Accurate measurement of three-dimensional deformations in deformable and rigid bodies using computer vision
- 1.5k Downloads
Recently, digital-image-correlation techniques have been used to accurately determine two-dimensional in-plane displacements and strains. An extension of the two-dimensional method to the acquisition of accurate, three-dimensional surfacedisplacement data from a stereo pair of CCD cameras is presented in this paper.
A pin-hole camera model is used to express the transformation relating three-dimensional world coordinates to two-dimensional computer-image coordinates by the use of camera extrinsic and intrinsic parameters. Accurate camera model parameters are obtained for each camera independently by (a) using several points which have three-dimensional world coordinates that are accurate within 0.001 mm and (b) using two-dimensional image-correlation methods that are accurate to within 0.05 pixels to obtain the computer-image coordinates of various object positions. A nonlinear, least-squares method is used to select the optimal camera parameters such that the deviations between the measured and estimated image positions are minimized.
Using multiple orientations of the cameras, the accuracy of the methodology is tested by performing translation tests. Using theoretical error estimates, error analyses are presented. To verify the methodology for actual tests both the displacement field for a cantilever beam and also the surface, three-dimensional displacement and strain fields for a 304L stainless-steel compact-tension specimen were experimentally obtained using stereo vision. Results indicate that the three-dimensional measurement methodology, when combined with two-dimensional digital correlation for subpixel accuracy, is a viable tool for the accurate measurement of surface displacements and strains.
KeywordsStereo Vision Camera Parameter Camera Model Stereo Pair Digital Correlation
Unable to display preview. Download preview PDF.
- 1.Riley, W.F. and Durelli, A.J., “Application of Moiré Methods to the Determination of Transient Stress and Strain Distributions,” J. Appl. Mech.,29 (1), (1962). 0416 0737 V 3Google Scholar
- 2.Oplinger, D.W., “Application of Moiré Methods to Evaluation of Structural Performance of Composite Materials,”Opt. Eng.,21 (4),626–632 (1982).Google Scholar
- 3.Ligtenberg, F.K., “The Moiré Method: A New Experimental method for the Determination of Moments in Small Slab Models,”Soc. for Exp. Stress Analysis,12 (2),83–98 (1955).Google Scholar
- 4.Kao, T.Y. andChiang, F.P., “Family of Grating Techniques of Slope and Curvature Measurements for Static and Dynamic Flexure of Plates,”Opt. Eng.,21 (4),721–742 (1982).Google Scholar
- 5.Pirodda L., “Shadow and Projection Moiré Techniques for Absolute or Relative Mapping of Surface Shapes,”Opt. Eng.,21 (4),640–649 (1982).Google Scholar
- 6.Haines, K.A. andHildebrand, B.P., “Surface-Deformation Measurement Using the Wavefront Reconstruction Technique,”Appl. Opt.,5 (4),595–602 (1966).Google Scholar
- 7.Gottenburg, W.G., “Some Applications of Holographic Interferometry,”Experimental Mechanics,8,281–285 (1969).Google Scholar
- 8.Post, D., Moiré Interferometry,” Handbook on Experimental Mechanics, ed. A.S. Kobayashi, Chapt. 8, Prentice-Hall (1987).Google Scholar
- 10.Archbold, E., Burch, J.M. andEnnos, A.E., “Recording of in-Plane Surface Displacements by Double Exposure Speckle Photography”,J. for Physics in Eng.: Scientific Instruments,4,277–279 (1971).Google Scholar
- 11.Stetson, K.A., “A Review of Speckle Photography and Interferometry,”Opt. Eng.,14, (5),482–489 (1975).Google Scholar
- 13.Peters, W.H., Ranson, W.F., “Digital Image Techniques in Experimental Stress Analysis,”Opt. Eng.,21, (3),427–431 (1982).Google Scholar
- 17.Sutton, M.A., McNeill, S.R., Jang, J. andBabai, M.K., “Effects of Subpixel Image Restoration on Digital Correlation Error Estimates,”Opt. Eng.,27 (10),870–877 (1988).Google Scholar
- 18.He, Y., Tan, Wan, Y., Ku C. and Smith, E.W., “3-D Displacement of the Crack Tip Measured by Microimage Plane Holography and Specklegraphy Separating Technique”, Proc. 1989 SEM Spring Conf. on Exp. Mech., 301–305 (1989).Google Scholar
- 20.Sobel, I.E., “Camera Models and Machine Perception,” Stanford Artificial Intelligence Project, PhD Thesis, Stanford Univ. (1970).Google Scholar
- 22.Parrish, E.A. andGoksel, A.K., “A Camera Model for Natural Scene Selection,”9,131–136 (1977).Google Scholar
- 24.Yakimovsky, Y., “A System for Extracting 3D Measurements from a Stereo Pair of TV Cameras,”Computer Graphics 7,195–210 (1978).Google Scholar
- 26.Tsai, R.Y., “A Versatile Camera Calibration Technique for High Accuracy 3D Machine Vision Metrology Using Off-the-Shelf TV Cameras and Lenses,”IEEE J. of Robotics and Automation,RA-3 (4),323–344 (1987).Google Scholar
- 27.Tsai, R.Y., “Accuracy Analysis and Prediction for 3D Robotics Vision Metrology,” IBM Research Rep. RC 11348.Google Scholar
- 28.IMSL Subroutine ZXSSQ, Problem Solving Software System for Mathematical and Statistical FORTRAN Programming, Edition 9.2 (1984).Google Scholar
- 30.Timoshenko, S.P. and Goodier, J.N., Theory of Elasticity, McGraw-Hill Book Company, Third Ed. (1970).Google Scholar
- 31.Luo, P.F., Chao, Y.J., Sutton, M.A., and Peters, W.H., “Optimal Arranbgements of the Stereo Imaging System for the Measurement of Three-Dimensional Displacement,”, Proc. of the XV Southeastern Conf. on Theoretical and Applied Mechanics, 789–795 (1990).Google Scholar
- 32.ASTM Metals Test Methods and Analytical Procedures, 03 01 (1990).Google Scholar
- 33.Sutton, M.A., Turner, J.L., Bruck, H.A., andChae, T.L., “Experimental Investigations of Three-Dimensional Effects Near a Crack Tip Using Computer Vision”,Int. J. of Fracture Mechanics,53,201–228 (1992).Google Scholar
- 34.Chae, T., Bruck, H.A., Sutton, M.A. andTurner, J.L., “Full-field Representation of Discretely Sampled Surface Deformation for Displacement and Strain Analysis,”Experimental Mechanics,31, (2),168–177 (1991).Google Scholar