Non-invasive ventilation is largely used to treat acute and chronic respiratory failure. This ventilation encounters a non-negligible rate of failure related to the used interface/mask, but the reasons for this failure remain unclear. In order to shed light on this issue and to better understand the effects of the geometrical design of interfaces, we aimed to quantify flow, pressure and gas composition in terms of CO2 and O2 at the passage through different types of interface (oronasal mask, integral mask and helmet). In particular, we postulated that due to specific gas flow passing throughout the interface, the effective dead space added by the interface is not always related to the whole gas volume included in the interface.
Numerical simulations, using computational fluid dynamics, were used to describe pressure, flow and gas composition during ventilation with the different interfaces.
Between the different interfaces the effective dead spaces differed only modestly (110–370 ml), whereas their internal volumes were markedly different (110–10,000 ml). Effective dead space was limited to half the tidal volume for the most voluminous interface, whereas it was close to the interface gas volume for the less voluminous interfaces. Pressure variations induced by the flow ventilation throughout the interface were negligible.
Effective dead space is not related to the internal gas volume included in the interface, suggesting that this internal volume should not be considered as a limiting factor for their efficacy during non-invasive ventilation. Patient’s comfort and synchrony have also to be taken into account.
This is a preview of subscription content, log in to check access.
Buy single article
Instant unlimited access to the full article PDF.
Price includes VAT for USA
Tarabini Fraticelli A, Lellouche F, L’Her E, Taille S, Mancebo J, Brochard L (2009) Physiological effects of different interfaces during noninvasive ventilation for acute respiratory failure. Crit Care Med 37:939–945
Mehta S, Hill NS (2001) Noninvasive ventilation. Am J Respir Crit Care Med 163:540–577
Brochard L, Mancebo J, Wysocki M, Lofaso F, Conti G, Rauss A, Simonneau G, Benito S, Gasparetto A, Lemaire F et al (1995) Noninvasive ventilation for acute exacerbations of chronic obstructive pulmonary disease. N Engl J Med 333:817–822
Nava S, Ambrosino N, Clini E, Prato M, Orlando G, Vitacca M, Brigada P, Fracchia C, Rubini F (1998) Noninvasive mechanical ventilation in the weaning of patients with respiratory failure due to chronic obstructive pulmonary disease. A randomized, controlled trial. Ann Intern Med 128:721–728
Carlucci A, Richard JC, Wysocki M, Lepage E, Brochard L (2001) Noninvasive versus conventional mechanical ventilation. An epidemiologic survey. Am J Respir Crit Care Med 163:874–880
Girault C, Briel A, Hellot MF, Tamion F, Woinet D, Leroy J, Bonmarchand G (2003) Noninvasive mechanical ventilation in clinical practice: a 2-year experience in a medical intensive care unit. Crit Care Med 31:552–559
Robino C, Faisy C, Diehl JL, Rezgui N, Labrousse J, Guerot E (2003) Effectiveness of non-invasive positive pressure ventilation differs between decompensated chronic restrictive and obstructive pulmonary disease patients. Intensive Care Med 29:603–610
Chiumello D (2006) Is the helmet different than the face mask in delivering noninvasive ventilation? Chest 129:1402–1403
Cuvelier A, Pujol W, Pramil S, Molano LC, Viacroze C, Muir JF (2009) Cephalic versus oronasal mask for noninvasive ventilation in acute hypercapnic respiratory failure. Intensive Care Med 35:519–526
Antonelli M, Conti G, Pelosi P, Gregoretti C, Pennisi MA, Costa R, Severgnini P, Chiaranda M, Proietti R (2002) New treatment of acute hypoxemic respiratory failure: noninvasive pressure support ventilation delivered by helmet—a pilot controlled trial. Crit Care Med 30:602–608
Bellani G, Patroniti N, Greco M, Foti G, Pesenti A (2008) The use of helmets to deliver non-invasive continuous positive airway pressure in hypoxemic acute respiratory failure. Minerva Anestesiol 74:651–656
Chiumello D, Pelosi P, Carlesso E, Severgnini P, Aspesi M, Gamberoni C, Antonelli M, Conti G, Chiaranda M, Gattinoni L (2003) Noninvasive positive pressure ventilation delivered by helmet vs standard face mask. Intensive Care Med 29:1671–1679
Mortimore IL, Whittle AT, Douglas NJ (1998) Comparison of nose and face mask CPAP therapy for sleep apnoea. Thorax 53:290–292
Navalesi P, Fanfulla F, Frigerio P, Gregoretti C, Nava S (2000) Physiologic evaluation of noninvasive mechanical ventilation delivered with three types of masks in patients with chronic hypercapnic respiratory failure. Crit Care Med 28:1785–1790
Pelosi P, Severgnini P, Aspesi M, Gamberoni C, Chiumello D, Fachinetti C, Introzzi L, Antonelli M, Chiaranda M (2003) Non-invasive ventilation delivered by conventional interfaces and helmet in the emergency department. Eur J Emerg Med 10:79–86
Racca F, Appendini L, Gregoretti C, Stra E, Patessio A, Donner CF, Ranieri VM (2005) Effectiveness of mask and helmet interfaces to deliver noninvasive ventilation in a human model of resistive breathing. J Appl Physiol 99:1262–1271
Taccone P, Hess D, Caironi P, Bigatello LM (2004) Continuous positive airway pressure delivered with a “helmet”: effects on carbon dioxide rebreathing. Crit Care Med 32:2090–2096
Vargas F, Thille A, Lyazidi A, Campo FR, Brochard L (2009) Helmet with specific settings versus facemask for noninvasive ventilation. Crit Care Med 37:1921–1928
Lellouche F, Maggiore SM, Deye N, Taille S, Pigeot J, Harf A, Brochard L (2002) Effect of the humidification device on the work of breathing during noninvasive ventilation. Intensive Care Med 28:1582–1589
Saatci E, Miller DM, Stell IM, Lee KC, Moxham J (2004) Dynamic dead space in face masks used with noninvasive ventilators: a lung model study. Eur Respir J 23:129–135
Anton A, Tarrega J, Giner J, Guell R, Sanchis J (2003) Acute physiologic effects of nasal and full-face masks during noninvasive positive-pressure ventilation in patients with acute exacerbations of chronic obstructive pulmonary disease. Respir Care 48:922–925
Gregoretti C, Confalonieri M, Navalesi P, Squadrone V, Frigerio P, Beltrame F, Carbone G, Conti G, Gamna F, Nava S, Calderini E, Skrobik Y, Antonelli M (2002) Evaluation of patient skin breakdown and comfort with a new face mask for non-invasive ventilation: a multi-center study. Intensive Care Med 28:278–284
Colaizy TT, Younis UM, Bell EF, Klein JM (2008) Nasal high-frequency ventilation for premature infants. Acta Paediatr 97:1518–1522
Fauroux B, Leroux K, Desmarais G, Isabey D, Clement A, Lofaso F, Louis B (2008) Performance of ventilators for noninvasive positive-pressure ventilation in children. Eur Respir J 31:1300–1307
Louis B, Leroux K, Isabey D, Fauroux B, Lofaso F (2009) Effect of manufacturers inserted mask leaks on ventilator performance. Eur Respir J 35:627–636
This article is discussed in the editorial available at: doi:10.1007/s00134-010-2072-5.
About this article
Cite this article
Fodil, R., Lellouche, F., Mancebo, J. et al. Comparison of patient–ventilator interfaces based on their computerized effective dead space. Intensive Care Med 37, 257–262 (2011). https://doi.org/10.1007/s00134-010-2066-3
- Non-invasive ventilation
- Dead space