A novel approach to SBRT patient quality assurance using EPID-based real-time transit dosimetry
- 52 Downloads
Intra- and inter-fraction organ motion is a major concern in stereotactic body radiation therapy (SBRT). It may cause substantial differences between the planned and delivered dose distribution. Such delivery errors may lead to medical harm and reduce life expectancy for patients. The project presented here investigates and improves a rapid method to detect such errors by performing online dose verification through the analysis of electronic portal imaging device (EPID) images.
To validate the method, a respiratory phantom with inhomogeneous insert was examined under various scenarios: no-error and error-simulated measurements. Simulation of respiratory motions was practiced for target ranges up to 2 cm. Three types of treatment planning technique – 3DCRT (three-dimensional conformal radiation therapy), IMRT (intensity modulated radiation therapy), and VMAT (volumetric modulated arc therapy – were generated for lung SBRT. A total of 54 plans were generated to assess the influence of techniques on the performance of portal dose images. Subsequently, EPID images of 52 SBRT patients were verified. Both for phantom and patient cases, dose distributions were compared using the gamma index method according to analysis protocols in the target volume.
The comparison of error-introduced EPID-measured images to reference images showed no significant differences with 3%/3 mm gamma evaluation, though target coverage was strongly underestimated. Gamma tolerance of 2%/2 mm reported noticeable detection in EPID sensitivity for simulated errors in 3DCRT and IMRT techniques. The passing rates for 3DCRT, IMRT, and VMAT with 1%/1 mm in open field were 84.86%, 92.91%, and 98.75%, and by considering MLC-CIAO + 1 cm (threshold 5%), were 68.25%, 83.19%, and 95.29%, respectively.
This study demonstrates the feasibility of EPID for detecting the interplay effects. We recommend using thin computed tomography slices and adding sufficient tumor margin in order to limit the dosimetric organ motion in hypofractionated irradiation with preserved plan quality. In the presence of respiratory and gastrointestinal motion, tighter criteria and consequently using local gamma evaluation should be considered, especially for VMAT. This methodology offers a substantial step forward in in vivo dosimetry and the potential to distinguish errors depending on the gamma tolerances. Thus, the approach/prototype provides a fast and easy quality assurance procedure for treatment delivery verification.
KeywordsElectronic portal imaging device Stereotactic body radiation therapy Quality assurance In vivo dosimetry Real time transit dosimetry
Compliance with ethical guidelines
Conflict of interest
C. Moustakis, F. Ebrahimi Tazehmahalleh, K. Elsayad, F. Fezeu, and S. Scobioala declare that they have no competing interests.
For this article, no studies with human participants or animals were performed by any of the authors. All procedures performed were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
- 3.Sveistrup J, Rosenschold PM, Deasy JO, Oh JH, Pommer T, Petersen PM et al (2014) Improvement in toxicity in high risk prostate cancer patients treated with image-guided intensity-modulated radiotherapy compared to 3D conformal radiotherapy without daily image guidance. Radiat Oncol 9:44PubMedPubMedCentralCrossRefGoogle Scholar
- 4.Chung HT, Lee B, Park E, Lu JJ, Xia P (2008) Can all centers plan intensity-modulated radiotherapy (IMRT) effectively? An external audit of dosimetric comparisons between three-dimensional conformal radiotherapy and IMRT for adjuvant chemoradiation for gastric cancer. Int J Radiat Oncol Biol Phys 71(4):1167–1174PubMedCrossRefGoogle Scholar
- 27.International Atomic Energy Agency (2012) Development of procedures for in vivo dosimetry in radiotherapy. Human health report no. 8. IAEA, ViennaGoogle Scholar
- 30.Seuntjens J, Lartigau EF, Cora S, Ding GX, Goetsch S, Nuyttens J et al (2014) Prescribing, recording, and reporting of stereotactic treatments with small photon beams. J ICRU 14:1–160Google Scholar
- 41.Chan MKH, Leung RWK, Lee VWY, Wong MYP, Chiang CL, Law GML et al (2019) Linking dose delivery accuracy and planning target margin in radiosurgery based on dose-volume histograms derived from measurement-guided dose reconstruction. Phys Med Biol 64(4):045009. https://doi.org/10.1088/1361-6560/aafd47 PubMedCrossRefGoogle Scholar