3D additive manufacture of oral and maxillofacial surgical models for preoperative planning
- 740 Downloads
This paper investigates the errors generated during the fabrication stage for producing complex anatomical replicas derived from computed tomography coupled with the 3D additive manufacturing methods. Based on this research work, it is shown that patient-specific model based on computed tomography data can be converted into computer numerically controlled G-code. It is shown that 3D extrusion-based additive manufacturing technology is accurate to ±3 % equating to ±200 μm surface deviations due to plastic shrinkage and distortion formed during the process. Polylactic acid plastic extrusion through a 200-μm bore nozzle has resulted in a model being produced with an Ra roughness of 35.5 μm. An evaluation on the errors generated during the fabrication process has been used to accurately produce an adult female mandible. Internal nerve channels and complex external bone geometry has been produced within the model. It is shown that using this process results in bone complexity and accuracy required for producing low-cost surgical grades models which is in comparison with traditional selective laser sintering manufacturing techniques. The surface accuracies suggest that the reproduction of anatomically complex representative structures by 3D plastic extrusion additive manufacturing which can be used for pre-surgical planning.
Keywords3D additive manufacture Oral and maxillofacial surgery Medical models 3D printing
Unable to display preview. Download preview PDF.
- 2.Jacobs P (1995) Stereolithography and other RP & M technologies from rapid prototyping to rapid tooling. ASME pressGoogle Scholar
- 3.Barker TM, Earwalker WJ, Frost H, Wakeley G (1993) Integration of 3D medical imaging and rapid prototyping to create stereolithographic models. Aust Phys Eng Sci Med 16:79–85Google Scholar
- 5.Milovanovic J (2005) Possibilities of using selective laser melting for tire mould manufacturing, Proceedings. Second international Conference on Manufacturing Engineering Icmen, Kallithea of Chalkidiki, Greece 187-93Google Scholar
- 6.Tukuru N, Gowda S, Mansoor KP, Ahmed S, Badami S (2008) Rapid prototype technique in medical field research. J Pharm Tech 1(4):341–344Google Scholar
- 7.Udupa JK, Goncalves RJ (1996) Imaging transforms for volume visualisation, in computer integrated surgery. Comput Integr Surg: Technol Clin Appl 3:33–57Google Scholar
- 8.Swalens B, Kruth JP (1993) Medical application of rapid prototyping techniques. Fourth International conference on rapid prototyping 107-20Google Scholar
- 10.Robiony M, Salvo I, Costa F, Zerman N, Bazzocchi M, Toso F, Bandera C, Filippi S, Felice M, Politi M (2007) Virtual reality surgical planning for maxillofacial distraction osteogenesis: the role of reverse engineering rapid prototyping and cooperative work. J Oral Maxillofac Surg 65(6):1198–1208CrossRefGoogle Scholar
- 12.Gibson I (2006) Advanced manufacturing technology for medical applications- reverse engineering, software conversion and rapid prototyping. Wiley 254Google Scholar
- 16.Kouhi E, Masood S, Morsi Y (2008) Design and fabrication of reconstructive mandibular models using fused deposition modelling. Rapid Prototyp J 28(3):246–254Google Scholar
- 17.Katatny IE, Masood SH, Morsi YS (2009) Evaluation and validation of the shape accuracy of FDM fabricated medical models. Adv Mater Res 275:83–86Google Scholar