.Neuschaefer-rube U, Neugebauer M, Dziomba T, et al (2013) New developments of measurement standards and procedures for micro and nanometrology at the PtbGoogle Scholar
- 35.Remondino F (2006) Detectors and descriptors for photogrammetric applications. Int Arch Photogramm Remote Sens Spat Inf Sci 2–7Google Scholar
- 36.Lowe DG (2004) Distinctive image features from scale-invariant keypoints. Int J Comput Vis 60:91–110. https://doi.org/10.1023/B:VISI.0000029664.99615.94 CrossRefGoogle Scholar
- 37.(2016) Agisoft Photoscan User Manual. http://www.agisoft.com/pdf/photoscan-pro_1_2_en.pdf
- 38.MountainsMap Software. http://www.digitalsurf.com/en/mntkey.html
- 39.ISO 5436-1:2000 Geometrical Product Specifications (GPS) – Surface texture: Profile method; Measurement standards – Part 1: Material measuresGoogle Scholar
- 40.JCFGIM. JIOSGI 2008;50:134. Evaluation of measurement data — Guide to the expression of uncertainty in measurementGoogle Scholar
- 41.ISO 15530-3:2011 Geometrical product specifications (GPS) -- Coordinate measuring machines (CMM): Technique for determining the uncertainty of measurement -- Part 3: Use of calibrated workpieces or measurement standardsGoogle Scholar
- 42.UNI EN ISO 4288:2000 Geometrical Product Specifications (GPS), Surface texture: Profile method, Rules and procedures for the assessment of surface textureGoogle Scholar
- 43.Accreditation AA for L. G104 - Guide for Estimation of Measurement Uncertainty In Testing December 2014. Am Assoc Lab Accredit 2014:1–31Google Scholar