Measuring diaphragm thickness with ultrasound in mechanically ventilated patients: feasibility, reproducibility and validity
- 3.4k Downloads
Ultrasound measurements of diaphragm thickness (T di) and thickening (TFdi) may be useful to monitor diaphragm activity and detect diaphragm atrophy in mechanically ventilated patients. We aimed to establish the reproducibility of measurements in ventilated patients and determine whether passive inflation by the ventilator might cause thickening apart from inspiratory effort.
Five observers measured T di and TFdi in 96 mechanically ventilated patients. The probe site was marked in 66 of the 96 patients. TFdi was measured at peak and end-inspiration (airway occluded and diaphragm relaxed) in nine healthy volunteers inhaling to varying lung volumes. The association with diaphragm electrical activity was quantified.
Right hemidiaphragm thickness was obtained on 95 % of attempts; left hemidiaphragm measurements could not be obtained consistently. Right hemidiaphragm thickness measurements were highly reproducible (mean ± SD 2.4 ± 0.8 mm, repeatability coefficient 0.2 mm, reproducibility coefficient 0.4 mm), particularly after marking the location of the probe. TFdi measurements were only moderately reproducible (median 11 %, IQR 3–17 %, repeatability coefficient 17 %, reproducibility coefficient 16 %). TFdi and diaphragm electrical activity were positively correlated, r 2 = 0.32, p < 0.01). At inspiratory volumes below 50 % of inspiratory capacity, passive inflation did not cause diaphragm thickening. TFdi was considerably lower in patients on either partially assisted or controlled ventilation compared to healthy subjects (median 11 vs. 35 %, p < 0.001).
Ultrasound measurements of right hemidiaphragm thickness are feasible and highly reproducible in ventilated patients. At clinically relevant inspiratory volumes, diaphragm thickening reflects muscular contraction and not passive inflation. This technique can be reliably employed to monitor diaphragm thickness, activity, and function during mechanical ventilation.
KeywordsDiaphragm Artificial respiration Ultrasonography Ventilator weaning
This study was funded by an operating grant from the Physician Services Incorporated Foundation (Toronto, Canada). E.G. is supported by a Post-Doctoral Fellowship from the Canadian Institutes of Health Research.
Conflicts of interest
On behalf of all the authors, the corresponding author states that there is no conflict of interest.
- 4.Meade MO, Cook DJ, Guyatt GH et al (2008) Ventilation strategy using low tidal volumes, recruitment maneuvers, and high positive end-expiratory pressure for acute lung injury and acute respiratory distress syndrome: a randomized controlled trial. JAMA 299:637–645. doi: 10.1001/jama.299.6.637 CrossRefPubMedGoogle Scholar
- 14.ATS/ERS statement on respiratory muscle testing. Am J Respir Crit Care Med. 2002;166:518–624. doi: 10.1164/rccm.166.4.518
- 25.Goligher EC, Detsky ME, Farias P, Murray A et al (2010) Validity and reproducibility of sonographic evaluation of diaphragm structure and activity in mechanically ventilated patients. Am J Respir Crit Care Med 181:A6346Google Scholar
- 33.McAlinden C, Khadka J, Pesudovs K (2011) Statistical methods for conducting agreement (comparison of clinical tests) and precision (repeatability or reproducibility) studies in optometry and ophthalmology. Ophthalmic Physiol Opt 31:330–338. doi: 10.1111/j.1475-1313.2011.00851.x CrossRefPubMedGoogle Scholar