Volume-targeted modes of modern neonatal ventilators: how stable is the delivered tidal volume?
Volume-targeted modes are designed to deliver a constant tidal volume (Vt) at lowest possible pressure independently of changes in compliance, resistance, and leak of the respiratory system. We examined whether these volume-targeted modes respond rapidly enough to sudden changes in respiratory mechanics (e.g., selective intubation, surfactant administration, endotracheal tube kinking, de-kinking, obstruction), resulting in insufficient or excessive Vt delivery.
Design and setting
Bench study of six neonatal ventilators in the volume-targeted mode simulating preterm and full-term infant settings on a test lung.
Measurements and results
Breath-to-breath expiratory Vt were measured after rapid compliance, resistance, and leak changes. Under our test settings all ventilators showed important volume overshooting following rapid increase in compliance or decrease in resistance. Between one and 16 inflations were required to return to the set Vt. Some ventilators delivered inaccurate Vt under steady state condition while others showed considerable breath-to-breath Vt variability.
We observed inaccurate Vt delivery under specific conditions as well as immediate and sometimes prolonged volume overshooting after a rapid respiratory system compliance increase or resistance decrease in volume-targeted modes of modern neonatal ventilators. Similar discrepancies between the set Vt and the delivered inflations can be harmful in clinical situations, especially in newborns. Their clinical relevance needs to be clarified with safety studies in the neonatal population and we encourage manufacturers to further improve the ventilators algorithms.
KeywordsMechanical ventilation Neonate Full-term Preterm Volume-targeted ventilation
- 5.Lista G, Marangione P, Azzali A, Castoldi F, Pogliani L, Compagnoni G (2000) [The “guaranteed volume” in pressure support ventilation reduces the risk of barotrauma in premature children with severe respiratory syndrome]. Acta Biomed Ateneo Parmense (Suppl 1) 71:453–456Google Scholar
- 10.Mrozek JD, Bendel-Stenzel EM, Meyers PA, Bing DR, Connett JE, Mammel MC (2000) Randomized controlled trial of volume-targeted synchronized ventilation and conventional intermittent mandatory ventilation following initial exogenous surfactant therapy. Pediatr Pulmonol 29:11–18PubMedCrossRefGoogle Scholar
- 15.McCallion N, Davis PG, Morley CJ (2004) Volume-targeted versus pressure-limited ventilation in the neonate. Cochrane Database of Systematic Reviews, Cochrane LibraryGoogle Scholar
- 17.Collins E, Faciane V, Abbruzzese L (2005) Comparison of Bird Vip Gold ventilator in conventional volume mode to Drager Babylog 8000 Plus in volume guarantee mode a bench study. http://www.viasyshealthcare.com/prod_serv/downloads/045_Volume_Accuracy_Monograph.pdf; accessed 18 September