Accuracy of an optically isolated tetra-polar impedance measurement system
- 84 Downloads
Accurate electrical transfer impedance measurement at the high frequencies (>1 MHz) required to characterise blood and intracellular structures is very difficult, owing to stray capacitances between lead wires. To solve this problem, an optically isolated measurement system has been developed using a phaselocked-loop technique for synchronisation between current injection (drive) and voltage measurement (receive) circuits. The synchronisation error between drive and receive circuits was less than 1 ns. The accuracy and reproducibility of the developed system was examined using a tissue equivalent Cole model consisting of two resistors and one capacitor. The absolute value Z and phase shift θ in impedance of the Cole model was measured at 1.25 MHz by both an LCR meter and the isolated measurement system. The difference between the values measured by the isolated measurement system and those measured by the LCR meter was less than 0.27 Ω (2.9%) in Z and 0.79 degree in θ. The standard deviation was less than 0.09Ω in Z and 0.60 degree in θ.
KeywordsImpedance measurement Biological tissue Stray capacitance Optical isolation Phase locked loop
Unable to display preview. Download preview PDF.
- Brown, B. H., Tidy, J. A., Boston, K., Blackett, A. D., Smallwood, R. H., andSharp, F. (2000): ‘Relation between tissue structure and imposed electrical current flow in cervical neoplasia’,Lancet,355, pp. 892–895Google Scholar
- Cole, K. S., andCole, R. H. (1941): ‘Dispersion and absorption in dielectrics’,J. Chem. Phys.,9, pp. 341–351Google Scholar
- Lu, L., Hamzaoui, L., Brown, B. H., Rigaud, B., Smallwood, R. H., Barber, D. C., andMorucci, J. P. (1996): ‘Parametric modelling for electrical impedance spectroscopy system’,Med. Biol. Eng. Comput.,34, pp. 122–126Google Scholar
- McAdams E. T., Henry, P., andAnderson J. M. C. C. (1992): ‘Optimal electrolytic chloriding of silver ink electrodes’,Clin. Phys. Physiol. Meas.,13, pp. 19–23Google Scholar