Rapidly evolving angioplasty balloon techniques need continual evaluation for the type or frequency of balloon-related complications. We recently encountered a case where a new narrow shaft polyethylene terephthalate balloon completely separated from the catheter following circumferential rupture. The balloon was successfully retrieved using grasping forceps and a basket. Patterns of balloon rupture and subsequent management are discussed. Problems of this type were more frequent in the early days of balloon angioplasty, but have since become rare. Close surveillance of new low profile balloons is recommended to determine whether this represents an isolated occurrence or a return to the earlier experience.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Yune HY, Klatte EC (1980) Circumferential tear of percutaneous transluminal angioplasty catheter balloon. AJR 135:395–396
Tegtmeyer CJ, Bezirdjian DR (1981) Removing the stuck, ruptured angioplasty balloon catheter. Radiology 139:231–232
Simpfendorfer CC, Dimas AP, Zaidi A, Hollman J, Franco I, Knezinek V, Hobbs RE (1986) Balloon rupture during coronary angioplasty. Angiology 37:828–831
Angelini P (1990) Balloon catheter coronary angioplasty: Balloon rupture. Cathet Cardiovasc Diagn 20:150–151
Bredlau CE, Roubin GS, Leimgruber PP, Douglas JS Jr, King SB III, Gruentzig AR (1985) In-hospital morbidity and mortality in patients undergoing elective coronary angioplasty. Circulation 72:1044–1052
Gardiner GA, Meyerovitz MF, Stokes KR, Clouse ME, Harrington DP, Bettmann MA (1986) Complications of transluminal angioplasty. Radiology 159:201–208
Rothschild R, Voda J (1990) Coronary artery dissection caused by angioplasty balloon rupture. Cathet Cardiovasc Diagn 19:26–29
LeMay MR, Beanlands DS (1990) Pinhole balloon rupture during coronary angioplasty causing rupture of the coronary artery. Cathet Cardiovasc Diagn 19:91–92
Zollikofer CL, Salomonowitz E, Castaneda-Zuniga WR, Bruhlmann WF, Amplatz K (1985) The relation between arterial and balloon rupture in experimental angioplasty. AJR 144:777–779
Rizzo TF, Werres R, Ciccone J, Karanam R, Shah S (1988) Entrapment of an angioplasty balloon catheter: A case report. Cathet Cardiovasc Diagn 14:255–257
Weinhaus L, Lababidi Z (1987) Catheter rupture during balloon valvuloplasty. Am Heart J 113:1035–1036
Quinn SF, Hallin R (1990) Endovascular retention of a ruptured polyester balloon in femoral angioplasty. Letter to the Editor. AJR 155:1141–1142
About this article
Cite this article
Selby, J.B., Oliva, V.L. & Tegtmeyer, C.J. Circumferential rupture of an angioplasty balloon with detachment from the shaft: Case report. Cardiovasc Intervent Radiol 15, 113–116 (1992). https://doi.org/10.1007/BF02734104
- Transluminal catheters, catheterization