Fusion of Preoperative MRI and Postoperative FD-CT for Direct Evaluation of Cochlear Implants
- 29 Downloads
This study was carried out to evaluate the diagnostic value of merging preoperative magnetic resonance imaging (MRI) with postoperative flat-panel computed tomography (FD-CT) and compare it to standard postoperative FD-CT for assessment of cochlear implant (CI) insertion.
The T2-weighted (T2w) constructive interference in steady state (CISS) data sets of preoperative 1.5 T and 3 T MRI scans of CI patients with both regular and adverse implant spiralization were co-registered with the corresponding postoperative FD-CT data sets using defined anatomic landmarks. These merged FD-CT/MRI volumes (CMV) were compared to the corresponding postoperative FD-CT MPRs in consensus reading with respect to qualitative, i.e. scala tympani spiralization, scala vestibuli spiralization, scalar translocation and quantitative, i.e. distance of the last electrode to the lateral cochlea wall (D1) distance of the 2nd/5th electrode to the basal cochlear wall (D2) and the transition point (TP) of the scalar translocation, parameters.
In total 30 patients (n 1.5T MRI = 18 patients; n 3T MRI = 12 patients) were included in the analysis. In all cases both CMVs and FD-CT MPRs were generated. Qualitative analysis of intracochlear CI position with CMVs (both 1.5 T and 3 T) and FD-CT was equivalent: In 20 patients the CI showed a regular implant spiralization, in 10 cases a scalar translocation was identified with both CMVs and FD-CT. Quantitative analysis showed a high level of congruency between CMVs (both 1.5 T and 3T) and FD-CT for fusion accuracy (D1: mean FD-CT D1 = 1.30 ± 0.7 mm; mean CMV D1 = 1.27 ± 0.77 mm, correlation r = 0.94, p < 0.0001; D2: mean FD-CT D2 = 1.17 ± 0.34 mm; mean CMV D2 = 1.10 ± 0.31 mm, correlation r = 0.89, p < 0.0001) and TP of the scalar translocation (mean FD-CT = 126.0 ± 59.25°, mean CMV = 117.0 ± 52.82°, correlation r = 0.95, p < 0.0001).
The co-registration of preoperative 1.5 and 3 T MRI with postoperative FD-CT enables a direct evaluation of the position of a CI equivalent to the current standard FD-CT. Despite the fact that CMV provided no additional diagnostic value in this series, regardless whether preoperative 1.5 or 3 T MRI was used for co-registration, it might help to simplify postoperative CI diagnostics.
KeywordsScalar translocation Merging Cochlear implant Flat panel computed tomography Magnetic resonance imaging
Compliance with ethical guidelines
Conflict of interest
F. Eisenhut, L. Taha, I. Kleibe, J. Hornung, H. Iro, A. Doerfler and S. Lang declare that they have no competing interests.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1975 Helsinki declaration and its later amendments or comparable ethical standards. Informed consent was obtained from all individual participants included in the study.
- 6.Booth TN, Wick C, Clarke R, Kutz JW, Medina M, Gorsage D, Xi Y, Isaacson B. Evaluation of the Normal Cochlear Second Interscalar Ridge Angle and Depth on 3D T2-Weighted Images: a Tool for the Diagnosis of Scala Communis and Incomplete Partition Type II. AJNR Am J Neuroradiol. 2018;39:923–7.CrossRefPubMedGoogle Scholar
- 9.Struffert T, Hertel V, Kyriakou Y, Krause J, Engelhorn T, Schick B, Iro H, Hornung J, Doerfler A. Imaging of cochlear implant electrode array with flat-detector CT and conventional multislice CT: comparison of image quality and radiation dose. Acta Otolaryngol. 2010;130:443–52.CrossRefPubMedGoogle Scholar
- 16.Verbist BM, Skinner MW, Cohen LT, Leake PA, James C, Boëx C, Holden TA, Finley CC, Roland PS, Roland JT Jr, Haller M, Patrick JF, Jolly CN, Faltys MA, Briaire JJ, Frijns JH. Consensus panel on a cochlear coordinate system applicable in histologic, physiologic, and radiologic studies of the human cochlea. Otol Neurotol. 2010;31:722–30.CrossRefPubMedPubMedCentralGoogle Scholar