Development of a platinized platinum/iridium electrode for use in vitro
Silver/silver chloride (Ag/AgCl) electrodes possess excellent electrical properties for measuring the electrical activity of gastrointestinal smooth muscle but exert toxic effects on this tissue in vitro. We thus developed a platinum electrode for use in vitro, the construction of these electrodes relying upon the formation of a glass-platinum/iridium seal. The platinum/iridium (Pt/Ir) electrodes were platinized using a current density of 0.45 mA mm−2. The electrode impedance at 0.01 Hz showed a minimum with platinization current-time products greater than 500 mA s mm−2. However, deposits in excess of 600 mA s mm−2 were readily removed by mechanical abrasion and proved unsatisfactory. Optimal platinization was obtained with a deposit of platinum-black corresponding to a current-time product of 550 mA s mm−2. Optimally-platinized electrodes (geometric surface area 0.11 mm2) had a stable and reproducible potential with a drift of less than 1 μV min−1 and a lower impedance than optimally chlorided silver electrodes (geometric surface area 0.46 mm2) at frequencies higher than 0.25 Hz. The platinized Pt/Ir electrodes were used to record the electrical activity of gastrointestinal smooth muscle in vitro.
KeywordsPlatinum/iridium electrodes Biological electrodes Platinization
Unable to display preview. Download preview PDF.
- 4.de Boer, R.W. and van Oosterom. Electrical properties of platinum electrodes: impedance measurements and time-domain analysis.Med. & Biol. Eng. & Comput. 16:1–10, 1978.Google Scholar
- 6.Fisher, G., R.G. Sayre, R.G. Bickford. Histological changes in cat's brain after introduction of metallic and coated wire used in electroencephalography.Proc. Mayo Clic. 32:14–21, 1957.Google Scholar
- 10.Jones, G. and G.M. Bollinger. The measurement of the conductance of electrolytes. VII on platinization.J. Am. Chem. Soc. 57:280–284, 1935.Google Scholar
- 11.Kingma, Y.J., J. Lenhart, N.G. Durdle, K.L. Bowes, M.M. Chambers. Improved Ag/AgCl pressure electrodes.Med. & Biol. Eng. & Comput. 21:351–357, 1983.Google Scholar
- 12.Kohlrausch, F. Uber platinirte Elektroden und Widerstandsbestimmung.Ann. Phys. Chem. 60:315–332, 1897.Google Scholar
- 13.Lukanov, J. and P. Atmadjov. Investigating the effect of silver ions on the contractile function of smooth-muscle preparation from guinea pig stomach in vitro.Folia Med. (Plovdiv) 21:11–19, 1979.Google Scholar
- 14.Onaral, B. and H.P. Schwan. Linear and onlinear properties of platinum electrode polarization. Part 1: frequency dependence at very low frequencies.Med. & Biol. Eng. & Comput. 20:299–306, 1982.Google Scholar
- 15.Petering, H.G. Pharmacology and toxicology of heavy metals: Silver.Pharmacol. Ther. 1:127–130, 1976.Google Scholar
- 16.Ray, C.D. and A.E. Walker. Platinized platinum wire brain depth probes: a comparison with electrodes of stainless steel.Proc. Am. EEG Soc. 19:90–91, 1965.Google Scholar
- 17.Schwan, H.P. Determination of biological impedances. In: Physical Techniques in Biological Research, Vol. 6. Nastuk, W.L. (Ed.), New York: Academic Press, 323–407, 1963.Google Scholar
- 19.Warburg, E. Ueber das Verhalten sogenanten und polisierbarer Electroden gegen Wechselstrom.Ann. Phy. Chem. 67:493–499, 1899.Google Scholar