Mechanical influences on fluid leakage past the tracheal tube cuff in a benchtop model
- 265 Downloads
High-volume low-pressure (HVLP) cuffs on endotracheal tubes do not fully protect the lower airway from leakage of potentially contaminated secretions down the longitudinal folds within the cuff. Here, our purpose was to evaluate potential effects of positive end-expiratory pressure (PEEP), inspiratory effort intensity, and tube characteristics on fluid leakage past the cuff.
This benchtop study at a research laboratory used a tracheal tube inserted into an artificial Plexiglas trachea connected to a ventilator and lung model. Methylene blue was deposited above the tube cuff to simulate subglottic secretions. Five PEEP levels (0, 5, 10, 15, and 20 cmH2O) were tested with volume-controlled ventilation and three simulated inspiratory effort levels with pressure-support ventilation. Several cuff materials and tube sizes were tested.
The leakage occurrence rate ranged from 91% with zero PEEP to 8% with 15 and 20 cmH2O PEEP and was indirectly proportional to the PEEP level with significant correlation (R 2 = 0.39, p < 0.001), an effect not explained by higher peak inspiratory pressure. Low, moderate, and high inspiratory effort intensities were associated with 38%, 46%, and 75% leakage rates, respectively (p = 0.024). Leakage flow was considerably less with polyurethane than with polyvinylchloride tubes (mean 0.5 versus 31.8 ml/h). Leakage increased with larger tube diameters.
This benchtop study shows that PEEP and a polyurethane cuff prevent leakage past the endotracheal tube cuff, whereas greater inspiratory effort and larger tube diameters for given tracheal size induce or worsen leakage.
KeywordsEndotracheal tube Benchtop model PEEP Ventilator-associated pneumonia Pressure-support ventilation
- 3.(2005) Guidelines for the management of adults with hospital-acquired, ventilator-associated, and healthcare-associated pneumonia. Am J Respir Crit Care Med 171:388–416Google Scholar
- 10.Ouanes I, Rana N, Lyazidi A, Thille A, Brochard L (2008) Influences mécaniques sur la fuite liquidienne autour de la sonde d’intubation: “étude sur banc”. Réanimation 17(Suppl 1):S1–S240Google Scholar
- 14.Lucangelo U, Zin WA, Antonaglia V, Petrucci L, Viviani M, Buscema G, Borelli M, Berlot G (2008) Effect of positive expiratory pressure and type of tracheal cuff on the incidence of aspiration in mechanically ventilated patients in an intensive care unit. Crit Care Med 36:409–413CrossRefPubMedGoogle Scholar
- 18.Orozco-Levi M, Torres A, Ferrer M, Piera C, el-Ebiary M, de la Bellacasa JP, Rodriguez-Roisin R (1995) Semirecumbent position protects from pulmonary aspiration but not completely from gastroesophageal reflux in mechanically ventilated patients. Am J Respir Crit Care Med 152:1387–1390PubMedGoogle Scholar