Development of the CAS-LIBB single-particle microbeam for localized irradiation of living cells
A single-particle microbeam facility has been constructed at the Key Laboratory of Ion Beam Bioengineering (LIBB), Chinese Academy of Sciences (CAS). The system was designed to deliver a defined numbers of hydrogen ions, produced by a van de Graaff accelerator, in an energy range of 2.0–3.0 MeV, into an area smaller than that of the nucleus of an individual living cell. The beam is collimated by a borosilicate glass capillary that forms the beam-line exit. An integrated computer program recognizes the cells and locates them one by one over the microbeam exit for irradiation. We present technical details of the CAS-LIBB microbeam facility, particularly on the collimator, hardware, control program, as well as cell irradiation protocols available. Various factors contributing to the targeting and positioning precision are discussed along with accuracy measurement results.
Keywordsmicrobeam facility single-particle irradiation particle detection precision target location accuracy living cells
Unable to display preview. Download preview PDF.
- 1.Zirkle, R. E., Advances in biological and medical physics, New York: Academic Press, 1957, 103–146.Google Scholar
- 8.Wu, L. J., Hei, T. K., Randers-Pehrson, G. et al., Columbia University microbeam: Development of an experimental system for targeting cells indiviually with counted particles, Nucl. Sci. & Tech. 1999, 10(3): 143–148.Google Scholar
- 9.Michelet, C., Moretto, Ph., Barberet, Ph. et al., A focused microbeam for targeting cells with counted multiple particles, Radiat. Res., 2002, 158: 370–371.Google Scholar
- 15.Brenner, D. J., Hall, E. J., Microbeams: A potent mix of physics and biology, Radiat. Prot. Dosim, 2002, 99: 283–286.Google Scholar
- 16.Yu, Z. L., Ion beam and biology science, Physics, 1997, 26: 333–338.Google Scholar
- 21.Geard, C. R., Randers-Pehrson, G., Marino, S. A. et al., Intra- and intercellular responses after cell site-specific microbeam irradiation, Radiat. Res., 2000, 153: 233.Google Scholar
- 24.Wang, X. H., Wang, S. H., Yu, Z. L., ·· Image feature extracting of cells for the ASIPP microbeam,· · Comput. Engin. (in Chinese), 2003, 29 (18): 9–10.··Google Scholar
- 25.Wang, X. H., Wang, S. H., Yu, Z. L., District partition in image acquiring f cells and image distinguishing, Comput. Engin. (in Chinese), 2003, 29 (20): 35–37.Google Scholar
- 26.Peng, S. X., Wang, S. H., Yu, Z. L., Ion Micro-beam Manipulation: Deposition of given number of particles into the pre-determined position of individual cells, Vac. Sci & Technol. (in Chinese), 1999, 18: 38–46. ··Google Scholar
- 27.Zhan, F. R., Yu, Z. L.,··The calculation of the ion trajectory in an accelerating tube,·· High Power Laser and Particle Beams (in Chinese), 2000, 12(1): 103–106.Google Scholar
- 28.Zhan, F. R., Yu, Z. L., Hu, C. D. et al.,··Construction of electrostatic accelerator RF ion source, Vac. Sci. & Technol. (in Chinese), 2000, 20(3): 226–228.Google Scholar